-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jul  1 09:22:29 2023 UTC
-- File 20230611_145948_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 11.535442 to 42.457151
-each separately from 11s to 42s and from 456862us to 546500us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.010480 in 1 packets: 18
      16.910362 in 1 packets: 36
MIU packet time from 2023-06-11T15:00:02.535.442.000 to 2023-06-11T15:00:33.457.151.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.010480 in 1 packets: 18
      16.910362 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.0797577 to 44.0796501
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999515 in 1 packets: 18
      16.9999409 in 1 packets: 36
Temperature FPGA: 
      01 -> 53.7 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      00 -> 55.0 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C4 -> 42.0 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20230611_145948_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 13.6152014 to 13.6154904
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.0104310  -- OK for SL --
   MIU time from 2023-06-11T15:00:02.535.442.000 to 2023-06-11T15:00:02.535.731.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.010480

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.6259214 to 27.6262104
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.9103030  -- OK for SL --
   MIU time from 2023-06-11T15:00:16.546.211.000 to 2023-06-11T15:00:16.546.500.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.910362

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 44.5365134 to 44.5368024
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-11T15:00:33.456.862.000 to 2023-06-11T15:00:33.457.151.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 11, 32bit 10MHz counter = 13.615201400 corrected by 40bit 10MHz counter of 13.079757700 obtained at 1Hz counter of 11. Result: 11.535445521
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-06-11T15:00:02.535.301.725
   Max 1st sample MIU time among all SL packets: 2023-06-11T15:00:33.457.010.725
   Min 1st sample XO time among all SL packets: 11.535305245
   Max 1st sample XO time among all SL packets: 42.457013625
   TM values from -266 to 221 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-11T15:00:02.535.301.725 (XO: 11.535305245 - 11.535611246, t=42.0 degC)  Rank: 00076
   TM values: Max =   195  Min =  -266  Ave =    -4.50  Std =    27.13
   Min TM value:-266 at an offset of 96.088 usec -> 2023-06-11T15:00:02.535.397.812 (XO: 11.535401333)
   Max TM value:195 at an offset of 65.325 usec -> 2023-06-11T15:00:02.535.367.050 (XO: 11.535370570)
   Max dTM/dt [1/us]:2.25e+03 at an offset of 96.200 usec -> 2023-06-11T15:00:02.535.397.925 (XO: 11.535401445)
   Min dTM/dt [1/us]:-2.16e+03 at an offset of 93.738 usec -> 2023-06-11T15:00:02.535.395.462 (XO: 11.535398983)

 - Block number: 2 starting at 2023-06-11T15:00:16.546.070.725 (XO: 25.546073836 - 25.546379837, t=42.0 degC)  Rank: 00081
   TM values: Max =   221  Min =  -233  Ave =    -4.43  Std =    25.20
   Min TM value:-233 at an offset of 160.163 usec -> 2023-06-11T15:00:16.546.230.888 (XO: 25.546233999)
   Max TM value:221 at an offset of 122.813 usec -> 2023-06-11T15:00:16.546.193.537 (XO: 25.546196649)
   Max dTM/dt [1/us]:1.51e+03 at an offset of 157.788 usec -> 2023-06-11T15:00:16.546.228.513 (XO: 25.546231624)
   Min dTM/dt [1/us]:-2.02e+03 at an offset of 122.925 usec -> 2023-06-11T15:00:16.546.193.650 (XO: 25.546196762)

 - Block number: 3 starting at 2023-06-11T15:00:33.456.721.725 (XO: 42.456724624 - 42.457030625, t=42.0 degC)  Rank: 00080
   TM values: Max =   216  Min =  -265  Ave =    -4.52  Std =    22.63
   Min TM value:-265 at an offset of 122.400 usec -> 2023-06-11T15:00:33.456.844.125 (XO: 42.456847024)
   Max TM value:216 at an offset of 119.913 usec -> 2023-06-11T15:00:33.456.841.637 (XO: 42.456844536)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 122.513 usec -> 2023-06-11T15:00:33.456.844.237 (XO: 42.456847136)
   Min dTM/dt [1/us]:-2.09e+03 at an offset of 120.025 usec -> 2023-06-11T15:00:33.456.841.750 (XO: 42.456844649)
