-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon Jun  5 23:51:29 2023 UTC
-- File 20230605_155954_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177028
-each separately from 0s to 14s and from 0us to 841254us.
   Differences in neigboring packets: 
      0.833939 in 2 packets: 1, 7
      0.833943 in 6 packets: 2, 4, 10, 12, 15, 17
      0.833944 in 8 packets: 3, 5, 6, 8, 9, 11, 14, 16
      0.833940 in 1 packets: 13
MIU packet time from 2023-06-05T15:59:57.000.000.000 to 2023-06-05T16:00:11.177.028.000
   Differences in neigboring packets: 
      0.833939 in 2 packets: 1, 7
      0.833943 in 6 packets: 2, 4, 10, 12, 15, 17
      0.833944 in 8 packets: 3, 5, 6, 8, 9, 11, 14, 16
      0.833940 in 1 packets: 13
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.1115489 to 16.1114993
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999965 in 8 packets: 2, 4, 5, 6, 9, 11, 14, 17
      0.9999964 in 6 packets: 3, 8, 10, 12, 15, 16
   XO PPM: 
      -3.5 in 8 packets: 2, 4, 5, 6, 9, 11, 14, 17
      -3.6 in 6 packets: 3, 8, 10, 12, 15, 16
Temperature FPGA: 
      01 -> 53.7 degC in 16 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 12, 13, 14, 15, 16, 17
      02 -> 52.4 degC in 2 packets: 11, 18
Temperature XO: 
      C3 -> 40.4 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230605_155954_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.1115490 to 16.2885290
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-05T15:59:57.000.000.000 to 2023-06-05T16:00:11.177.028.000
      Difference between neigboring packets: 
      0.833939 in 2 packets: 1, 7
      0.833943 in 6 packets: 2, 4, 10, 12, 15, 17
      0.833944 in 8 packets: 3, 5, 6, 8, 9, 11, 14, 16
      0.833940 in 1 packets: 13
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.111549000 corrected by 40bit 10MHz counter of 2.111548900 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SR packets: 2023-06-05T15:59:56.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-06-05T16:00:11.177.027.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177030245
   TM values from -102 to 92 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-06-05T15:59:56.999.999.925 (XO: 0.000000025 - 0.000017025, t=40.4 degC)
   TM values: Max =    81  Min =   -94  Ave =    -4.49  Std =    38.29

 - Packet number: 2 starting at 2023-06-05T15:59:57.833.938.925 (XO: 0.833942944 - 0.833959944, t=40.4 degC)
   TM values: Max =    76  Min =   -88  Ave =    -4.55  Std =    35.68

 - Packet number: 3 starting at 2023-06-05T15:59:58.667.881.925 (XO: 1.667885929 - 1.667902929, t=40.4 degC)
   TM values: Max =    52  Min =   -63  Ave =    -4.62  Std =    24.59

 - Packet number: 4 starting at 2023-06-05T15:59:59.501.825.925 (XO: 2.501828881 - 2.501845881, t=40.4 degC)
   TM values: Max =    45  Min =   -54  Ave =    -4.32  Std =    22.10

 - Packet number: 5 starting at 2023-06-05T16:00:00.335.768.925 (XO: 3.335771800 - 3.335788800, t=40.4 degC)
   TM values: Max =    54  Min =   -66  Ave =    -4.69  Std =    24.14

 - Packet number: 6 starting at 2023-06-05T16:00:01.169.712.925 (XO: 4.169714719 - 4.169731719, t=40.4 degC)
   TM values: Max =    65  Min =   -66  Ave =    -4.58  Std =    26.44

 - Packet number: 7 starting at 2023-06-05T16:00:02.003.656.925 (XO: 5.003657638 - 5.003674638, t=40.4 degC)
   TM values: Max =    83  Min =   -95  Ave =    -4.63  Std =    32.50

 - Packet number: 8 starting at 2023-06-05T16:00:02.837.595.925 (XO: 5.837600640 - 5.837617640, t=40.4 degC)
   TM values: Max =    79  Min =   -91  Ave =    -4.42  Std =    30.80

 - Packet number: 9 starting at 2023-06-05T16:00:03.671.539.925 (XO: 6.671543575 - 6.671560575, t=40.4 degC)
   TM values: Max =    74  Min =   -86  Ave =    -4.61  Std =    31.51

 - Packet number: 10 starting at 2023-06-05T16:00:04.505.483.925 (XO: 7.505486545 - 7.505503545, t=40.4 degC)
   TM values: Max =    77  Min =   -87  Ave =    -4.65  Std =    33.70

 - Packet number: 11 starting at 2023-06-05T16:00:05.339.426.925 (XO: 8.339429513 - 8.339446513, t=40.4 degC)
   TM values: Max =    62  Min =   -72  Ave =    -4.30  Std =    28.00

 - Packet number: 12 starting at 2023-06-05T16:00:06.173.370.925 (XO: 9.173372449 - 9.173389449, t=40.4 degC)
   TM values: Max =    78  Min =   -86  Ave =    -4.62  Std =    32.01

 - Packet number: 13 starting at 2023-06-05T16:00:07.007.313.925 (XO: 10.007315451 - 10.007332451, t=40.4 degC)
   TM values: Max =    44  Min =   -52  Ave =    -4.69  Std =    19.83

 - Packet number: 14 starting at 2023-06-05T16:00:07.841.253.925 (XO: 10.841258369 - 10.841275369, t=40.4 degC)
   TM values: Max =    64  Min =   -73  Ave =    -4.54  Std =    28.86

 - Packet number: 15 starting at 2023-06-05T16:00:08.675.197.925 (XO: 11.675201356 - 11.675218356, t=40.4 degC)
   TM values: Max =    76  Min =   -88  Ave =    -4.79  Std =    38.18

 - Packet number: 16 starting at 2023-06-05T16:00:09.509.140.925 (XO: 12.509144358 - 12.509161358, t=40.4 degC)
   TM values: Max =    92  Min =  -102  Ave =    -4.56  Std =    48.63

 - Packet number: 17 starting at 2023-06-05T16:00:10.343.084.925 (XO: 13.343087326 - 13.343104326, t=40.4 degC)
   TM values: Max =    82  Min =   -95  Ave =    -4.36  Std =    42.87

 - Packet number: 18 starting at 2023-06-05T16:00:11.177.027.925 (XO: 14.177030245 - 14.177047245, t=40.4 degC)
   TM values: Max =    78  Min =   -85  Ave =    -4.66  Std =    36.33
