-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Jan  4 21:58:09 2024 UTC
-- File 20230720_094621_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177029
-each separately from 0s to 14s and from 0us to 841254us.
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833944 in 11 packets: 2, 3, 5, 6, 8, 10, 11, 12, 14, 16, 17
      0.833943 in 3 packets: 4, 9, 15
      0.833939 in 2 packets: 7, 13
MIU packet time from 2023-07-20T09:46:24.000.000.000 to 2023-07-20T09:46:38.177.029.000
   Differences in neigboring packets: 
      0.833938 in 1 packets: 1
      0.833944 in 11 packets: 2, 3, 5, 6, 8, 10, 11, 12, 14, 16, 17
      0.833943 in 3 packets: 4, 9, 15
      0.833939 in 2 packets: 7, 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.7195527 to 15.7195029
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999965 in 6 packets: 2, 4, 6, 11, 14, 16
      0.9999964 in 8 packets: 3, 5, 8, 9, 10, 12, 15, 17
   XO PPM: 
      -3.5 in 6 packets: 2, 4, 6, 11, 14, 16
      -3.6 in 8 packets: 3, 5, 8, 9, 10, 12, 15, 17
Temperature FPGA: 
      00 -> 55.0 degC in 15 packets: 1, 2, 4, 5, 6, 7, 8, 9, 10, 13, 14, 15, 16, 17, 18
      01 -> 53.7 degC in 1 packets: 3
      FF -> 56.3 degC in 2 packets: 11, 12
Temperature XO: 
      C5 -> 43.6 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 4 packets: 1, 2, 3, 4
      1 in 14 packets: 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
Number of generated alerts: 
      0 in 4 packets: 1, 2, 3, 4
      1 in 14 packets: 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230720_094621_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.7195528 to 15.8965328
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-20T09:46:24.000.000.000 to 2023-07-20T09:46:38.177.029.000
      Difference between neigboring packets: 
      0.833938 in 1 packets: 1
      0.833944 in 11 packets: 2, 3, 5, 6, 8, 10, 11, 12, 14, 16, 17
      0.833943 in 3 packets: 4, 9, 15
      0.833939 in 2 packets: 7, 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.719552800 corrected by 40bit 10MHz counter of 1.719552700 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SR packets: 2023-07-20T09:46:23.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-07-20T09:46:38.177.028.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177030462
   TM values from -13 to 5 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-07-20T09:46:23.999.999.925 (XO: 0.000000025 - 0.000017025, t=43.6 degC)
   TM values: Max =     3  Min =   -13  Ave =    -4.41  Std =     1.95

 - Packet number: 2 starting at 2023-07-20T09:46:24.833.937.925 (XO: 0.833942944 - 0.833959944, t=43.6 degC)
   TM values: Max =     3  Min =   -11  Ave =    -4.43  Std =     1.97

 - Packet number: 3 starting at 2023-07-20T09:46:25.667.881.925 (XO: 1.667885929 - 1.667902929, t=43.6 degC)
   TM values: Max =     4  Min =   -11  Ave =    -4.37  Std =     2.00

 - Packet number: 4 starting at 2023-07-20T09:46:26.501.825.925 (XO: 2.501828881 - 2.501845881, t=43.6 degC)
   TM values: Max =     2  Min =   -10  Ave =    -4.48  Std =     2.03

 - Packet number: 5 starting at 2023-07-20T09:46:27.335.768.925 (XO: 3.335771834 - 3.335788834, t=43.6 degC)
   TM values: Max =     3  Min =   -12  Ave =    -4.48  Std =     2.19

 - Packet number: 6 starting at 2023-07-20T09:46:28.169.712.925 (XO: 4.169714819 - 4.169731819, t=43.6 degC)
   TM values: Max =     5  Min =   -13  Ave =    -4.19  Std =     2.04

 - Packet number: 7 starting at 2023-07-20T09:46:29.003.656.925 (XO: 5.003657738 - 5.003674738, t=43.6 degC)
   TM values: Max =     1  Min =   -10  Ave =    -4.41  Std =     1.92

 - Packet number: 8 starting at 2023-07-20T09:46:29.837.595.925 (XO: 5.837600740 - 5.837617740, t=43.6 degC)
   TM values: Max =     4  Min =   -10  Ave =    -4.40  Std =     2.07

 - Packet number: 9 starting at 2023-07-20T09:46:30.671.539.925 (XO: 6.671543743 - 6.671560743, t=43.6 degC)
   TM values: Max =     3  Min =   -11  Ave =    -4.44  Std =     2.03

 - Packet number: 10 starting at 2023-07-20T09:46:31.505.482.925 (XO: 7.505486745 - 7.505503745, t=43.6 degC)
   TM values: Max =     3  Min =   -12  Ave =    -4.23  Std =     1.87

 - Packet number: 11 starting at 2023-07-20T09:46:32.339.426.925 (XO: 8.339429713 - 8.339446713, t=43.6 degC)
   TM values: Max =     3  Min =   -10  Ave =    -4.53  Std =     2.07

 - Packet number: 12 starting at 2023-07-20T09:46:33.173.370.925 (XO: 9.173372649 - 9.173389649, t=43.6 degC)
   TM values: Max =     2  Min =   -11  Ave =    -4.24  Std =     1.97

 - Packet number: 13 starting at 2023-07-20T09:46:34.007.314.925 (XO: 10.007315651 - 10.007332651, t=43.6 degC)
   TM values: Max =     1  Min =   -10  Ave =    -4.52  Std =     1.99

 - Packet number: 14 starting at 2023-07-20T09:46:34.841.253.925 (XO: 10.841258569 - 10.841275569, t=43.6 degC)
   TM values: Max =     5  Min =   -10  Ave =    -4.32  Std =     2.07

 - Packet number: 15 starting at 2023-07-20T09:46:35.675.197.925 (XO: 11.675201556 - 11.675218556, t=43.6 degC)
   TM values: Max =     4  Min =   -12  Ave =    -4.50  Std =     2.06

 - Packet number: 16 starting at 2023-07-20T09:46:36.509.140.925 (XO: 12.509144507 - 12.509161507, t=43.6 degC)
   TM values: Max =     3  Min =   -10  Ave =    -4.39  Std =     1.79

 - Packet number: 17 starting at 2023-07-20T09:46:37.343.084.925 (XO: 13.343087460 - 13.343104460, t=43.6 degC)
   TM values: Max =     2  Min =   -10  Ave =    -4.47  Std =     1.84

 - Packet number: 18 starting at 2023-07-20T09:46:38.177.028.925 (XO: 14.177030462 - 14.177047462, t=43.6 degC)
   TM values: Max =     5  Min =   -12  Ave =    -4.37  Std =     1.97
