-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jun  3 07:37:05 2023 UTC
-- File 20230602_184014_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.755301 to 27.680496
-each separately from 2s to 27s and from 680207us to 755590us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.924617 in 1 packets: 18
MIU packet time from 2023-06-02T18:40:19.755.301.000 to 2023-06-02T18:40:44.680.496.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.924617 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.4306149 to 28.4305247
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      24.9999098 in 1 packets: 18
Temperature FPGA: 
      00 -> 55.0 degC in 36 packets (all packets)
Temperature XO: 
      C4 -> 42.0 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 17 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230602_184014_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      64 in 18 packets (all packets)  -- OK --
   10 MHz counter from 4.1859175 to 4.1862065
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.9245270  -- OK for SL --
   MIU time from 2023-06-02T18:40:19.755.301.000 to 2023-06-02T18:40:19.755.590.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.924617

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.1107335 to 29.1110225
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-02T18:40:44.680.207.000 to 2023-06-02T18:40:44.680.496.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 4.185917500 corrected by 40bit 10MHz counter of 3.430614900 obtained at 1Hz counter of 2. Result: 2.755305319
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2023-06-02T18:40:19.755.160.725
   Max 1st sample MIU time among all SL packets: 2023-06-02T18:40:44.680.355.725
   Min 1st sample XO time among all SL packets: 2.755165044
   Max 1st sample XO time among all SL packets: 27.680360042
   TM values from -207 to 201 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-02T18:40:19.755.160.725 (XO: 2.755165044 - 2.755471045, t=42.0 degC)  Rank: 00064
   TM values: Max =   154  Min =  -167  Ave =    -4.42  Std =    49.96
   Min TM value:-167 at an offset of 149.076 usec -> 2023-06-02T18:40:19.755.309.800 (XO: 2.755314119)
   Max TM value:154 at an offset of 146.788 usec -> 2023-06-02T18:40:19.755.307.513 (XO: 2.755311832)
   Max dTM/dt [1/us]:1.10e+03 at an offset of 117.825 usec -> 2023-06-02T18:40:19.755.278.550 (XO: 2.755282869)
   Min dTM/dt [1/us]:-9.93e+02 at an offset of 156.113 usec -> 2023-06-02T18:40:19.755.316.838 (XO: 2.755321157)

 - Block number: 2 starting at 2023-06-02T18:40:44.680.066.725 (XO: 27.680071041 - 27.680377042, t=42.0 degC)  Rank: 00081
   TM values: Max =   201  Min =  -207  Ave =    -4.41  Std =    66.40
   Min TM value:-207 at an offset of 123.213 usec -> 2023-06-02T18:40:44.680.189.937 (XO: 27.680194254)
   Max TM value:201 at an offset of 146.501 usec -> 2023-06-02T18:40:44.680.213.225 (XO: 27.680217542)
   Max dTM/dt [1/us]:1.86e+03 at an offset of 123.650 usec -> 2023-06-02T18:40:44.680.190.375 (XO: 27.680194692)
   Min dTM/dt [1/us]:-1.78e+03 at an offset of 154.976 usec -> 2023-06-02T18:40:44.680.221.700 (XO: 27.680226017)
