-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun Jun  4 00:06:27 2023 UTC
-- File 20230603_103054_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.381996 to 24.558320
-each separately from 14s to 24s and from 381996us to 558320us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.175746 in 1 packets: 18
MIU packet time from 2023-06-03T10:31:11.381.996.000 to 2023-06-03T10:31:21.558.320.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.175746 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.6152276 to 25.6151926
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999650 in 1 packets: 18
Temperature FPGA: 
      01 -> 53.7 degC in 36 packets (all packets)
Temperature XO: 
      C3 -> 40.4 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230603_103054_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.9972250 to 15.9975140
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.1757110  -- OK for SL --
   MIU time from 2023-06-03T10:31:11.381.996.000 to 2023-06-03T10:31:11.382.285.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.175746

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      97 in 18 packets (all packets)  -- OK --
   10 MHz counter from 26.1732250 to 26.1735140
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-03T10:31:21.558.031.000 to 2023-06-03T10:31:21.558.320.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 15.997225000 corrected by 40bit 10MHz counter of 15.615227600 obtained at 1Hz counter of 14. Result: 14.381998737
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-06-03T10:31:11.381.855.725
   Max 1st sample MIU time among all SL packets: 2023-06-03T10:31:21.558.179.725
   Min 1st sample XO time among all SL packets: 14.381858462
   Max 1st sample XO time among all SL packets: 24.558183079
   TM values from -379 to 309 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-03T10:31:11.381.855.725 (XO: 14.381858462 - 14.382164463, t=40.4 degC)  Rank: 00078
   TM values: Max =   204  Min =  -248  Ave =    -4.59  Std =    17.04
   Min TM value:-248 at an offset of 111.513 usec -> 2023-06-03T10:31:11.381.967.237 (XO: 14.381969974)
   Max TM value:204 at an offset of 111.250 usec -> 2023-06-03T10:31:11.381.966.975 (XO: 14.381969712)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 113.863 usec -> 2023-06-03T10:31:11.381.969.587 (XO: 14.381972324)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 111.375 usec -> 2023-06-03T10:31:11.381.967.100 (XO: 14.381969837)

 - Block number: 2 starting at 2023-06-03T10:31:21.557.890.725 (XO: 24.557894078 - 24.558200079, t=40.4 degC)  Rank: 00097
   TM values: Max =   309  Min =  -379  Ave =    -4.20  Std =    31.19
   Min TM value:-379 at an offset of 296.651 usec -> 2023-06-03T10:31:21.558.187.376 (XO: 24.558190729)
   Max TM value:309 at an offset of 124.363 usec -> 2023-06-03T10:31:21.558.015.087 (XO: 24.558018441)
   Max dTM/dt [1/us]:3.09e+03 at an offset of 292.001 usec -> 2023-06-03T10:31:21.558.182.726 (XO: 24.558186079)
   Min dTM/dt [1/us]:-3.48e+03 at an offset of 247.463 usec -> 2023-06-03T10:31:21.558.138.188 (XO: 24.558141541)
