-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jan  5 22:11:57 2024 UTC
-- File 20230720_160547_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177031
-each separately from 0s to 14s and from 0us to 841255us.
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833944 in 13 packets: 2, 3, 4, 5, 6, 8, 10, 11, 12, 14, 15, 16, 17
      0.833939 in 2 packets: 7, 13
      0.833943 in 1 packets: 9
MIU packet time from 2023-07-20T16:05:50.000.000.000 to 2023-07-20T16:06:04.177.031.000
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833944 in 13 packets: 2, 3, 4, 5, 6, 8, 10, 11, 12, 14, 15, 16, 17
      0.833939 in 2 packets: 7, 13
      0.833943 in 1 packets: 9
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.3293179 to 15.3292660
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999962 in 2 packets: 2, 16
      0.9999964 in 1 packets: 3
      0.9999963 in 11 packets: 4, 5, 6, 8, 9, 10, 11, 12, 14, 15, 17
   XO PPM: 
      -3.8 in 2 packets: 2, 16
      -3.6 in 1 packets: 3
      -3.7 in 11 packets: 4, 5, 6, 8, 9, 10, 11, 12, 14, 15, 17
Temperature FPGA: 
      FF -> 56.3 degC in 7 packets: 1, 2, 7, 8, 9, 12, 17
      FE -> 57.6 degC in 11 packets: 3, 4, 5, 6, 10, 11, 13, 14, 15, 16, 18
Temperature XO: 
      C6 -> 45.2 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 2 packets: 1, 2
      1 in 7 packets: 3, 4, 5, 6, 7, 8, 9
      2 in 9 packets: 10, 11, 12, 13, 14, 15, 16, 17, 18
Number of generated alerts: 
      0 in 2 packets: 1, 2
      1 in 7 packets: 3, 4, 5, 6, 7, 8, 9
      2 in 9 packets: 10, 11, 12, 13, 14, 15, 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230720_160547_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.3293180 to 15.5062980
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-20T16:05:50.000.000.000 to 2023-07-20T16:06:04.177.031.000
      Difference between neigboring packets: 
      0.833938 in 1 packets: 1
      0.833944 in 13 packets: 2, 3, 4, 5, 6, 8, 10, 11, 12, 14, 15, 16, 17
      0.833939 in 2 packets: 7, 13
      0.833943 in 1 packets: 9
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.329318000 corrected by 40bit 10MHz counter of 1.329317900 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996200 - 0.999996400
   Min 1st sample MIU time among all SR packets: 2023-07-20T16:05:49.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-07-20T16:06:04.177.030.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177032580
   TM values from -43 to 36 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-07-20T16:05:49.999.999.925 (XO: 0.000000025 - 0.000017025, t=45.2 degC)
   TM values: Max =    23  Min =   -30  Ave =    -4.21  Std =    11.28

 - Packet number: 2 starting at 2023-07-20T16:05:50.833.937.925 (XO: 0.833943194 - 0.833960194, t=45.2 degC)
   TM values: Max =    28  Min =   -34  Ave =    -4.31  Std =    12.19

 - Packet number: 3 starting at 2023-07-20T16:05:51.667.881.925 (XO: 1.667886229 - 1.667903229, t=45.2 degC)
   TM values: Max =    24  Min =   -34  Ave =    -4.48  Std =    11.29

 - Packet number: 4 starting at 2023-07-20T16:05:52.501.825.925 (XO: 2.501829282 - 2.501846282, t=45.2 degC)
   TM values: Max =    13  Min =   -20  Ave =    -4.47  Std =     6.59

 - Packet number: 5 starting at 2023-07-20T16:05:53.335.769.925 (XO: 3.335772367 - 3.335789367, t=45.2 degC)
   TM values: Max =    29  Min =   -36  Ave =    -4.42  Std =    13.00

 - Packet number: 6 starting at 2023-07-20T16:05:54.169.713.925 (XO: 4.169715453 - 4.169732453, t=45.2 degC)
   TM values: Max =    23  Min =   -33  Ave =    -4.49  Std =    11.33

 - Packet number: 7 starting at 2023-07-20T16:05:55.003.657.925 (XO: 5.003658539 - 5.003675539, t=45.2 degC)
   TM values: Max =    17  Min =   -25  Ave =    -4.26  Std =     8.94

 - Packet number: 8 starting at 2023-07-20T16:05:55.837.596.925 (XO: 5.837601624 - 5.837618624, t=45.2 degC)
   TM values: Max =    20  Min =   -32  Ave =    -4.36  Std =     9.16

 - Packet number: 9 starting at 2023-07-20T16:05:56.671.540.925 (XO: 6.671544710 - 6.671561710, t=45.2 degC)
   TM values: Max =    15  Min =   -24  Ave =    -4.46  Std =     7.74

 - Packet number: 10 starting at 2023-07-20T16:05:57.505.483.925 (XO: 7.505487795 - 7.505504795, t=45.2 degC)
   TM values: Max =    12  Min =   -20  Ave =    -4.44  Std =     4.89

 - Packet number: 11 starting at 2023-07-20T16:05:58.339.427.925 (XO: 8.339430881 - 8.339447881, t=45.2 degC)
   TM values: Max =    15  Min =   -23  Ave =    -4.43  Std =     7.27

 - Packet number: 12 starting at 2023-07-20T16:05:59.173.371.925 (XO: 9.173373966 - 9.173390967, t=45.2 degC)
   TM values: Max =    16  Min =   -26  Ave =    -4.41  Std =     7.47

 - Packet number: 13 starting at 2023-07-20T16:06:00.007.315.925 (XO: 10.007317052 - 10.007334052, t=45.2 degC)
   TM values: Max =    25  Min =   -32  Ave =    -4.29  Std =    11.11

 - Packet number: 14 starting at 2023-07-20T16:06:00.841.254.925 (XO: 10.841260138 - 10.841277138, t=45.2 degC)
   TM values: Max =    17  Min =   -27  Ave =    -4.42  Std =     8.39

 - Packet number: 15 starting at 2023-07-20T16:06:01.675.198.925 (XO: 11.675203223 - 11.675220223, t=45.2 degC)
   TM values: Max =    22  Min =   -29  Ave =    -4.68  Std =    12.15

 - Packet number: 16 starting at 2023-07-20T16:06:02.509.142.925 (XO: 12.509146360 - 12.509163360, t=45.2 degC)
   TM values: Max =    19  Min =   -27  Ave =    -4.26  Std =     8.16

 - Packet number: 17 starting at 2023-07-20T16:06:03.343.086.925 (XO: 13.343089494 - 13.343106494, t=45.2 degC)
   TM values: Max =    36  Min =   -43  Ave =    -4.40  Std =    17.84

 - Packet number: 18 starting at 2023-07-20T16:06:04.177.030.925 (XO: 14.177032580 - 14.177049580, t=45.2 degC)
   TM values: Max =    25  Min =   -31  Ave =    -4.56  Std =    11.26
