-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jul  1 09:59:59 2023 UTC
-- File 20230611_155652_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.431729 to 59.475948
-each separately from 10s to 59s and from 431729us to 724204us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.051181 in 1 packets: 18
      8.240427 in 1 packets: 36
      22.751455 in 1 packets: 54
MIU packet time from 2023-06-11T15:57:05.431.729.000 to 2023-06-11T15:57:54.475.948.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.051181 in 1 packets: 18
      8.240427 in 1 packets: 36
      22.751455 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.6906520 to 60.6904822
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999378 in 1 packets: 18
      7.9999722 in 1 packets: 36
      22.9999202 in 1 packets: 54
Temperature FPGA: 
      01 -> 53.7 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      00 -> 55.0 degC in 18 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Temperature XO: 
      C4 -> 42.0 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20230611_155652_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.1223824 to 12.1226714
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.0511190  -- OK for SL --
   MIU time from 2023-06-11T15:57:05.431.729.000 to 2023-06-11T15:57:05.432.018.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.051181

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.1737904 to 30.1740794
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.2403990  -- OK for SL --
   MIU time from 2023-06-11T15:57:23.483.199.000 to 2023-06-11T15:57:23.483.488.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.240427

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 38.4144784 to 38.4147674
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.7513750  -- OK for SL --
   MIU time from 2023-06-11T15:57:31.723.915.000 to 2023-06-11T15:57:31.724.204.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.751455

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 61.1661424 to 61.1664314
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-11T15:57:54.475.659.000 to 2023-06-11T15:57:54.475.948.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.122382400 corrected by 40bit 10MHz counter of 11.690652000 obtained at 1Hz counter of 10. Result: 10.431731911
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-06-11T15:57:05.431.588.725
   Max 1st sample MIU time among all SL packets: 2023-06-11T15:57:54.475.807.725
   Min 1st sample XO time among all SL packets: 10.431591636
   Max 1st sample XO time among all SL packets: 59.475810590
   TM values from -266 to 199 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-11T15:57:05.431.588.725 (XO: 10.431591636 - 10.431897637, t=42.0 degC)  Rank: 00084
   TM values: Max =   167  Min =  -228  Ave =    -4.52  Std =    19.85
   Min TM value:-228 at an offset of 112.988 usec -> 2023-06-11T15:57:05.431.701.712 (XO: 10.431704623)
   Max TM value:167 at an offset of 118.100 usec -> 2023-06-11T15:57:05.431.706.825 (XO: 10.431709736)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 115.538 usec -> 2023-06-11T15:57:05.431.704.262 (XO: 10.431707173)
   Min dTM/dt [1/us]:-1.50e+03 at an offset of 112.863 usec -> 2023-06-11T15:57:05.431.701.587 (XO: 10.431704498)

 - Block number: 2 starting at 2023-06-11T15:57:23.483.058.725 (XO: 28.483062016 - 28.483368017, t=42.0 degC)  Rank: 00078
   TM values: Max =   190  Min =  -262  Ave =    -4.49  Std =    19.33
   Min TM value:-262 at an offset of 126.500 usec -> 2023-06-11T15:57:23.483.185.225 (XO: 28.483188516)
   Max TM value:190 at an offset of 126.238 usec -> 2023-06-11T15:57:23.483.184.962 (XO: 28.483188254)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 128.863 usec -> 2023-06-11T15:57:23.483.187.587 (XO: 28.483190879)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 126.350 usec -> 2023-06-11T15:57:23.483.185.075 (XO: 28.483188366)

 - Block number: 3 starting at 2023-06-11T15:57:31.723.774.725 (XO: 36.723778658 - 36.724084659, t=42.0 degC)  Rank: 00085
   TM values: Max =   199  Min =  -266  Ave =    -4.31  Std =    19.49
   Min TM value:-266 at an offset of 113.238 usec -> 2023-06-11T15:57:31.723.887.962 (XO: 36.723891896)
   Max TM value:199 at an offset of 118.213 usec -> 2023-06-11T15:57:31.723.892.937 (XO: 36.723896871)
   Max dTM/dt [1/us]:1.62e+03 at an offset of 115.700 usec -> 2023-06-11T15:57:31.723.890.425 (XO: 36.723894359)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 113.113 usec -> 2023-06-11T15:57:31.723.887.837 (XO: 36.723891771)

 - Block number: 4 starting at 2023-06-11T15:57:54.475.518.725 (XO: 59.475521589 - 59.475827590, t=42.0 degC)  Rank: 00075
   TM values: Max =   194  Min =  -245  Ave =    -4.42  Std =    17.64
   Min TM value:-245 at an offset of 115.075 usec -> 2023-06-11T15:57:54.475.633.800 (XO: 59.475636665)
   Max TM value:194 at an offset of 115.325 usec -> 2023-06-11T15:57:54.475.634.050 (XO: 59.475636915)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 115.188 usec -> 2023-06-11T15:57:54.475.633.912 (XO: 59.475636777)
   Min dTM/dt [1/us]:-1.71e+03 at an offset of 112.688 usec -> 2023-06-11T15:57:54.475.631.412 (XO: 59.475634277)
