-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jul  2 08:24:48 2023 UTC
-- File 20230624_195633_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.193684 to 48.683772
-each separately from 3s to 48s and from 193684us to 683772us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.119691 in 1 packets: 18
      23.149564 in 1 packets: 36
      9.219677 in 1 packets: 54
MIU packet time from 2023-06-24T19:56:39.193.684.000 to 2023-06-24T19:57:24.683.772.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.119691 in 1 packets: 18
      23.149564 in 1 packets: 36
      9.219677 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.7040617 to 49.7039103
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999563 in 1 packets: 18
      22.9999226 in 1 packets: 36
      8.9999697 in 1 packets: 54
Temperature FPGA: 
      FF -> 56.3 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      FE -> 57.6 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C6 -> 45.2 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      2 in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20230624_195633_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.8977473 to 4.8980363
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.1196470  -- OK for SL --
   MIU time from 2023-06-24T19:56:39.193.684.000 to 2023-06-24T19:56:39.193.973.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.119691

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.0176833 to 18.0179723
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 23.1494870  -- OK for SL --
   MIU time from 2023-06-24T19:56:52.313.664.000 to 2023-06-24T19:56:52.313.953.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 23.149564

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 41.1674593 to 41.1677483
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.2196470  -- OK for SL --
   MIU time from 2023-06-24T19:57:15.463.517.000 to 2023-06-24T19:57:15.463.806.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.219677

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 50.3873953 to 50.3876843
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-24T19:57:24.683.483.000 to 2023-06-24T19:57:24.683.772.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 4.897747300 corrected by 40bit 10MHz counter of 4.704061700 obtained at 1Hz counter of 3. Result: 3.193686239
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2023-06-24T19:56:39.193.543.725
   Max 1st sample MIU time among all SL packets: 2023-06-24T19:57:24.683.631.725
   Min 1st sample XO time among all SL packets: 3.193545964
   Max 1st sample XO time among all SL packets: 48.683635981
   TM values from -309 to 247 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-24T19:56:39.193.543.725 (XO: 3.193545964 - 3.193851965, t=45.2 degC)  Rank: 00078
   TM values: Max =   233  Min =  -309  Ave =    -4.45  Std =    30.83
   Min TM value:-309 at an offset of 129.688 usec -> 2023-06-24T19:56:39.193.673.412 (XO: 3.193675652)
   Max TM value:233 at an offset of 129.963 usec -> 2023-06-24T19:56:39.193.673.687 (XO: 3.193675927)
   Max dTM/dt [1/us]:2.74e+03 at an offset of 129.813 usec -> 2023-06-24T19:56:39.193.673.537 (XO: 3.193675777)
   Min dTM/dt [1/us]:-2.50e+03 at an offset of 127.338 usec -> 2023-06-24T19:56:39.193.671.062 (XO: 3.193673302)

 - Block number: 2 starting at 2023-06-24T19:56:52.313.523.725 (XO: 16.313526060 - 16.313832061, t=45.2 degC)  Rank: 00084
   TM values: Max =   247  Min =  -292  Ave =    -4.47  Std =    27.50
   Min TM value:-292 at an offset of 99.600 usec -> 2023-06-24T19:56:52.313.623.325 (XO: 16.313625660)
   Max TM value:247 at an offset of 104.450 usec -> 2023-06-24T19:56:52.313.628.175 (XO: 16.313630510)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 102.050 usec -> 2023-06-24T19:56:52.313.625.775 (XO: 16.313628110)
   Min dTM/dt [1/us]:-2.52e+03 at an offset of 99.438 usec -> 2023-06-24T19:56:52.313.623.162 (XO: 16.313625497)

 - Block number: 3 starting at 2023-06-24T19:57:15.463.376.725 (XO: 39.463380000 - 39.463686002, t=45.2 degC)  Rank: 00078
   TM values: Max =   217  Min =  -246  Ave =    -4.40  Std =    29.04
   Min TM value:-246 at an offset of 111.413 usec -> 2023-06-24T19:57:15.463.488.137 (XO: 39.463491413)
   Max TM value:217 at an offset of 116.338 usec -> 2023-06-24T19:57:15.463.493.062 (XO: 39.463496338)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 113.913 usec -> 2023-06-24T19:57:15.463.490.637 (XO: 39.463493913)
   Min dTM/dt [1/us]:-2.18e+03 at an offset of 98.475 usec -> 2023-06-24T19:57:15.463.475.200 (XO: 39.463478476)

 - Block number: 4 starting at 2023-06-24T19:57:24.683.342.725 (XO: 48.683346980 - 48.683652981, t=45.2 degC)  Rank: 00078
   TM values: Max =   225  Min =  -247  Ave =    -4.44  Std =    28.07
   Min TM value:-247 at an offset of 114.000 usec -> 2023-06-24T19:57:24.683.456.725 (XO: 48.683460980)
   Max TM value:225 at an offset of 116.600 usec -> 2023-06-24T19:57:24.683.459.325 (XO: 48.683463580)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 114.113 usec -> 2023-06-24T19:57:24.683.456.837 (XO: 48.683461093)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 116.725 usec -> 2023-06-24T19:57:24.683.459.450 (XO: 48.683463705)
