-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jul  1 09:43:13 2023 UTC
-- File 20230611_153642_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.175605 to 22.594773
-each separately from 13s to 22s and from 175605us to 594773us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.418590 in 1 packets: 18
MIU packet time from 2023-06-11T15:36:58.175.605.000 to 2023-06-11T15:37:07.594.773.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.418590 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.7011699 to 23.7011384
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999685 in 1 packets: 18
Temperature FPGA: 
      00 -> 55.0 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      01 -> 53.7 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C4 -> 42.0 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230611_153642_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.8767757 to 14.8770647
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.4185590  -- OK for SL --
   MIU time from 2023-06-11T15:36:58.175.605.000 to 2023-06-11T15:36:58.175.894.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.418590

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      91 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.2956237 to 24.2959127
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-11T15:37:07.594.484.000 to 2023-06-11T15:37:07.594.773.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 14.876775700 corrected by 40bit 10MHz counter of 14.701169900 obtained at 1Hz counter of 13. Result: 13.175606415
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-06-11T15:36:58.175.464.725
   Max 1st sample MIU time among all SL packets: 2023-06-11T15:37:07.594.632.725
   Min 1st sample XO time among all SL packets: 13.175466139
   Max 1st sample XO time among all SL packets: 22.594636106
   TM values from -302 to 216 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-11T15:36:58.175.464.725 (XO: 13.175466139 - 13.175772140, t=42.0 degC)  Rank: 00078
   TM values: Max =   216  Min =  -271  Ave =    -4.32  Std =    18.55
   Min TM value:-271 at an offset of 120.413 usec -> 2023-06-11T15:36:58.175.585.137 (XO: 13.175586552)
   Max TM value:216 at an offset of 120.688 usec -> 2023-06-11T15:36:58.175.585.412 (XO: 13.175586827)
   Max dTM/dt [1/us]:2.19e+03 at an offset of 120.525 usec -> 2023-06-11T15:36:58.175.585.250 (XO: 13.175586665)
   Min dTM/dt [1/us]:-2.27e+03 at an offset of 118.050 usec -> 2023-06-11T15:36:58.175.582.775 (XO: 13.175584190)

 - Block number: 2 starting at 2023-06-11T15:37:07.594.343.725 (XO: 22.594347105 - 22.594653106, t=42.0 degC)  Rank: 00091
   TM values: Max =   208  Min =  -302  Ave =    -4.40  Std =    22.01
   Min TM value:-302 at an offset of 127.563 usec -> 2023-06-11T15:37:07.594.471.287 (XO: 22.594474668)
   Max TM value:208 at an offset of 114.988 usec -> 2023-06-11T15:37:07.594.458.712 (XO: 22.594462093)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 129.938 usec -> 2023-06-11T15:37:07.594.473.662 (XO: 22.594477043)
   Min dTM/dt [1/us]:-2.48e+03 at an offset of 127.450 usec -> 2023-06-11T15:37:07.594.471.175 (XO: 22.594474556)
