-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jul  1 09:37:53 2023 UTC
-- File 20230611_151558_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 4.628028 to 19.768193
-each separately from 4s to 19s and from 628028us to 768193us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.139587 in 1 packets: 18
MIU packet time from 2023-06-11T15:16:05.628.028.000 to 2023-06-11T15:16:20.768.193.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.139587 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.6523059 to 20.6522539
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999480 in 1 packets: 18
Temperature FPGA: 
      02 -> 52.4 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      01 -> 53.7 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C4 -> 42.0 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230611_151558_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.2803352 to 6.2806242
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.1395350  -- OK for SL --
   MIU time from 2023-06-11T15:16:05.628.028.000 to 2023-06-11T15:16:05.628.317.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.139587

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.4201592 to 21.4204482
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-11T15:16:20.767.904.000 to 2023-06-11T15:16:20.768.193.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 4, 32bit 10MHz counter = 6.280335200 corrected by 40bit 10MHz counter of 5.652305900 obtained at 1Hz counter of 4. Result: 4.628031498
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-06-11T15:16:05.627.887.725
   Max 1st sample MIU time among all SL packets: 2023-06-11T15:16:20.768.052.725
   Min 1st sample XO time among all SL packets: 4.627891223
   Max 1st sample XO time among all SL packets: 19.768056713
   TM values from -291 to 211 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-11T15:16:05.627.887.725 (XO: 4.627891223 - 4.628197224, t=42.0 degC)  Rank: 00081
   TM values: Max =   211  Min =  -291  Ave =    -4.37  Std =    20.36
   Min TM value:-291 at an offset of 104.488 usec -> 2023-06-11T15:16:05.627.992.212 (XO: 4.627995710)
   Max TM value:211 at an offset of 92.088 usec -> 2023-06-11T15:16:05.627.979.812 (XO: 4.627983310)
   Max dTM/dt [1/us]:1.54e+03 at an offset of 107.000 usec -> 2023-06-11T15:16:05.627.994.725 (XO: 4.627998223)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 92.200 usec -> 2023-06-11T15:16:05.627.979.925 (XO: 4.627983423)

 - Block number: 2 starting at 2023-06-11T15:16:20.767.763.725 (XO: 19.767767712 - 19.768073713, t=42.0 degC)  Rank: 00082
   TM values: Max =   192  Min =  -256  Ave =    -4.29  Std =    25.19
   Min TM value:-256 at an offset of 132.888 usec -> 2023-06-11T15:16:20.767.896.612 (XO: 19.767900600)
   Max TM value:192 at an offset of 130.350 usec -> 2023-06-11T15:16:20.767.894.075 (XO: 19.767898063)
   Max dTM/dt [1/us]:2.10e+03 at an offset of 133.000 usec -> 2023-06-11T15:16:20.767.896.725 (XO: 19.767900713)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 130.500 usec -> 2023-06-11T15:16:20.767.894.225 (XO: 19.767898213)
