-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jul  9 05:44:13 2023 UTC
-- File 20230704_032328_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000096 to 15.011289
-each separately from 0s to 15s and from 96us to 11289us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.010615 in 1 packets: 18
MIU packet time from 2023-07-04T03:23:31.000.096.000 to 2023-07-04T03:23:46.011.289.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.010615 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9755899 to 16.9755343
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999444 in 1 packets: 18
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230704_032328_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.9756875 to 1.9759765
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.0105590  -- OK for SL --
   MIU time from 2023-07-04T03:23:31.000.096.000 to 2023-07-04T03:23:31.000.385.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.010615

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      1 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.9865355 to 16.9868245
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-04T03:23:46.011.000.000 to 2023-07-04T03:23:46.011.289.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.975687500 corrected by 40bit 10MHz counter of 1.975589900 obtained at 1Hz counter of 0. Result: 0.000097600
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996200 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2023-07-04T03:23:30.999.955.725
   Max 1st sample MIU time among all SL packets: 2023-07-04T03:23:46.011.148.725
   Min 1st sample XO time among all SL packets: -0.000042675
   Max 1st sample XO time among all SL packets: 15.011149967
   TM values from -5 to -3 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-07-04T03:23:30.999.955.725 (XO: -0.000042675 - 0.000263326, t=37.2 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -5  Ave =    -4.20  Std =     0.41
   Min TM value:-5 at an offset of 0.100 usec -> 2023-07-04T03:23:30.999.955.825 (XO: -0.000042575)
   Max TM value:-3 at an offset of 42.650 usec -> 2023-07-04T03:23:30.999.998.375 (XO: -0.000000025)
   Max dTM/dt [1/us]:1.50e+01 at an offset of 106.163 usec -> 2023-07-04T03:23:31.000.061.887 (XO: 0.000063488)
   Min dTM/dt [1/us]:-1.40e+01 at an offset of 223.138 usec -> 2023-07-04T03:23:31.000.178.863 (XO: 0.000180463)

 - Block number: 2 starting at 2023-07-04T03:23:46.010.859.725 (XO: 15.010860966 - 15.011166967, t=37.2 degC)  Rank: 00001
   TM values: Max =    -3  Min =    -5  Ave =    -4.18  Std =     0.39
   Min TM value:-5 at an offset of 0.038 usec -> 2023-07-04T03:23:46.010.859.762 (XO: 15.010861004)
   Max TM value:-3 at an offset of 5.663 usec -> 2023-07-04T03:23:46.010.865.387 (XO: 15.010866629)
   Max dTM/dt [1/us]:1.30e+01 at an offset of 150.963 usec -> 2023-07-04T03:23:46.011.010.688 (XO: 15.011011929)
   Min dTM/dt [1/us]:-1.60e+01 at an offset of 14.300 usec -> 2023-07-04T03:23:46.010.874.025 (XO: 15.010875266)
