-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jun  4 03:13:22 2023 UTC
-- File 20230603_154014_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.694388 to 22.885988
-each separately from 13s to 22s and from 694388us to 885988us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.191022 in 1 packets: 18
MIU packet time from 2023-06-03T15:40:30.694.388.000 to 2023-06-03T15:40:39.885.988.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.191022 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.8930677 to 23.8930359
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      8.9999682 in 1 packets: 18
Temperature FPGA: 
      02 -> 52.4 degC in 36 packets (all packets)
Temperature XO: 
      C3 -> 40.4 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230603_154014_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.5874567 to 15.5877457
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.1909910  -- OK for SL --
   MIU time from 2023-06-03T15:40:30.694.388.000 to 2023-06-03T15:40:30.694.677.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.191022

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.7787367 to 24.7790257
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-03T15:40:39.885.699.000 to 2023-06-03T15:40:39.885.988.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.587456700 corrected by 40bit 10MHz counter of 14.893067700 obtained at 1Hz counter of 13. Result: 13.694391430
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-06-03T15:40:30.694.247.725
   Max 1st sample MIU time among all SL packets: 2023-06-03T15:40:39.885.847.725
   Min 1st sample XO time among all SL packets: 13.694251155
   Max 1st sample XO time among all SL packets: 22.885852714
   TM values from -253 to 211 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-03T15:40:30.694.247.725 (XO: 13.694251155 - 13.694557156, t=40.4 degC)  Rank: 00078
   TM values: Max =   211  Min =  -253  Ave =    -4.43  Std =    25.42
   Min TM value:-253 at an offset of 127.275 usec -> 2023-06-03T15:40:30.694.375.000 (XO: 13.694378430)
   Max TM value:211 at an offset of 126.988 usec -> 2023-06-03T15:40:30.694.374.712 (XO: 13.694378143)
   Max dTM/dt [1/us]:1.94e+03 at an offset of 129.638 usec -> 2023-06-03T15:40:30.694.377.362 (XO: 13.694380793)
   Min dTM/dt [1/us]:-2.15e+03 at an offset of 127.100 usec -> 2023-06-03T15:40:30.694.374.825 (XO: 13.694378255)

 - Block number: 2 starting at 2023-06-03T15:40:39.885.558.725 (XO: 22.885563713 - 22.885869714, t=40.4 degC)  Rank: 00082
   TM values: Max =   176  Min =  -226  Ave =    -4.44  Std =    18.63
   Min TM value:-226 at an offset of 125.775 usec -> 2023-06-03T15:40:39.885.684.500 (XO: 22.885689488)
   Max TM value:176 at an offset of 111.200 usec -> 2023-06-03T15:40:39.885.669.925 (XO: 22.885674913)
   Max dTM/dt [1/us]:1.75e+03 at an offset of 125.888 usec -> 2023-06-03T15:40:39.885.684.612 (XO: 22.885689601)
   Min dTM/dt [1/us]:-1.59e+03 at an offset of 111.313 usec -> 2023-06-03T15:40:39.885.670.037 (XO: 22.885675026)
