-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jul  2 03:41:55 2023 UTC
-- File 20230624_093634_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.495312 to 20.175574
-each separately from 0s to 20s and from 175285us to 495601us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.679684 in 1 packets: 18
MIU packet time from 2023-06-24T09:36:37.495.312.000 to 2023-06-24T09:36:57.175.574.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.679684 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.7257041 to 21.7256356
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      19.9999315 in 1 packets: 18
Temperature FPGA: 
      00 -> 55.0 degC in 36 packets (all packets)
Temperature XO: 
      C5 -> 43.6 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230624_093634_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.2210174 to 2.2213064
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.6796150  -- OK for SL --
   MIU time from 2023-06-24T09:36:37.495.312.000 to 2023-06-24T09:36:37.495.601.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.679684

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.9009214 to 21.9012104
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-24T09:36:57.175.285.000 to 2023-06-24T09:36:57.175.574.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.221017400 corrected by 40bit 10MHz counter of 1.725704100 obtained at 1Hz counter of 0. Result: 0.495314984
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-06-24T09:36:37.495.171.725
   Max 1st sample MIU time among all SL packets: 2023-06-24T09:36:57.175.433.725
   Min 1st sample XO time among all SL packets: 0.495174709
   Max 1st sample XO time among all SL packets: 20.175435121
   TM values from -236 to 202 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-24T09:36:37.495.171.725 (XO: 0.495174709 - 0.495480710, t=43.6 degC)  Rank: 00077
   TM values: Max =   196  Min =  -215  Ave =    -4.52  Std =    26.78
   Min TM value:-215 at an offset of 137.550 usec -> 2023-06-24T09:36:37.495.309.275 (XO: 0.495312259)
   Max TM value:196 at an offset of 125.050 usec -> 2023-06-24T09:36:37.495.296.775 (XO: 0.495299759)
   Max dTM/dt [1/us]:1.77e+03 at an offset of 152.426 usec -> 2023-06-24T09:36:37.495.324.150 (XO: 0.495327134)
   Min dTM/dt [1/us]:-1.97e+03 at an offset of 137.425 usec -> 2023-06-24T09:36:37.495.309.150 (XO: 0.495312134)

 - Block number: 2 starting at 2023-06-24T09:36:57.175.144.725 (XO: 20.175146120 - 20.175452122, t=43.6 degC)  Rank: 00086
   TM values: Max =   202  Min =  -236  Ave =    -4.41  Std =    22.47
   Min TM value:-236 at an offset of 121.225 usec -> 2023-06-24T09:36:57.175.265.950 (XO: 20.175267346)
   Max TM value:202 at an offset of 126.175 usec -> 2023-06-24T09:36:57.175.270.900 (XO: 20.175272296)
   Max dTM/dt [1/us]:1.61e+03 at an offset of 123.763 usec -> 2023-06-24T09:36:57.175.268.487 (XO: 20.175269883)
   Min dTM/dt [1/us]:-1.65e+03 at an offset of 108.788 usec -> 2023-06-24T09:36:57.175.253.512 (XO: 20.175254908)
