-- OK --
-----------------> BLESK ver.2021-02-27 run on Tue Jun  6 02:05:37 2023 UTC
-- File 20230606_015716_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.427957 to 76.871974
-each separately from 12s to 76s and from 115756us to 871974us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.315495 in 1 packets: 18
      3.371726 in 1 packets: 36
      20.021780 in 1 packets: 54
      8.213307 in 1 packets: 72
      16.519975 in 1 packets: 90
MIU packet time from 2023-06-06T01:57:31.427.957.000 to 2023-06-06T01:58:35.871.974.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.315495 in 1 packets: 18
      3.371726 in 1 packets: 36
      20.021780 in 1 packets: 54
      8.213307 in 1 packets: 72
      16.519975 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.7551143 to 77.7548906
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999441 in 1 packets: 18
      3.9999860 in 1 packets: 36
      19.9999301 in 1 packets: 54
      7.9999721 in 1 packets: 72
      15.9999440 in 1 packets: 90
Temperature FPGA: 
      00 -> 55.0 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      01 -> 53.7 degC in 72 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C3 -> 40.4 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 72 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Number of generated alerts: 
      0 in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      3 in 1 packets: 37
      4 in 71 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20230606_015716_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      64 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.1830727 to 14.1833617
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.3154390  -- OK for SL --
   MIU time from 2023-06-06T01:57:31.427.957.000 to 2023-06-06T01:57:31.428.246.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.315495

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      67 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.4988007 to 30.4990897
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 3.3717110  -- OK for SL --
   MIU time from 2023-06-06T01:57:47.743.741.000 to 2023-06-06T01:57:47.744.030.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 3.371726

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      187 in 18 packets (all packets)  -- OK --
   10 MHz counter from 33.8708007 to 33.8710897
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.0217110  -- OK for SL --
   MIU time from 2023-06-06T01:57:51.115.756.000 to 2023-06-06T01:57:51.116.045.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.021780

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      65 in 18 packets (all packets)  -- OK --
   10 MHz counter from 53.8928007 to 53.8930897
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 8.2132790  -- OK for SL --
   MIU time from 2023-06-06T01:58:11.137.825.000 to 2023-06-06T01:58:11.138.114.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 8.213307

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 62.1063687 to 62.1066577
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.5199190  -- OK for SL --
   MIU time from 2023-06-06T01:58:19.351.421.000 to 2023-06-06T01:58:19.351.710.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.519975

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      64 in 18 packets (all packets)  -- OK --
   10 MHz counter from 78.6265767 to 78.6268657
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-06T01:58:35.871.685.000 to 2023-06-06T01:58:35.871.974.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.183072700 corrected by 40bit 10MHz counter of 13.755114300 obtained at 1Hz counter of 12. Result: 12.427959898
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-06-06T01:57:31.427.816.725
   Max 1st sample MIU time among all SL packets: 2023-06-06T01:58:35.871.833.725
   Min 1st sample XO time among all SL packets: 12.427819622
   Max 1st sample XO time among all SL packets: 76.871837876
   TM values from -955 to 513 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-06T01:57:31.427.816.725 (XO: 12.427819622 - 12.428125623, t=40.4 degC)  Rank: 00064
   TM values: Max =   171  Min =  -200  Ave =    -4.43  Std =    18.20
   Min TM value:-200 at an offset of 110.988 usec -> 2023-06-06T01:57:31.427.927.712 (XO: 12.427930610)
   Max TM value:171 at an offset of 110.675 usec -> 2023-06-06T01:57:31.427.927.400 (XO: 12.427930298)
   Max dTM/dt [1/us]:1.52e+03 at an offset of 113.313 usec -> 2023-06-06T01:57:31.427.930.037 (XO: 12.427932935)
   Min dTM/dt [1/us]:-1.62e+03 at an offset of 110.813 usec -> 2023-06-06T01:57:31.427.927.537 (XO: 12.427930435)

