-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jun  2 22:37:25 2023 UTC
-- File 20230602_035214_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 1.989890 to 72.830414
-each separately from 1s to 72s and from 430041us to 990179us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.459842 in 1 packets: 18
      0.979731 in 1 packets: 36
      26.259787 in 1 packets: 54
      16.139719 in 1 packets: 72
MIU packet time from 2023-06-02T03:52:18.989.890.000 to 2023-06-02T03:53:29.830.414.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.459842 in 1 packets: 18
      0.979731 in 1 packets: 36
      26.259787 in 1 packets: 54
      16.139719 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 2.5961955 to 73.5959443
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      27.9999008 in 1 packets: 18
      0.9999966 in 1 packets: 36
      25.9999080 in 1 packets: 54
      15.9999434 in 1 packets: 72
Temperature FPGA: 
      01 -> 53.7 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      00 -> 55.0 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C3 -> 40.4 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C4 -> 42.0 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 71 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 71 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20230602_035214_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.5860868 to 3.5863758
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 27.4597430  -- OK for SL --
   MIU time from 2023-06-02T03:52:18.989.890.000 to 2023-06-02T03:52:18.990.179.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 27.459842

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 31.0461188 to 31.0464078
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 0.9797270  -- OK for SL --
   MIU time from 2023-06-02T03:52:46.450.021.000 to 2023-06-02T03:52:46.450.310.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 0.979731

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 32.0261348 to 32.0264238
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.2596950  -- OK for SL --
   MIU time from 2023-06-02T03:52:47.430.041.000 to 2023-06-02T03:52:47.430.330.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.259787

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 58.2861188 to 58.2864078
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.1396630  -- OK for SL --
   MIU time from 2023-06-02T03:53:13.690.117.000 to 2023-06-02T03:53:13.690.406.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.139719

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 74.4260708 to 74.4263598
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-02T03:53:29.830.125.000 to 2023-06-02T03:53:29.830.414.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 1, 32bit 10MHz counter = 3.586086800 corrected by 40bit 10MHz counter of 2.596195500 obtained at 1Hz counter of 1. Result: 1.989894864
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-06-02T03:52:18.989.749.725
   Max 1st sample MIU time among all SL packets: 2023-06-02T03:53:29.830.273.725
   Min 1st sample XO time among all SL packets: 1.989754588
   Max 1st sample XO time among all SL packets: 72.830278214
   TM values from -257 to 200 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-02T03:52:18.989.749.725 (XO: 1.989754588 - 1.990060589, t=40.4 degC)  Rank: 00077
   TM values: Max =   182  Min =  -250  Ave =    -4.39  Std =    21.36
   Min TM value:-250 at an offset of 124.713 usec -> 2023-06-02T03:52:18.989.874.437 (XO: 1.989879301)
   Max TM value:182 at an offset of 129.700 usec -> 2023-06-02T03:52:18.989.879.425 (XO: 1.989884289)
   Max dTM/dt [1/us]:1.72e+03 at an offset of 127.300 usec -> 2023-06-02T03:52:18.989.877.025 (XO: 1.989881889)
   Min dTM/dt [1/us]:-1.89e+03 at an offset of 124.588 usec -> 2023-06-02T03:52:18.989.874.312 (XO: 1.989879176)

 - Block number: 2 starting at 2023-06-02T03:52:46.449.880.725 (XO: 29.449883755 - 29.450189756, t=40.4 degC)  Rank: 00081
   TM values: Max =   200  Min =  -257  Ave =    -4.39  Std =    17.85
   Min TM value:-257 at an offset of 115.463 usec -> 2023-06-02T03:52:46.449.996.187 (XO: 29.449999217)
   Max TM value:200 at an offset of 115.188 usec -> 2023-06-02T03:52:46.449.995.912 (XO: 29.449998942)
   Max dTM/dt [1/us]:1.98e+03 at an offset of 117.838 usec -> 2023-06-02T03:52:46.449.998.562 (XO: 29.450001593)
   Min dTM/dt [1/us]:-2.19e+03 at an offset of 115.313 usec -> 2023-06-02T03:52:46.449.996.037 (XO: 29.449999067)

 - Block number: 3 starting at 2023-06-02T03:52:47.429.900.725 (XO: 30.429903173 - 30.430209174, t=42.0 degC)  Rank: 00075
   TM values: Max =   176  Min =  -246  Ave =    -4.46  Std =    19.06
   Min TM value:-246 at an offset of 111.550 usec -> 2023-06-02T03:52:47.430.012.275 (XO: 30.430014723)
   Max TM value:176 at an offset of 99.075 usec -> 2023-06-02T03:52:47.429.999.800 (XO: 30.430002248)
   Max dTM/dt [1/us]:1.59e+03 at an offset of 114.138 usec -> 2023-06-02T03:52:47.430.014.862 (XO: 30.430017311)
   Min dTM/dt [1/us]:-1.80e+03 at an offset of 99.213 usec -> 2023-06-02T03:52:47.429.999.937 (XO: 30.430002386)

 - Block number: 4 starting at 2023-06-02T03:53:13.689.976.725 (XO: 56.689980040 - 56.690286041, t=40.4 degC)  Rank: 00075
   TM values: Max =   169  Min =  -253  Ave =    -4.35  Std =    20.76
   Min TM value:-253 at an offset of 123.925 usec -> 2023-06-02T03:53:13.690.100.650 (XO: 56.690103965)
   Max TM value:169 at an offset of 126.550 usec -> 2023-06-02T03:53:13.690.103.275 (XO: 56.690106590)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 126.400 usec -> 2023-06-02T03:53:13.690.103.125 (XO: 56.690106440)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 123.775 usec -> 2023-06-02T03:53:13.690.100.500 (XO: 56.690103815)

 - Block number: 5 starting at 2023-06-02T03:53:29.829.984.725 (XO: 72.829989213 - 72.830295214, t=40.4 degC)  Rank: 00077
   TM values: Max =   163  Min =  -205  Ave =    -4.37  Std =    18.23
   Min TM value:-205 at an offset of 124.150 usec -> 2023-06-02T03:53:29.830.108.875 (XO: 72.830113363)
   Max TM value:163 at an offset of 126.863 usec -> 2023-06-02T03:53:29.830.111.587 (XO: 72.830116076)
   Max dTM/dt [1/us]:1.63e+03 at an offset of 126.713 usec -> 2023-06-02T03:53:29.830.111.437 (XO: 72.830115926)
   Min dTM/dt [1/us]:-1.53e+03 at an offset of 124.000 usec -> 2023-06-02T03:53:29.830.108.725 (XO: 72.830113213)
