-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jul  1 12:16:02 2023 UTC
-- File 20230611_230042_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.991015 to 21.048793
-each separately from 7s to 21s and from 48504us to 991304us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.057200 in 1 packets: 18
MIU packet time from 2023-06-11T23:00:52.991.015.000 to 2023-06-11T23:01:06.048.793.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.057200 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.7056650 to 22.7056164
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999514 in 1 packets: 18
Temperature FPGA: 
      00 -> 55.0 degC in 36 packets (all packets)
Temperature XO: 
      C4 -> 42.0 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230611_230042_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.6966817 to 9.6969707
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.0571510  -- OK for SL --
   MIU time from 2023-06-11T23:00:52.991.015.000 to 2023-06-11T23:00:52.991.304.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.057200

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      87 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.7541217 to 22.7544107
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-11T23:01:06.048.504.000 to 2023-06-11T23:01:06.048.793.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.696681700 corrected by 40bit 10MHz counter of 8.705665000 obtained at 1Hz counter of 7. Result: 7.991020069
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996600 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-06-11T23:00:52.990.874.725
   Max 1st sample MIU time among all SL packets: 2023-06-11T23:01:06.048.652.725
   Min 1st sample XO time among all SL packets: 7.990879794
   Max 1st sample XO time among all SL packets: 21.048654190
   TM values from -307 to 248 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-11T23:00:52.990.874.725 (XO: 7.990879794 - 7.991185795, t=42.0 degC)  Rank: 00073
   TM values: Max =   248  Min =  -307  Ave =    -4.49  Std =    28.37
   Min TM value:-307 at an offset of 126.688 usec -> 2023-06-11T23:00:52.991.001.412 (XO: 7.991006482)
   Max TM value:248 at an offset of 126.413 usec -> 2023-06-11T23:00:52.991.001.137 (XO: 7.991006207)
   Max dTM/dt [1/us]:2.29e+03 at an offset of 124.425 usec -> 2023-06-11T23:00:52.990.999.150 (XO: 7.991004219)
   Min dTM/dt [1/us]:-2.64e+03 at an offset of 126.525 usec -> 2023-06-11T23:00:52.991.001.250 (XO: 7.991006319)

 - Block number: 2 starting at 2023-06-11T23:01:06.048.363.725 (XO: 21.048365189 - 21.048671190, t=42.0 degC)  Rank: 00087
   TM values: Max =   205  Min =  -278  Ave =    -4.47  Std =    23.43
   Min TM value:-278 at an offset of 110.388 usec -> 2023-06-11T23:01:06.048.474.112 (XO: 21.048475577)
   Max TM value:205 at an offset of 110.113 usec -> 2023-06-11T23:01:06.048.473.837 (XO: 21.048475302)
   Max dTM/dt [1/us]:2.14e+03 at an offset of 112.713 usec -> 2023-06-11T23:01:06.048.476.437 (XO: 21.048477902)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 110.250 usec -> 2023-06-11T23:01:06.048.473.975 (XO: 21.048475440)
