-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jun  4 03:28:58 2023 UTC
-- File 20230603_155234_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.168711 to 57.506546
-each separately from 0s to 57s and from 168711us to 506546us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.048904 in 1 packets: 18
      19.078866 in 1 packets: 36
      22.208909 in 1 packets: 54
MIU packet time from 2023-06-03T15:52:37.168.711.000 to 2023-06-03T15:53:34.506.546.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.048904 in 1 packets: 18
      19.078866 in 1 packets: 36
      22.208909 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.8541531 to 58.8539512
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.9999432 in 1 packets: 18
      18.9999327 in 1 packets: 36
      21.9999222 in 1 packets: 54
Temperature FPGA: 
      02 -> 52.4 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      01 -> 53.7 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature XO: 
      C3 -> 40.4 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 53 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 1 packets: 55
      3 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 53 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 1 packets: 55
      3 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20230603_155234_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.0228652 to 2.0231542
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.0488470  -- OK for SL --
   MIU time from 2023-06-03T15:52:37.168.711.000 to 2023-06-03T15:52:37.169.000.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.048904

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.0720012 to 18.0722902
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.0787990  -- OK for SL --
   MIU time from 2023-06-03T15:52:53.217.904.000 to 2023-06-03T15:52:53.218.193.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.078866

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 37.1510892 to 37.1513782
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.2088310  -- OK for SL --
   MIU time from 2023-06-03T15:53:12.297.059.000 to 2023-06-03T15:53:12.297.348.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.208909

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      86 in 18 packets (all packets)  -- OK --
   10 MHz counter from 59.3602092 to 59.3604982
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-03T15:53:34.506.257.000 to 2023-06-03T15:53:34.506.546.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.022865200 corrected by 40bit 10MHz counter of 1.854153100 obtained at 1Hz counter of 0. Result: 0.168712707
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-06-03T15:52:37.168.570.725
   Max 1st sample MIU time among all SL packets: 2023-06-03T15:53:34.506.405.725
   Min 1st sample XO time among all SL packets: 0.168572432
   Max 1st sample XO time among all SL packets: 57.506408497
   TM values from -251 to 193 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-03T15:52:37.168.570.725 (XO: 0.168572432 - 0.168878433, t=40.4 degC)  Rank: 00083
   TM values: Max =   182  Min =  -233  Ave =    -4.43  Std =    18.38
   Min TM value:-233 at an offset of 117.125 usec -> 2023-06-03T15:52:37.168.687.850 (XO: 0.168689557)
   Max TM value:182 at an offset of 122.075 usec -> 2023-06-03T15:52:37.168.692.800 (XO: 0.168694507)
   Max dTM/dt [1/us]:1.69e+03 at an offset of 119.738 usec -> 2023-06-03T15:52:37.168.690.462 (XO: 0.168692170)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 104.688 usec -> 2023-06-03T15:52:37.168.675.412 (XO: 0.168677120)

 - Block number: 2 starting at 2023-06-03T15:52:53.217.763.725 (XO: 16.217765387 - 16.218071388, t=40.4 degC)  Rank: 00076
   TM values: Max =   186  Min =  -223  Ave =    -4.54  Std =    20.63
   Min TM value:-223 at an offset of 108.363 usec -> 2023-06-03T15:52:53.217.872.087 (XO: 16.217873750)
   Max TM value:186 at an offset of 103.950 usec -> 2023-06-03T15:52:53.217.867.675 (XO: 16.217869338)
   Max dTM/dt [1/us]:1.68e+03 at an offset of 108.475 usec -> 2023-06-03T15:52:53.217.872.200 (XO: 16.217873863)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 105.850 usec -> 2023-06-03T15:52:53.217.869.575 (XO: 16.217871238)

 - Block number: 3 starting at 2023-06-03T15:53:12.296.918.725 (XO: 35.296920964 - 35.297226965, t=40.4 degC)  Rank: 00078
   TM values: Max =   193  Min =  -251  Ave =    -4.41  Std =    16.30
   Min TM value:-251 at an offset of 110.650 usec -> 2023-06-03T15:53:12.297.029.375 (XO: 35.297031615)
   Max TM value:193 at an offset of 108.125 usec -> 2023-06-03T15:53:12.297.026.850 (XO: 35.297029090)
   Max dTM/dt [1/us]:2.04e+03 at an offset of 110.763 usec -> 2023-06-03T15:53:12.297.029.487 (XO: 35.297031727)
   Min dTM/dt [1/us]:-1.85e+03 at an offset of 108.213 usec -> 2023-06-03T15:53:12.297.026.937 (XO: 35.297029177)

 - Block number: 4 starting at 2023-06-03T15:53:34.506.116.725 (XO: 57.506119496 - 57.506425497, t=40.4 degC)  Rank: 00086
   TM values: Max =   192  Min =  -237  Ave =    -4.49  Std =    19.06
   Min TM value:-237 at an offset of 117.225 usec -> 2023-06-03T15:53:34.506.233.950 (XO: 57.506236722)
   Max TM value:192 at an offset of 122.263 usec -> 2023-06-03T15:53:34.506.238.987 (XO: 57.506241759)
   Max dTM/dt [1/us]:1.73e+03 at an offset of 119.838 usec -> 2023-06-03T15:53:34.506.236.562 (XO: 57.506239334)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 104.788 usec -> 2023-06-03T15:53:34.506.221.512 (XO: 57.506224284)