 - Block number: 2 starting at 2023-06-06T01:57:47.743.600.725 (XO: 28.743604628 - 28.743910629, t=40.4 degC)  Rank: 00067
   TM values: Max =   162  Min =  -189  Ave =    -4.51  Std =    15.50
   Min TM value:-189 at an offset of 122.100 usec -> 2023-06-06T01:57:47.743.722.825 (XO: 28.743726728)
   Max TM value:162 at an offset of 109.588 usec -> 2023-06-06T01:57:47.743.710.312 (XO: 28.743714215)
   Max dTM/dt [1/us]:1.38e+03 at an offset of 124.500 usec -> 2023-06-06T01:57:47.743.725.225 (XO: 28.743729128)
   Min dTM/dt [1/us]:-1.46e+03 at an offset of 109.700 usec -> 2023-06-06T01:57:47.743.710.425 (XO: 28.743714328)

 - Block number: 3 starting at 2023-06-06T01:57:51.115.615.725 (XO: 32.115616430 - 32.115922431, t=40.4 degC)  Rank: 00187
   TM values: Max =   513  Min =  -955  Ave =    -4.49  Std =    66.70
   Min TM value:-955 at an offset of 110.813 usec -> 2023-06-06T01:57:51.115.726.537 (XO: 32.115727243)
   Max TM value:513 at an offset of 111.200 usec -> 2023-06-06T01:57:51.115.726.925 (XO: 32.115727630)
   Max dTM/dt [1/us]:8.82e+03 at an offset of 111.013 usec -> 2023-06-06T01:57:51.115.726.737 (XO: 32.115727443)
   Min dTM/dt [1/us]:-5.46e+03 at an offset of 110.763 usec -> 2023-06-06T01:57:51.115.726.487 (XO: 32.115727193)

 - Block number: 4 starting at 2023-06-06T01:58:11.137.684.725 (XO: 52.137686407 - 52.137992408, t=40.4 degC)  Rank: 00065
   TM values: Max =   168  Min =  -180  Ave =    -4.52  Std =    20.25
   Min TM value:-180 at an offset of 125.438 usec -> 2023-06-06T01:58:11.137.810.162 (XO: 52.137811845)
   Max TM value:168 at an offset of 125.138 usec -> 2023-06-06T01:58:11.137.809.862 (XO: 52.137811545)
   Max dTM/dt [1/us]:1.49e+03 at an offset of 127.763 usec -> 2023-06-06T01:58:11.137.812.487 (XO: 52.137814170)
   Min dTM/dt [1/us]:-1.67e+03 at an offset of 125.288 usec -> 2023-06-06T01:58:11.137.810.012 (XO: 52.137811695)

 - Block number: 5 starting at 2023-06-06T01:58:19.351.280.725 (XO: 60.351283054 - 60.351589056, t=40.4 degC)  Rank: 00072
   TM values: Max =   493  Min =  -250  Ave =    -4.59  Std =    20.65
   Min TM value:-250 at an offset of 102.163 usec -> 2023-06-06T01:58:19.351.382.887 (XO: 60.351385217)
   Max TM value:493 at an offset of 101.688 usec -> 2023-06-06T01:58:19.351.382.412 (XO: 60.351384742)
   Max dTM/dt [1/us]:2.57e+03 at an offset of 101.600 usec -> 2023-06-06T01:58:19.351.382.325 (XO: 60.351384655)
   Min dTM/dt [1/us]:-2.64e+03 at an offset of 90.838 usec -> 2023-06-06T01:58:19.351.371.562 (XO: 60.351373892)

 - Block number: 6 starting at 2023-06-06T01:58:35.871.544.725 (XO: 76.871548875 - 76.871854876, t=40.4 degC)  Rank: 00064
   TM values: Max =   182  Min =  -179  Ave =    -4.54  Std =    12.64
   Min TM value:-179 at an offset of 112.413 usec -> 2023-06-06T01:58:35.871.657.137 (XO: 76.871661288)
   Max TM value:182 at an offset of 112.138 usec -> 2023-06-06T01:58:35.871.656.862 (XO: 76.871661013)
   Max dTM/dt [1/us]:1.42e+03 at an offset of 112.038 usec -> 2023-06-06T01:58:35.871.656.762 (XO: 76.871660913)
   Min dTM/dt [1/us]:-1.67e+03 at an offset of 112.238 usec -> 2023-06-06T01:58:35.871.656.962 (XO: 76.871661113)
