-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jul  1 06:32:41 2023 UTC
-- File 20230611_110341_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.384287 to 38.724371
-each separately from 0s to 38s and from 384287us to 884758us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.499893 in 1 packets: 18
      12.839324 in 1 packets: 36
MIU packet time from 2023-06-11T11:03:44.384.287.000 to 2023-06-11T11:04:22.724.371.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      25.499893 in 1 packets: 18
      12.839324 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.6996543 to 39.6995227
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.9999134 in 1 packets: 18
      12.9999550 in 1 packets: 36
Temperature FPGA: 
      01 -> 53.7 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      00 -> 55.0 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      FF -> 56.3 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C4 -> 42.0 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      2 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20230611_110341_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.0839428 to 2.0842318
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.4998070  -- OK for SL --
   MIU time from 2023-06-11T11:03:44.384.287.000 to 2023-06-11T11:03:44.384.576.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.499893

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      88 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.5840388 to 27.5843278
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.8392790  -- OK for SL --
   MIU time from 2023-06-11T11:04:09.884.469.000 to 2023-06-11T11:04:09.884.758.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.839324

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.4236068 to 40.4238958
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-11T11:04:22.724.082.000 to 2023-06-11T11:04:22.724.371.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.083942800 corrected by 40bit 10MHz counter of 1.699654300 obtained at 1Hz counter of 0. Result: 0.384289845
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-06-11T11:03:44.384.146.725
   Max 1st sample MIU time among all SL packets: 2023-06-11T11:04:22.724.230.725
   Min 1st sample XO time among all SL packets: 0.384149570
   Max 1st sample XO time among all SL packets: 38.724235287
   TM values from -280 to 208 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-11T11:03:44.384.146.725 (XO: 0.384149570 - 0.384455571, t=42.0 degC)  Rank: 00079
   TM values: Max =   205  Min =  -280  Ave =    -4.50  Std =    25.78
   Min TM value:-280 at an offset of 113.350 usec -> 2023-06-11T11:03:44.384.260.075 (XO: 0.384262920)
   Max TM value:205 at an offset of 113.600 usec -> 2023-06-11T11:03:44.384.260.325 (XO: 0.384263170)
   Max dTM/dt [1/us]:2.27e+03 at an offset of 113.450 usec -> 2023-06-11T11:03:44.384.260.175 (XO: 0.384263020)
   Min dTM/dt [1/us]:-2.23e+03 at an offset of 111.000 usec -> 2023-06-11T11:03:44.384.257.725 (XO: 0.384260570)

 - Block number: 2 starting at 2023-06-11T11:04:09.884.328.725 (XO: 25.884333920 - 25.884639921, t=42.0 degC)  Rank: 00088
   TM values: Max =   179  Min =  -252  Ave =    -4.44  Std =    17.99
   Min TM value:-252 at an offset of 122.488 usec -> 2023-06-11T11:04:09.884.451.212 (XO: 25.884456408)
   Max TM value:179 at an offset of 119.950 usec -> 2023-06-11T11:04:09.884.448.675 (XO: 25.884453871)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 122.588 usec -> 2023-06-11T11:04:09.884.451.312 (XO: 25.884456508)
   Min dTM/dt [1/us]:-2.15e+03 at an offset of 120.075 usec -> 2023-06-11T11:04:09.884.448.800 (XO: 25.884453996)

 - Block number: 3 starting at 2023-06-11T11:04:22.723.941.725 (XO: 38.723946286 - 38.724252287, t=42.0 degC)  Rank: 00082
   TM values: Max =   208  Min =  -272  Ave =    -4.43  Std =    25.91
   Min TM value:-272 at an offset of 111.563 usec -> 2023-06-11T11:04:22.724.053.287 (XO: 38.724057849)
   Max TM value:208 at an offset of 109.088 usec -> 2023-06-11T11:04:22.724.050.812 (XO: 38.724055374)
   Max dTM/dt [1/us]:2.15e+03 at an offset of 111.663 usec -> 2023-06-11T11:04:22.724.053.387 (XO: 38.724057949)
   Min dTM/dt [1/us]:-2.32e+03 at an offset of 109.200 usec -> 2023-06-11T11:04:22.724.050.925 (XO: 38.724055487)
