-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jul  2 08:21:38 2023 UTC
-- File 20230624_195302_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.414201 to 18.534204
-each separately from 3s to 18s and from 414201us to 534204us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.119425 in 1 packets: 18
MIU packet time from 2023-06-24T19:53:08.414.201.000 to 2023-06-24T19:53:23.534.204.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.119425 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.7096440 to 19.7095936
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999496 in 1 packets: 18
Temperature FPGA: 
      FF -> 56.3 degC in 36 packets (all packets)
Temperature XO: 
      C6 -> 45.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 35 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230624_195302_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.1238465 to 5.1241355
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.1193750  -- OK for SL --
   MIU time from 2023-06-24T19:53:08.414.201.000 to 2023-06-24T19:53:08.414.490.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.119425

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.2435105 to 20.2437995
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-24T19:53:23.533.915.000 to 2023-06-24T19:53:23.534.204.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.123846500 corrected by 40bit 10MHz counter of 4.709644000 obtained at 1Hz counter of 3. Result: 3.414203867
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996700 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2023-06-24T19:53:08.414.060.725
   Max 1st sample MIU time among all SL packets: 2023-06-24T19:53:23.534.063.725
   Min 1st sample XO time among all SL packets: 3.414063591
   Max 1st sample XO time among all SL packets: 18.534067387
   TM values from -307 to 234 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-24T19:53:08.414.060.725 (XO: 3.414063591 - 3.414369592, t=45.2 degC)  Rank: 00082
   TM values: Max =   234  Min =  -307  Ave =    -4.40  Std =    25.42
   Min TM value:-307 at an offset of 134.638 usec -> 2023-06-24T19:53:08.414.195.362 (XO: 3.414198229)
   Max TM value:234 at an offset of 134.913 usec -> 2023-06-24T19:53:08.414.195.637 (XO: 3.414198504)
   Max dTM/dt [1/us]:2.42e+03 at an offset of 134.750 usec -> 2023-06-24T19:53:08.414.195.475 (XO: 3.414198342)
   Min dTM/dt [1/us]:-2.62e+03 at an offset of 132.338 usec -> 2023-06-24T19:53:08.414.193.062 (XO: 3.414195929)

 - Block number: 2 starting at 2023-06-24T19:53:23.533.774.725 (XO: 18.533778386 - 18.534084387, t=45.2 degC)  Rank: 00080
   TM values: Max =   208  Min =  -237  Ave =    -4.46  Std =    32.15
   Min TM value:-237 at an offset of 169.688 usec -> 2023-06-24T19:53:23.533.944.413 (XO: 18.533948075)
   Max TM value:208 at an offset of 131.400 usec -> 2023-06-24T19:53:23.533.906.125 (XO: 18.533909787)
   Max dTM/dt [1/us]:1.86e+03 at an offset of 167.288 usec -> 2023-06-24T19:53:23.533.942.013 (XO: 18.533945675)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 169.576 usec -> 2023-06-24T19:53:23.533.944.300 (XO: 18.533947962)
