-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan  5 22:39:35 2024 UTC
-- File 20230720_174047_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.813822 to 25.194317
-each separately from 12s to 25s and from 194028us to 814111us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.379917 in 1 packets: 18
MIU packet time from 2023-07-20T17:41:02.813.822.000 to 2023-07-20T17:41:15.194.317.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.379917 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.8860949 to 26.8860496
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999547 in 1 packets: 18
Temperature FPGA: 
      FE -> 57.6 degC in 36 packets (all packets)
Temperature XO: 
      C6 -> 45.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 1 packets: 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230720_174047_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.6999194 to 14.7002084
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.3798710  -- OK for SL --
   MIU time from 2023-07-20T17:41:02.813.822.000 to 2023-07-20T17:41:02.814.111.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.379917

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.0800794 to 27.0803684
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-20T17:41:15.194.028.000 to 2023-07-20T17:41:15.194.317.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.699919400 corrected by 40bit 10MHz counter of 13.886094900 obtained at 1Hz counter of 12. Result: 12.813827348
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-07-20T17:41:02.813.681.725
   Max 1st sample MIU time among all SL packets: 2023-07-20T17:41:15.194.176.725
   Min 1st sample XO time among all SL packets: 12.813687073
   Max 1st sample XO time among all SL packets: 25.194179185
   TM values from -333 to 242 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-07-20T17:41:02.813.681.725 (XO: 12.813687073 - 12.813993074, t=45.2 degC)  Rank: 00076
   TM values: Max =   242  Min =  -333  Ave =    -4.33  Std =    29.63
   Min TM value:-333 at an offset of 129.913 usec -> 2023-07-20T17:41:02.813.811.637 (XO: 12.813816986)
   Max TM value:242 at an offset of 132.350 usec -> 2023-07-20T17:41:02.813.814.075 (XO: 12.813819423)
   Max dTM/dt [1/us]:2.85e+03 at an offset of 132.200 usec -> 2023-07-20T17:41:02.813.813.925 (XO: 12.813819273)
   Min dTM/dt [1/us]:-2.62e+03 at an offset of 129.800 usec -> 2023-07-20T17:41:02.813.811.525 (XO: 12.813816873)

 - Block number: 2 starting at 2023-07-20T17:41:15.193.887.725 (XO: 25.193890184 - 25.194196185, t=45.2 degC)  Rank: 00088
   TM values: Max =   226  Min =  -248  Ave =    -4.48  Std =    31.83
   Min TM value:-248 at an offset of 159.101 usec -> 2023-07-20T17:41:15.194.046.825 (XO: 25.194049285)
   Max TM value:226 at an offset of 132.888 usec -> 2023-07-20T17:41:15.194.020.612 (XO: 25.194023072)
   Max dTM/dt [1/us]:1.95e+03 at an offset of 159.213 usec -> 2023-07-20T17:41:15.194.046.938 (XO: 25.194049397)
   Min dTM/dt [1/us]:-2.09e+03 at an offset of 133.038 usec -> 2023-07-20T17:41:15.194.020.762 (XO: 25.194023222)
