-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jul  1 05:39:02 2023 UTC
-- File 20230611_091941_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.998343 to 54.085820
-each separately from 12s to 54s and from 85531us to 998632us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.748320 in 1 packets: 18
      21.043225 in 1 packets: 36
      15.294776 in 1 packets: 54
MIU packet time from 2023-06-11T09:19:56.998.343.000 to 2023-06-11T09:20:38.085.820.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.748320 in 1 packets: 18
      21.043225 in 1 packets: 36
      15.294776 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.6915265 to 55.6913788
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      4.9999824 in 1 packets: 18
      20.9999261 in 1 packets: 36
      15.9999438 in 1 packets: 54
Temperature FPGA: 
      02 -> 52.4 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      01 -> 53.7 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C4 -> 42.0 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20230611_091941_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.6898706 to 14.6901596
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 4.7483030  -- OK for SL --
   MIU time from 2023-06-11T09:19:56.998.343.000 to 2023-06-11T09:19:56.998.632.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 4.748320

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.4384626 to 19.4387516
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.0431510  -- OK for SL --
   MIU time from 2023-06-11T09:20:01.746.952.000 to 2023-06-11T09:20:01.747.241.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.043225

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      90 in 18 packets (all packets)  -- OK --
   10 MHz counter from 40.4819026 to 40.4821916
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.2947190  -- OK for SL --
   MIU time from 2023-06-11T09:20:22.790.466.000 to 2023-06-11T09:20:22.790.755.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.294776

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 55.7769106 to 55.7771996
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-11T09:20:38.085.531.000 to 2023-06-11T09:20:38.085.820.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.689870600 corrected by 40bit 10MHz counter of 13.691526500 obtained at 1Hz counter of 12. Result: 12.998347694
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-06-11T09:19:56.998.202.725
   Max 1st sample MIU time among all SL packets: 2023-06-11T09:20:38.085.679.725
   Min 1st sample XO time among all SL packets: 12.998207419
   Max 1st sample XO time among all SL packets: 54.085680825
   TM values from -291 to 209 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-11T09:19:56.998.202.725 (XO: 12.998207419 - 12.998513420, t=42.0 degC)  Rank: 00085
   TM values: Max =   198  Min =  -291  Ave =    -4.40  Std =    21.98
   Min TM value:-291 at an offset of 119.813 usec -> 2023-06-11T09:19:56.998.322.537 (XO: 12.998327231)
   Max TM value:198 at an offset of 107.488 usec -> 2023-06-11T09:19:56.998.310.212 (XO: 12.998314906)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 122.238 usec -> 2023-06-11T09:19:56.998.324.962 (XO: 12.998329656)
   Min dTM/dt [1/us]:-2.26e+03 at an offset of 119.688 usec -> 2023-06-11T09:19:56.998.322.412 (XO: 12.998327106)

 - Block number: 2 starting at 2023-06-11T09:20:01.746.811.725 (XO: 17.746816114 - 17.747122115, t=42.0 degC)  Rank: 00079
   TM values: Max =   183  Min =  -264  Ave =    -4.28  Std =    22.59
   Min TM value:-264 at an offset of 121.400 usec -> 2023-06-11T09:20:01.746.933.125 (XO: 17.746937514)
   Max TM value:183 at an offset of 108.563 usec -> 2023-06-11T09:20:01.746.920.287 (XO: 17.746924676)
   Max dTM/dt [1/us]:1.62e+03 at an offset of 123.888 usec -> 2023-06-11T09:20:01.746.935.612 (XO: 17.746940001)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 108.675 usec -> 2023-06-11T09:20:01.746.920.400 (XO: 17.746924789)

 - Block number: 3 starting at 2023-06-11T09:20:22.790.325.725 (XO: 38.790330091 - 38.790636092, t=42.0 degC)  Rank: 00090
   TM values: Max =   209  Min =  -283  Ave =    -4.47  Std =    19.12
   Min TM value:-283 at an offset of 121.050 usec -> 2023-06-11T09:20:22.790.446.775 (XO: 38.790451142)
   Max TM value:209 at an offset of 108.838 usec -> 2023-06-11T09:20:22.790.434.562 (XO: 38.790438929)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 123.413 usec -> 2023-06-11T09:20:22.790.449.137 (XO: 38.790453504)
   Min dTM/dt [1/us]:-2.01e+03 at an offset of 108.963 usec -> 2023-06-11T09:20:22.790.434.687 (XO: 38.790439054)

 - Block number: 4 starting at 2023-06-11T09:20:38.085.390.725 (XO: 54.085391824 - 54.085697825, t=42.0 degC)  Rank: 00080
   TM values: Max =   174  Min =  -260  Ave =    -4.34  Std =    18.72
   Min TM value:-260 at an offset of 96.600 usec -> 2023-06-11T09:20:38.085.487.325 (XO: 54.085488424)
   Max TM value:174 at an offset of 96.338 usec -> 2023-06-11T09:20:38.085.487.062 (XO: 54.085488162)
   Max dTM/dt [1/us]:1.77e+03 at an offset of 99.038 usec -> 2023-06-11T09:20:38.085.489.762 (XO: 54.085490862)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 96.475 usec -> 2023-06-11T09:20:38.085.487.200 (XO: 54.085488299)
