-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jun  3 01:56:26 2023 UTC
-- File 20230602_110035_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.310431 to 68.732585
-each separately from 6s to 68s and from 310431us to 911313us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.600304 in 1 packets: 18
      12.579981 in 1 packets: 36
      14.040048 in 1 packets: 54
      21.200376 in 1 packets: 72
MIU packet time from 2023-06-02T11:00:44.310.431.000 to 2023-06-02T11:01:46.732.585.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.600304 in 1 packets: 18
      12.579981 in 1 packets: 36
      14.040048 in 1 packets: 54
      21.200376 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.8975218 to 69.8973050
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999510 in 2 packets: 18, 54
      12.9999545 in 1 packets: 36
      20.9999267 in 1 packets: 72
Temperature FPGA: 
      02 -> 52.4 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      01 -> 53.7 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C3 -> 40.4 degC in 90 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 53 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20230602_110035_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.2079543 to 8.2082433
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.6002550  -- OK for SL --
   MIU time from 2023-06-02T11:00:44.310.431.000 to 2023-06-02T11:00:44.310.720.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.600304

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      74 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.8084983 to 22.8087873
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.5799350  -- OK for SL --
   MIU time from 2023-06-02T11:00:58.911.024.000 to 2023-06-02T11:00:58.911.313.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.579981

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.3887223 to 35.3890113
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.0399990  -- OK for SL --
   MIU time from 2023-06-02T11:01:11.491.294.000 to 2023-06-02T11:01:11.491.583.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.040048

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 49.4290103 to 49.4292993
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.2003030  -- OK for SL --
   MIU time from 2023-06-02T11:01:25.531.631.000 to 2023-06-02T11:01:25.531.920.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.200376

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 70.6296023 to 70.6298913
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-02T11:01:46.732.296.000 to 2023-06-02T11:01:46.732.585.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.207954300 corrected by 40bit 10MHz counter of 7.897521800 obtained at 1Hz counter of 6. Result: 6.310433587
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-06-02T11:00:44.310.290.725
   Max 1st sample MIU time among all SL packets: 2023-06-02T11:01:46.732.444.725
   Min 1st sample XO time among all SL packets: 6.310293311
   Max 1st sample XO time among all SL packets: 68.732448589
   TM values from -270 to 196 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-02T11:00:44.310.290.725 (XO: 6.310293311 - 6.310599312, t=40.4 degC)  Rank: 00079
   TM values: Max =   191  Min =  -264  Ave =    -4.43  Std =    23.85
   Min TM value:-264 at an offset of 92.875 usec -> 2023-06-02T11:00:44.310.383.600 (XO: 6.310386186)
   Max TM value:191 at an offset of 92.613 usec -> 2023-06-02T11:00:44.310.383.337 (XO: 6.310385924)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 95.238 usec -> 2023-06-02T11:00:44.310.385.962 (XO: 6.310388549)
   Min dTM/dt [1/us]:-2.38e+03 at an offset of 92.725 usec -> 2023-06-02T11:00:44.310.383.450 (XO: 6.310386036)

 - Block number: 2 starting at 2023-06-02T11:00:58.910.883.725 (XO: 20.910888504 - 20.911194505, t=40.4 degC)  Rank: 00074
   TM values: Max =   181  Min =  -245  Ave =    -4.49  Std =    19.30
   Min TM value:-245 at an offset of 107.113 usec -> 2023-06-02T11:00:58.910.990.837 (XO: 20.910995617)
   Max TM value:181 at an offset of 94.825 usec -> 2023-06-02T11:00:58.910.978.550 (XO: 20.910983330)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 109.488 usec -> 2023-06-02T11:00:58.910.993.212 (XO: 20.910997992)
   Min dTM/dt [1/us]:-2.03e+03 at an offset of 106.963 usec -> 2023-06-02T11:00:58.910.990.687 (XO: 20.910995467)

 - Block number: 3 starting at 2023-06-02T11:01:11.491.153.725 (XO: 33.491156444 - 33.491462445, t=40.4 degC)  Rank: 00081
   TM values: Max =   196  Min =  -254  Ave =    -4.58  Std =    18.11
   Min TM value:-254 at an offset of 116.263 usec -> 2023-06-02T11:01:11.491.269.987 (XO: 33.491272707)
   Max TM value:196 at an offset of 103.925 usec -> 2023-06-02T11:01:11.491.257.650 (XO: 33.491260369)
   Max dTM/dt [1/us]:1.46e+03 at an offset of 103.850 usec -> 2023-06-02T11:01:11.491.257.575 (XO: 33.491260294)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 104.025 usec -> 2023-06-02T11:01:11.491.257.750 (XO: 33.491260469)

 - Block number: 4 starting at 2023-06-02T11:01:25.531.490.725 (XO: 47.531493532 - 47.531799533, t=40.4 degC)  Rank: 00078
   TM values: Max =   169  Min =  -222  Ave =    -4.48  Std =    16.73
   Min TM value:-222 at an offset of 116.550 usec -> 2023-06-02T11:01:25.531.607.275 (XO: 47.531610082)
   Max TM value:169 at an offset of 121.550 usec -> 2023-06-02T11:01:25.531.612.275 (XO: 47.531615082)
   Max dTM/dt [1/us]:1.47e+03 at an offset of 119.038 usec -> 2023-06-02T11:01:25.531.609.762 (XO: 47.531612570)
   Min dTM/dt [1/us]:-1.73e+03 at an offset of 116.388 usec -> 2023-06-02T11:01:25.531.607.112 (XO: 47.531609920)

 - Block number: 5 starting at 2023-06-02T11:01:46.732.155.725 (XO: 68.732159588 - 68.732465589, t=40.4 degC)  Rank: 00078
   TM values: Max =   193  Min =  -270  Ave =    -4.56  Std =    18.42
   Min TM value:-270 at an offset of 93.925 usec -> 2023-06-02T11:01:46.732.249.650 (XO: 68.732253513)
   Max TM value:193 at an offset of 98.863 usec -> 2023-06-02T11:01:46.732.254.587 (XO: 68.732258450)
   Max dTM/dt [1/us]:1.43e+03 at an offset of 96.413 usec -> 2023-06-02T11:01:46.732.252.137 (XO: 68.732256000)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 81.625 usec -> 2023-06-02T11:01:46.732.237.350 (XO: 68.732241213)
