-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat Jun  3 01:22:23 2023 UTC
-- File 20230602_100051_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.685920 to 18.026838
-each separately from 3s to 18s and from 26549us to 686209us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.340340 in 1 packets: 18
MIU packet time from 2023-06-02T10:00:57.685.920.000 to 2023-06-02T10:01:12.026.838.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.340340 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 5.2544966 to 20.2544442
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999476 in 1 packets: 18
Temperature FPGA: 
      01 -> 53.7 degC in 36 packets (all packets)
Temperature XO: 
      C3 -> 40.4 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230602_100051_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.9404185 to 5.9407075
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.3402870  -- OK for SL --
   MIU time from 2023-06-02T10:00:57.685.920.000 to 2023-06-02T10:00:57.686.209.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.340340

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.2809945 to 20.2812835
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-02T10:01:12.026.549.000 to 2023-06-02T10:01:12.026.838.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.940418500 corrected by 40bit 10MHz counter of 5.254496600 obtained at 1Hz counter of 3. Result: 3.685924301
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-06-02T10:00:57.685.779.725
   Max 1st sample MIU time among all SL packets: 2023-06-02T10:01:12.026.697.725
   Min 1st sample XO time among all SL packets: 3.685784025
   Max 1st sample XO time among all SL packets: 18.026699118
   TM values from -242 to 183 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-02T10:00:57.685.779.725 (XO: 3.685784025 - 3.686090026, t=40.4 degC)  Rank: 00077
   TM values: Max =   183  Min =  -242  Ave =    -4.48  Std =    17.78
   Min TM value:-242 at an offset of 111.525 usec -> 2023-06-02T10:00:57.685.891.250 (XO: 3.685895551)
   Max TM value:183 at an offset of 109.000 usec -> 2023-06-02T10:00:57.685.888.725 (XO: 3.685893026)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 111.638 usec -> 2023-06-02T10:00:57.685.891.362 (XO: 3.685895663)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 109.125 usec -> 2023-06-02T10:00:57.685.888.850 (XO: 3.685893151)

 - Block number: 2 starting at 2023-06-02T10:01:12.026.408.725 (XO: 18.026410117 - 18.026716119, t=40.4 degC)  Rank: 00073
   TM values: Max =   172  Min =  -233  Ave =    -4.53  Std =    16.08
   Min TM value:-233 at an offset of 111.488 usec -> 2023-06-02T10:01:12.026.520.212 (XO: 18.026521605)
   Max TM value:172 at an offset of 99.563 usec -> 2023-06-02T10:01:12.026.508.287 (XO: 18.026509680)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 113.875 usec -> 2023-06-02T10:01:12.026.522.600 (XO: 18.026523993)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 111.338 usec -> 2023-06-02T10:01:12.026.520.062 (XO: 18.026521455)
