-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jul  9 02:06:04 2023 UTC
-- File 20230630_011125_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.513606 to 75.984272
-each separately from 11s to 75s and from 293698us to 984272us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.209708 in 1 packets: 18
      12.569806 in 1 packets: 36
      21.339819 in 1 packets: 54
      20.019864 in 1 packets: 72
      2.329735 in 1 packets: 90
MIU packet time from 2023-06-30T01:11:39.513.606.000 to 2023-06-30T01:12:43.984.272.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.209708 in 1 packets: 18
      12.569806 in 1 packets: 36
      21.339819 in 1 packets: 54
      20.019864 in 1 packets: 72
      2.329735 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.8385697 to 76.8383370
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      7.9999709 in 1 packets: 18
      12.9999528 in 1 packets: 36
      20.9999237 in 1 packets: 54
      19.9999272 in 1 packets: 72
      1.9999927 in 1 packets: 90
Temperature FPGA: 
      05 -> 48.5 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      04 -> 49.8 degC in 90 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
      2 in 35 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
      2 in 35 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, 91, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20230630_011125_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.3521763 to 13.3524653
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.2096790  -- OK for SL --
   MIU time from 2023-06-30T01:11:39.513.606.000 to 2023-06-30T01:11:39.513.895.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.209708

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.5621443 to 21.5624333
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.5697590  -- OK for SL --
   MIU time from 2023-06-30T01:11:47.723.603.000 to 2023-06-30T01:11:47.723.892.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.569806

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.1321923 to 34.1324813
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.3397430  -- OK for SL --
   MIU time from 2023-06-30T01:12:00.293.698.000 to 2023-06-30T01:12:00.293.987.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.339819

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 55.4722243 to 55.4725133
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.0197910  -- OK for SL --
   MIU time from 2023-06-30T01:12:21.633.806.000 to 2023-06-30T01:12:21.634.095.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.019864

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 75.4923043 to 75.4925933
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 2.3297270  -- OK for SL --
   MIU time from 2023-06-30T01:12:41.653.959.000 to 2023-06-30T01:12:41.654.248.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 2.329735

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 77.8223203 to 77.8226093
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-30T01:12:43.983.983.000 to 2023-06-30T01:12:43.984.272.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.352176300 corrected by 40bit 10MHz counter of 12.838569700 obtained at 1Hz counter of 11. Result: 11.513608449
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2023-06-30T01:11:39.513.465.725
   Max 1st sample MIU time among all SL packets: 2023-06-30T01:12:43.984.131.725
   Min 1st sample XO time among all SL packets: 11.513468173
   Max 1st sample XO time among all SL packets: 75.984135568
   TM values from -273 to 222 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-30T01:11:39.513.465.725 (XO: 11.513468173 - 11.513774175, t=37.2 degC)  Rank: 00079
   TM values: Max =   215  Min =  -241  Ave =    -4.48  Std =    20.71
   Min TM value:-241 at an offset of 128.475 usec -> 2023-06-30T01:11:39.513.594.200 (XO: 11.513596649)
   Max TM value:215 at an offset of 128.238 usec -> 2023-06-30T01:11:39.513.593.962 (XO: 11.513596411)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 131.000 usec -> 2023-06-30T01:11:39.513.596.725 (XO: 11.513599174)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 116.200 usec -> 2023-06-30T01:11:39.513.581.925 (XO: 11.513584374)

 - Block number: 2 starting at 2023-06-30T01:11:47.723.462.725 (XO: 19.723466029 - 19.723772031, t=37.2 degC)  Rank: 00079
   TM values: Max =   202  Min =  -273  Ave =    -4.38  Std =    21.39
   Min TM value:-273 at an offset of 131.175 usec -> 2023-06-30T01:11:47.723.593.900 (XO: 19.723597205)
   Max TM value:202 at an offset of 130.888 usec -> 2023-06-30T01:11:47.723.593.612 (XO: 19.723596917)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 133.713 usec -> 2023-06-30T01:11:47.723.596.437 (XO: 19.723599742)
   Min dTM/dt [1/us]:-2.09e+03 at an offset of 130.988 usec -> 2023-06-30T01:11:47.723.593.712 (XO: 19.723597017)

 - Block number: 3 starting at 2023-06-30T01:12:00.293.557.725 (XO: 32.293559682 - 32.293865683, t=37.2 degC)  Rank: 00083
   TM values: Max =   222  Min =  -257  Ave =    -4.32  Std =    19.98
   Min TM value:-257 at an offset of 130.100 usec -> 2023-06-30T01:12:00.293.687.825 (XO: 32.293689782)
   Max TM value:222 at an offset of 129.800 usec -> 2023-06-30T01:12:00.293.687.525 (XO: 32.293689482)
   Max dTM/dt [1/us]:1.68e+03 at an offset of 127.600 usec -> 2023-06-30T01:12:00.293.685.325 (XO: 32.293687282)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 129.925 usec -> 2023-06-30T01:12:00.293.687.650 (XO: 32.293689607)

 - Block number: 4 starting at 2023-06-30T01:12:21.633.665.725 (XO: 53.633669270 - 53.633975271, t=37.2 degC)  Rank: 00075
   TM values: Max =   196  Min =  -223  Ave =    -4.40  Std =    19.38
   Min TM value:-223 at an offset of 117.413 usec -> 2023-06-30T01:12:21.633.783.137 (XO: 53.633786683)
   Max TM value:196 at an offset of 114.650 usec -> 2023-06-30T01:12:21.633.780.375 (XO: 53.633783920)
   Max dTM/dt [1/us]:2.03e+03 at an offset of 117.513 usec -> 2023-06-30T01:12:21.633.783.237 (XO: 53.633786783)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 114.813 usec -> 2023-06-30T01:12:21.633.780.537 (XO: 53.633784083)

 - Block number: 5 starting at 2023-06-30T01:12:41.653.818.725 (XO: 73.653822079 - 73.654128080, t=37.2 degC)  Rank: 00089
   TM values: Max =   180  Min =  -226  Ave =    -4.38  Std =    20.59
   Min TM value:-226 at an offset of 129.913 usec -> 2023-06-30T01:12:41.653.948.637 (XO: 73.653951992)
   Max TM value:180 at an offset of 132.713 usec -> 2023-06-30T01:12:41.653.951.437 (XO: 73.653954792)
   Max dTM/dt [1/us]:1.68e+03 at an offset of 130.038 usec -> 2023-06-30T01:12:41.653.948.762 (XO: 73.653952117)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 114.375 usec -> 2023-06-30T01:12:41.653.933.100 (XO: 73.653936454)

 - Block number: 6 starting at 2023-06-30T01:12:43.983.842.725 (XO: 75.983846567 - 75.984152568, t=37.2 degC)  Rank: 00078
   TM values: Max =   188  Min =  -258  Ave =    -4.38  Std =    19.38
   Min TM value:-258 at an offset of 128.663 usec -> 2023-06-30T01:12:43.983.971.387 (XO: 75.983975230)
   Max TM value:188 at an offset of 131.363 usec -> 2023-06-30T01:12:43.983.974.087 (XO: 75.983977930)
   Max dTM/dt [1/us]:1.89e+03 at an offset of 131.200 usec -> 2023-06-30T01:12:43.983.973.925 (XO: 75.983977767)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 128.500 usec -> 2023-06-30T01:12:43.983.971.225 (XO: 75.983975067)
