-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jul  1 06:34:24 2023 UTC
-- File 20230611_110535_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 9.205516 to 22.846249
-each separately from 9s to 22s and from 205516us to 846249us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.640155 in 1 packets: 18
MIU packet time from 2023-06-11T11:05:47.205.516.000 to 2023-06-11T11:06:00.846.249.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.640155 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.7795239 to 23.7794790
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999551 in 1 packets: 18
Temperature FPGA: 
      00 -> 55.0 degC in 36 packets (all packets)
Temperature XO: 
      C4 -> 42.0 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230611_110535_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.9850411 to 10.9853301
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.6401110  -- OK for SL --
   MIU time from 2023-06-11T11:05:47.205.516.000 to 2023-06-11T11:05:47.205.805.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.640155

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      89 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.6254411 to 24.6257301
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-11T11:06:00.845.960.000 to 2023-06-11T11:06:00.846.249.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 9, 32bit 10MHz counter = 10.985041100 corrected by 40bit 10MHz counter of 10.779523900 obtained at 1Hz counter of 9. Result: 9.205517899
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-06-11T11:05:47.205.375.725
   Max 1st sample MIU time among all SL packets: 2023-06-11T11:06:00.846.108.725
   Min 1st sample XO time among all SL packets: 9.205377623
   Max 1st sample XO time among all SL packets: 22.846113786
   TM values from -293 to 194 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-11T11:05:47.205.375.725 (XO: 9.205377623 - 9.205683624, t=42.0 degC)  Rank: 00081
   TM values: Max =   194  Min =  -293  Ave =    -4.59  Std =    23.60
   Min TM value:-293 at an offset of 105.463 usec -> 2023-06-11T11:05:47.205.481.187 (XO: 9.205483086)
   Max TM value:194 at an offset of 105.200 usec -> 2023-06-11T11:05:47.205.480.925 (XO: 9.205482824)
   Max dTM/dt [1/us]:2.14e+03 at an offset of 107.800 usec -> 2023-06-11T11:05:47.205.483.525 (XO: 9.205485424)
   Min dTM/dt [1/us]:-2.43e+03 at an offset of 105.338 usec -> 2023-06-11T11:05:47.205.481.062 (XO: 9.205482961)

 - Block number: 2 starting at 2023-06-11T11:06:00.845.819.725 (XO: 22.845824785 - 22.846130786, t=42.0 degC)  Rank: 00089
   TM values: Max =   192  Min =  -279  Ave =    -4.47  Std =    19.09
   Min TM value:-279 at an offset of 105.838 usec -> 2023-06-11T11:06:00.845.925.562 (XO: 22.845930623)
   Max TM value:192 at an offset of 110.725 usec -> 2023-06-11T11:06:00.845.930.450 (XO: 22.845935511)
   Max dTM/dt [1/us]:1.60e+03 at an offset of 108.263 usec -> 2023-06-11T11:06:00.845.927.987 (XO: 22.845933048)
   Min dTM/dt [1/us]:-1.84e+03 at an offset of 105.700 usec -> 2023-06-11T11:06:00.845.925.425 (XO: 22.845930486)
