-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jul  1 11:21:57 2023 UTC
-- File 20230611_211822_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.344837 to 15.844227
-each separately from 11s to 15s and from 344837us to 844227us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.498812 in 1 packets: 18
MIU packet time from 2023-06-11T21:18:36.344.837.000 to 2023-06-11T21:18:40.844.227.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.498812 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 12.5590085 to 16.5589945
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      3.9999860 in 1 packets: 18
Temperature FPGA: 
      00 -> 55.0 degC in 36 packets (all packets)
Temperature XO: 
      C5 -> 43.6 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      C4 -> 42.0 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230611_211822_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.9038467 to 12.9041357
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.4987990  -- OK for SL --
   MIU time from 2023-06-11T21:18:36.344.837.000 to 2023-06-11T21:18:36.345.126.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.498812

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.4029347 to 17.4032237
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-11T21:18:40.843.938.000 to 2023-06-11T21:18:40.844.227.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 12.903846700 corrected by 40bit 10MHz counter of 12.559008500 obtained at 1Hz counter of 11. Result: 11.344839407
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-06-11T21:18:36.344.696.725
   Max 1st sample MIU time among all SL packets: 2023-06-11T21:18:40.844.086.725
   Min 1st sample XO time among all SL packets: 11.344699131
   Max 1st sample XO time among all SL packets: 15.844091795
   TM values from -321 to 260 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-11T21:18:36.344.696.725 (XO: 11.344699131 - 11.345005133, t=43.6 degC)  Rank: 00078
   TM values: Max =   250  Min =  -307  Ave =    -4.37  Std =    22.32
   Min TM value:-307 at an offset of 103.275 usec -> 2023-06-11T21:18:36.344.800.000 (XO: 11.344802407)
   Max TM value:250 at an offset of 102.988 usec -> 2023-06-11T21:18:36.344.799.712 (XO: 11.344802119)
   Max dTM/dt [1/us]:2.17e+03 at an offset of 105.600 usec -> 2023-06-11T21:18:36.344.802.325 (XO: 11.344804732)
   Min dTM/dt [1/us]:-2.69e+03 at an offset of 103.113 usec -> 2023-06-11T21:18:36.344.799.837 (XO: 11.344802244)

 - Block number: 2 starting at 2023-06-11T21:18:40.843.797.725 (XO: 15.843802794 - 15.844108795, t=42.0 degC)  Rank: 00076
   TM values: Max =   260  Min =  -321  Ave =    -4.53  Std =    32.05
   Min TM value:-321 at an offset of 127.275 usec -> 2023-06-11T21:18:40.843.925.000 (XO: 15.843930069)
   Max TM value:260 at an offset of 126.988 usec -> 2023-06-11T21:18:40.843.924.712 (XO: 15.843929782)
   Max dTM/dt [1/us]:2.19e+03 at an offset of 129.550 usec -> 2023-06-11T21:18:40.843.927.275 (XO: 15.843932344)
   Min dTM/dt [1/us]:-2.44e+03 at an offset of 127.138 usec -> 2023-06-11T21:18:40.843.924.862 (XO: 15.843929932)
