-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jul  1 08:14:35 2023 UTC
-- File 20230611_130219_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.587310 to 35.377414
-each separately from 5s to 35s and from 87354us to 587599us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.499755 in 1 packets: 18
      8.289482 in 1 packets: 36
MIU packet time from 2023-06-11T13:02:27.587.310.000 to 2023-06-11T13:02:57.377.414.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.499755 in 1 packets: 18
      8.289482 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.5636740 to 36.5635703
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999240 in 1 packets: 18
      7.9999723 in 1 packets: 36
Temperature FPGA: 
      01 -> 53.7 degC in 54 packets (all packets)
Temperature XO: 
      C4 -> 42.0 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20230611_130219_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.1509851 to 7.1512741
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.4996790  -- OK for SL --
   MIU time from 2023-06-11T13:02:27.587.310.000 to 2023-06-11T13:02:27.587.599.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.499755

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.6509531 to 28.6512421
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.2894550  -- OK for SL --
   MIU time from 2023-06-11T13:02:49.087.354.000 to 2023-06-11T13:02:49.087.643.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.289482

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.9406971 to 36.9409861
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-11T13:02:57.377.125.000 to 2023-06-11T13:02:57.377.414.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.150985100 corrected by 40bit 10MHz counter of 6.563674000 obtained at 1Hz counter of 5. Result: 5.587313097
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-06-11T13:02:27.587.169.725
   Max 1st sample MIU time among all SL packets: 2023-06-11T13:02:57.377.273.725
   Min 1st sample XO time among all SL packets: 5.587172821
   Max 1st sample XO time among all SL packets: 35.377276845
   TM values from -269 to 202 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-11T13:02:27.587.169.725 (XO: 5.587172821 - 5.587478822, t=42.0 degC)  Rank: 00076
   TM values: Max =   176  Min =  -186  Ave =    -4.46  Std =    24.70
   Min TM value:-186 at an offset of 115.825 usec -> 2023-06-11T13:02:27.587.285.550 (XO: 5.587288647)
   Max TM value:176 at an offset of 118.550 usec -> 2023-06-11T13:02:27.587.288.275 (XO: 5.587291372)
   Max dTM/dt [1/us]:1.43e+03 at an offset of 171.676 usec -> 2023-06-11T13:02:27.587.341.400 (XO: 5.587344497)
   Min dTM/dt [1/us]:-1.59e+03 at an offset of 118.688 usec -> 2023-06-11T13:02:27.587.288.412 (XO: 5.587291509)

 - Block number: 2 starting at 2023-06-11T13:02:49.087.213.725 (XO: 27.087215130 - 27.087521131, t=42.0 degC)  Rank: 00076
   TM values: Max =   202  Min =  -230  Ave =    -4.38  Std =    20.03
   Min TM value:-230 at an offset of 118.913 usec -> 2023-06-11T13:02:49.087.332.637 (XO: 27.087334043)
   Max TM value:202 at an offset of 118.638 usec -> 2023-06-11T13:02:49.087.332.362 (XO: 27.087333768)
   Max dTM/dt [1/us]:1.96e+03 at an offset of 121.288 usec -> 2023-06-11T13:02:49.087.335.012 (XO: 27.087336418)
   Min dTM/dt [1/us]:-1.78e+03 at an offset of 118.738 usec -> 2023-06-11T13:02:49.087.332.462 (XO: 27.087333868)

 - Block number: 3 starting at 2023-06-11T13:02:57.376.984.725 (XO: 35.376987844 - 35.377293846, t=42.0 degC)  Rank: 00088
   TM values: Max =   191  Min =  -269  Ave =    -4.41  Std =    19.21
   Min TM value:-269 at an offset of 126.888 usec -> 2023-06-11T13:02:57.377.111.612 (XO: 35.377114732)
   Max TM value:191 at an offset of 126.650 usec -> 2023-06-11T13:02:57.377.111.375 (XO: 35.377114495)
   Max dTM/dt [1/us]:2.12e+03 at an offset of 129.275 usec -> 2023-06-11T13:02:57.377.114.000 (XO: 35.377117120)
   Min dTM/dt [1/us]:-2.41e+03 at an offset of 126.763 usec -> 2023-06-11T13:02:57.377.111.487 (XO: 35.377114607)
