-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jul  9 05:37:13 2023 UTC
-- File 20230703_212733_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177033
-each separately from 0s to 14s and from 0us to 841258us.
   Differences in neigboring packets: 
      0.833940 in 2 packets: 1, 13
      0.833943 in 4 packets: 2, 6, 9, 17
      0.833944 in 10 packets: 3, 4, 5, 8, 10, 11, 12, 14, 15, 16
      0.833941 in 1 packets: 7
MIU packet time from 2023-07-03T21:27:36.000.000.000 to 2023-07-03T21:27:50.177.033.000
   Differences in neigboring packets: 
      0.833940 in 2 packets: 1, 13
      0.833943 in 4 packets: 2, 6, 9, 17
      0.833944 in 10 packets: 3, 4, 5, 8, 10, 11, 12, 14, 15, 16
      0.833941 in 1 packets: 7
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.0893674 to 16.0893132
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999962 in 4 packets: 2, 8, 10, 16
      0.9999961 in 10 packets: 3, 4, 5, 6, 9, 11, 12, 14, 15, 17
   XO PPM: 
      -3.8 in 4 packets: 2, 8, 10, 16
      -3.9 in 10 packets: 3, 4, 5, 6, 9, 11, 12, 14, 15, 17
Temperature FPGA: 
      03 -> 51.1 degC in 5 packets: 1, 10, 13, 14, 17
      04 -> 49.8 degC in 12 packets: 2, 3, 4, 5, 6, 7, 8, 9, 12, 15, 16, 18
      05 -> 48.5 degC in 1 packets: 11
Temperature XO: 
      C1 -> 37.2 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230703_212733_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.0893675 to 16.2663475
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-07-03T21:27:36.000.000.000 to 2023-07-03T21:27:50.177.033.000
      Difference between neigboring packets: 
      0.833940 in 2 packets: 1, 13
      0.833943 in 4 packets: 2, 6, 9, 17
      0.833944 in 10 packets: 3, 4, 5, 8, 10, 11, 12, 14, 15, 16
      0.833941 in 1 packets: 7
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.089367500 corrected by 40bit 10MHz counter of 2.089367400 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996100 - 0.999996200
   Min 1st sample MIU time among all SR packets: 2023-07-03T21:27:35.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-07-03T21:27:50.177.032.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177034915
   TM values from -6 to -3 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-07-03T21:27:35.999.999.925 (XO: 0.000000025 - 0.000017025, t=37.2 degC)
   TM values: Max =    -3  Min =    -6  Ave =    -4.48  Std =     0.52

 - Packet number: 2 starting at 2023-07-03T21:27:36.833.939.925 (XO: 0.833943194 - 0.833960194, t=37.2 degC)
   TM values: Max =    -3  Min =    -5  Ave =    -4.43  Std =     0.50

 - Packet number: 3 starting at 2023-07-03T21:27:37.667.882.925 (XO: 1.667886430 - 1.667903430, t=37.2 degC)
   TM values: Max =    -4  Min =    -6  Ave =    -4.52  Std =     0.50

 - Packet number: 4 starting at 2023-07-03T21:27:38.501.826.925 (XO: 2.501829682 - 2.501846682, t=37.2 degC)
   TM values: Max =    -4  Min =    -5  Ave =    -4.44  Std =     0.50

 - Packet number: 5 starting at 2023-07-03T21:27:39.335.770.925 (XO: 3.335772935 - 3.335789935, t=37.2 degC)
   TM values: Max =    -3  Min =    -5  Ave =    -4.39  Std =     0.49

 - Packet number: 6 starting at 2023-07-03T21:27:40.169.714.925 (XO: 4.169716187 - 4.169733187, t=37.2 degC)
   TM values: Max =    -4  Min =    -6  Ave =    -4.46  Std =     0.50

 - Packet number: 7 starting at 2023-07-03T21:27:41.003.657.925 (XO: 5.003659439 - 5.003676439, t=37.2 degC)
   TM values: Max =    -3  Min =    -5  Ave =    -4.25  Std =     0.45

 - Packet number: 8 starting at 2023-07-03T21:27:41.837.598.925 (XO: 5.837602608 - 5.837619608, t=37.2 degC)
   TM values: Max =    -3  Min =    -5  Ave =    -4.32  Std =     0.47

 - Packet number: 9 starting at 2023-07-03T21:27:42.671.542.925 (XO: 6.671545844 - 6.671562844, t=37.2 degC)
   TM values: Max =    -4  Min =    -5  Ave =    -4.48  Std =     0.50

 - Packet number: 10 starting at 2023-07-03T21:27:43.505.485.925 (XO: 7.505489046 - 7.505506046, t=37.2 degC)
   TM values: Max =    -4  Min =    -5  Ave =    -4.49  Std =     0.50

 - Packet number: 11 starting at 2023-07-03T21:27:44.339.429.925 (XO: 8.339432249 - 8.339449249, t=37.2 degC)
   TM values: Max =    -4  Min =    -5  Ave =    -4.46  Std =     0.50

 - Packet number: 12 starting at 2023-07-03T21:27:45.173.373.925 (XO: 9.173375501 - 9.173392501, t=37.2 degC)
   TM values: Max =    -4  Min =    -5  Ave =    -4.49  Std =     0.50

 - Packet number: 13 starting at 2023-07-03T21:27:46.007.317.925 (XO: 10.007318754 - 10.007335754, t=37.2 degC)
   TM values: Max =    -3  Min =    -5  Ave =    -4.38  Std =     0.49

 - Packet number: 14 starting at 2023-07-03T21:27:46.841.257.925 (XO: 10.841262006 - 10.841279006, t=37.2 degC)
   TM values: Max =    -3  Min =    -6  Ave =    -4.34  Std =     0.48

 - Packet number: 15 starting at 2023-07-03T21:27:47.675.201.925 (XO: 11.675205258 - 11.675222258, t=37.2 degC)
   TM values: Max =    -3  Min =    -5  Ave =    -4.40  Std =     0.49

 - Packet number: 16 starting at 2023-07-03T21:27:48.509.145.925 (XO: 12.509148460 - 12.509165460, t=37.2 degC)
   TM values: Max =    -3  Min =    -5  Ave =    -4.47  Std =     0.50

 - Packet number: 17 starting at 2023-07-03T21:27:49.343.089.925 (XO: 13.343091663 - 13.343108663, t=37.2 degC)
   TM values: Max =    -4  Min =    -5  Ave =    -4.42  Std =     0.49

 - Packet number: 18 starting at 2023-07-03T21:27:50.177.032.925 (XO: 14.177034915 - 14.177051916, t=37.2 degC)
   TM values: Max =    -4  Min =    -5  Ave =    -4.57  Std =     0.49
