-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jul  9 02:18:51 2023 UTC
-- File 20230630_013427_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.114237 to 35.404495
-each separately from 4s to 35s and from 114237us to 404495us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.199687 in 1 packets: 18
      20.089704 in 1 packets: 36
MIU packet time from 2023-06-30T01:34:34.114.237.000 to 2023-06-30T01:35:05.404.495.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.199687 in 1 packets: 18
      20.089704 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.8471074 to 36.8469946
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999600 in 1 packets: 18
      19.9999272 in 1 packets: 36
Temperature FPGA: 
      04 -> 49.8 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      05 -> 48.5 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20230630_013427_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.9613453 to 5.9616343
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.1996470  -- OK for SL --
   MIU time from 2023-06-30T01:34:34.114.237.000 to 2023-06-30T01:34:34.114.526.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.199687

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.1612813 to 17.1615703
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.0896310  -- OK for SL --
   MIU time from 2023-06-30T01:34:45.314.213.000 to 2023-06-30T01:34:45.314.502.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.089704

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.2512013 to 37.2514903
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-30T01:35:05.404.206.000 to 2023-06-30T01:35:05.404.495.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 5.961345300 corrected by 40bit 10MHz counter of 5.847107400 obtained at 1Hz counter of 4. Result: 4.114238311
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2023-06-30T01:34:34.114.096.725
   Max 1st sample MIU time among all SL packets: 2023-06-30T01:35:05.404.354.725
   Min 1st sample XO time among all SL packets: 4.114098036
   Max 1st sample XO time among all SL packets: 35.404356881
   TM values from -275 to 222 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-30T01:34:34.114.096.725 (XO: 4.114098036 - 4.114404037, t=37.2 degC)  Rank: 00082
   TM values: Max =   185  Min =  -242  Ave =    -4.55  Std =    20.34
   Min TM value:-242 at an offset of 127.638 usec -> 2023-06-30T01:34:34.114.224.362 (XO: 4.114225674)
   Max TM value:185 at an offset of 127.888 usec -> 2023-06-30T01:34:34.114.224.612 (XO: 4.114225924)
   Max dTM/dt [1/us]:1.99e+03 at an offset of 127.713 usec -> 2023-06-30T01:34:34.114.224.437 (XO: 4.114225749)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 125.000 usec -> 2023-06-30T01:34:34.114.221.725 (XO: 4.114223036)

 - Block number: 2 starting at 2023-06-30T01:34:45.314.072.725 (XO: 15.314074787 - 15.314380788, t=37.2 degC)  Rank: 00082
   TM values: Max =   191  Min =  -221  Ave =    -4.47  Std =    21.14
   Min TM value:-221 at an offset of 124.638 usec -> 2023-06-30T01:34:45.314.197.362 (XO: 15.314199425)
   Max TM value:191 at an offset of 108.825 usec -> 2023-06-30T01:34:45.314.181.550 (XO: 15.314183612)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 124.775 usec -> 2023-06-30T01:34:45.314.197.500 (XO: 15.314199563)
   Min dTM/dt [1/us]:-1.73e+03 at an offset of 108.988 usec -> 2023-06-30T01:34:45.314.181.712 (XO: 15.314183775)

 - Block number: 3 starting at 2023-06-30T01:35:05.404.065.725 (XO: 35.404067880 - 35.404373881, t=37.2 degC)  Rank: 00077
   TM values: Max =   222  Min =  -275  Ave =    -4.50  Std =    21.60
   Min TM value:-275 at an offset of 124.463 usec -> 2023-06-30T01:35:05.404.190.187 (XO: 35.404192343)
   Max TM value:222 at an offset of 124.200 usec -> 2023-06-30T01:35:05.404.189.925 (XO: 35.404192080)
   Max dTM/dt [1/us]:1.51e+03 at an offset of 127.000 usec -> 2023-06-30T01:35:05.404.192.725 (XO: 35.404194880)
   Min dTM/dt [1/us]:-2.30e+03 at an offset of 124.300 usec -> 2023-06-30T01:35:05.404.190.025 (XO: 35.404192180)
