-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jul  2 08:10:43 2023 UTC
-- File 20230624_193917_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.197591 to 62.017986
-each separately from 2s to 62s and from 17697us to 997847us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.619640 in 1 packets: 18
      16.179749 in 1 packets: 36
      6.929734 in 1 packets: 54
      14.089827 in 1 packets: 72
MIU packet time from 2023-06-24T19:39:22.197.591.000 to 2023-06-24T19:40:22.017.986.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      22.619640 in 1 packets: 18
      16.179749 in 1 packets: 36
      6.929734 in 1 packets: 54
      14.089827 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.5709995 to 63.5707982
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999261 in 1 packets: 18
      15.9999464 in 1 packets: 36
      6.9999765 in 1 packets: 54
      14.9999497 in 1 packets: 72
Temperature FPGA: 
      FF -> 56.3 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      FE -> 57.6 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C6 -> 45.2 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20230624_193917_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.7685921 to 3.7688811
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.6195670  -- OK for SL --
   MIU time from 2023-06-24T19:39:22.197.591.000 to 2023-06-24T19:39:22.197.880.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.619640

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.3884481 to 26.3887371
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.1796950  -- OK for SL --
   MIU time from 2023-06-24T19:39:44.817.520.000 to 2023-06-24T19:39:44.817.809.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.179749

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.5684321 to 42.5687211
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.9297110  -- OK for SL --
   MIU time from 2023-06-24T19:40:00.997.558.000 to 2023-06-24T19:40:00.997.847.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.929734

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 49.4984321 to 49.4987211
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.0897750  -- OK for SL --
   MIU time from 2023-06-24T19:40:07.927.581.000 to 2023-06-24T19:40:07.927.870.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.089827

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 63.5884961 to 63.5887851
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-24T19:40:22.017.697.000 to 2023-06-24T19:40:22.017.986.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 3.768592100 corrected by 40bit 10MHz counter of 3.570999500 obtained at 1Hz counter of 2. Result: 2.197593272
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996700
   Min 1st sample MIU time among all SL packets: 2023-06-24T19:39:22.197.450.725
   Max 1st sample MIU time among all SL packets: 2023-06-24T19:40:22.017.845.725
   Min 1st sample XO time among all SL packets: 2.197452996
   Max 1st sample XO time among all SL packets: 62.017846684
   TM values from -333 to 256 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-24T19:39:22.197.450.725 (XO: 2.197452996 - 2.197758997, t=45.2 degC)  Rank: 00078
   TM values: Max =   256  Min =  -322  Ave =    -4.44  Std =    27.40
   Min TM value:-322 at an offset of 117.000 usec -> 2023-06-24T19:39:22.197.567.725 (XO: 2.197569997)
   Max TM value:256 at an offset of 116.750 usec -> 2023-06-24T19:39:22.197.567.475 (XO: 2.197569747)
   Max dTM/dt [1/us]:2.26e+03 at an offset of 119.250 usec -> 2023-06-24T19:39:22.197.569.975 (XO: 2.197572247)
   Min dTM/dt [1/us]:-2.89e+03 at an offset of 116.813 usec -> 2023-06-24T19:39:22.197.567.537 (XO: 2.197569809)

 - Block number: 2 starting at 2023-06-24T19:39:44.817.379.725 (XO: 24.817384922 - 24.817690923, t=45.2 degC)  Rank: 00082
   TM values: Max =   234  Min =  -323  Ave =    -4.38  Std =    30.46
   Min TM value:-323 at an offset of 102.800 usec -> 2023-06-24T19:39:44.817.482.525 (XO: 24.817487723)
   Max TM value:234 at an offset of 102.513 usec -> 2023-06-24T19:39:44.817.482.237 (XO: 24.817487435)
   Max dTM/dt [1/us]:2.48e+03 at an offset of 105.150 usec -> 2023-06-24T19:39:44.817.484.875 (XO: 24.817490073)
   Min dTM/dt [1/us]:-2.48e+03 at an offset of 102.675 usec -> 2023-06-24T19:39:44.817.482.400 (XO: 24.817487598)

 - Block number: 3 starting at 2023-06-24T19:40:00.997.417.725 (XO: 40.997423216 - 40.997729217, t=45.2 degC)  Rank: 00081
   TM values: Max =   205  Min =  -316  Ave =    -4.52  Std =    27.89
   Min TM value:-316 at an offset of 126.675 usec -> 2023-06-24T19:40:00.997.544.400 (XO: 40.997549892)
   Max TM value:205 at an offset of 129.113 usec -> 2023-06-24T19:40:00.997.546.837 (XO: 40.997552329)
   Max dTM/dt [1/us]:2.49e+03 at an offset of 128.988 usec -> 2023-06-24T19:40:00.997.546.712 (XO: 40.997552204)
   Min dTM/dt [1/us]:-2.50e+03 at an offset of 126.538 usec -> 2023-06-24T19:40:00.997.544.262 (XO: 40.997549754)

 - Block number: 4 starting at 2023-06-24T19:40:07.927.440.725 (XO: 47.927446386 - 47.927752387, t=45.2 degC)  Rank: 00075
   TM values: Max =   233  Min =  -254  Ave =    -4.43  Std =    30.79
   Min TM value:-254 at an offset of 147.900 usec -> 2023-06-24T19:40:07.927.588.625 (XO: 47.927594286)
   Max TM value:233 at an offset of 120.488 usec -> 2023-06-24T19:40:07.927.561.212 (XO: 47.927566873)
   Max dTM/dt [1/us]:2.45e+03 at an offset of 117.975 usec -> 2023-06-24T19:40:07.927.558.700 (XO: 47.927564361)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 120.588 usec -> 2023-06-24T19:40:07.927.561.312 (XO: 47.927566973)

 - Block number: 5 starting at 2023-06-24T19:40:22.017.556.725 (XO: 62.017557683 - 62.017863684, t=45.2 degC)  Rank: 00077
   TM values: Max =   254  Min =  -333  Ave =    -4.39  Std =    31.47
   Min TM value:-333 at an offset of 95.663 usec -> 2023-06-24T19:40:22.017.652.387 (XO: 62.017653346)
   Max TM value:254 at an offset of 93.225 usec -> 2023-06-24T19:40:22.017.649.950 (XO: 62.017650908)
   Max dTM/dt [1/us]:2.63e+03 at an offset of 95.788 usec -> 2023-06-24T19:40:22.017.652.512 (XO: 62.017653471)
   Min dTM/dt [1/us]:-2.75e+03 at an offset of 93.388 usec -> 2023-06-24T19:40:22.017.650.112 (XO: 62.017651071)
