-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jun  2 06:57:54 2023 UTC
-- File 20230602_021332_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 6.902174 to 55.662379
-each separately from 6s to 55s and from 302113us to 902463us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.439641 in 1 packets: 18
      15.959720 in 1 packets: 36
      21.359688 in 1 packets: 54
MIU packet time from 2023-06-02T02:13:41.902.174.000 to 2023-06-02T02:14:30.662.379.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.439641 in 1 packets: 18
      15.959720 in 1 packets: 36
      21.359688 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 7.6791555 to 56.6789832
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999577 in 1 packets: 18
      15.9999438 in 1 packets: 36
      20.9999262 in 1 packets: 54
Temperature FPGA: 
      01 -> 53.7 degC in 72 packets (all packets)
Temperature XO: 
      C3 -> 40.4 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      C4 -> 42.0 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20230602_021332_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 8.5813309 to 8.5816199
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.4395990  -- OK for SL --
   MIU time from 2023-06-02T02:13:41.902.174.000 to 2023-06-02T02:13:41.902.463.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.439641

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 20.0212189 to 20.0215079
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.9596630  -- OK for SL --
   MIU time from 2023-06-02T02:13:53.342.104.000 to 2023-06-02T02:13:53.342.393.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.959720

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.9811709 to 35.9814599
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.3596150  -- OK for SL --
   MIU time from 2023-06-02T02:14:09.302.113.000 to 2023-06-02T02:14:09.302.402.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.359688

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 57.3410749 to 57.3413639
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-02T02:14:30.662.090.000 to 2023-06-02T02:14:30.662.379.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 6, 32bit 10MHz counter = 8.581330900 corrected by 40bit 10MHz counter of 7.679155500 obtained at 1Hz counter of 6. Result: 6.902178648
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-06-02T02:13:41.902.033.725
   Max 1st sample MIU time among all SL packets: 2023-06-02T02:14:30.662.238.725
   Min 1st sample XO time among all SL packets: 6.902038372
   Max 1st sample XO time among all SL packets: 55.662242743
   TM values from -269 to 200 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-02T02:13:41.902.033.725 (XO: 6.902038372 - 6.902344373, t=40.4 degC)  Rank: 00082
   TM values: Max =   200  Min =  -246  Ave =    -4.52  Std =    17.77
   Min TM value:-246 at an offset of 120.475 usec -> 2023-06-02T02:13:41.902.154.200 (XO: 6.902158848)
   Max TM value:200 at an offset of 120.200 usec -> 2023-06-02T02:13:41.902.153.925 (XO: 6.902158573)
   Max dTM/dt [1/us]:1.92e+03 at an offset of 122.863 usec -> 2023-06-02T02:13:41.902.156.587 (XO: 6.902161235)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 120.325 usec -> 2023-06-02T02:13:41.902.154.050 (XO: 6.902158698)

 - Block number: 2 starting at 2023-06-02T02:13:53.341.963.725 (XO: 18.341966622 - 18.342272623, t=42.0 degC)  Rank: 00078
   TM values: Max =   178  Min =  -243  Ave =    -4.44  Std =    23.58
   Min TM value:-243 at an offset of 119.313 usec -> 2023-06-02T02:13:53.342.083.037 (XO: 18.342085935)
   Max TM value:178 at an offset of 119.038 usec -> 2023-06-02T02:13:53.342.082.762 (XO: 18.342085660)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 121.750 usec -> 2023-06-02T02:13:53.342.085.475 (XO: 18.342088372)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 119.200 usec -> 2023-06-02T02:13:53.342.082.925 (XO: 18.342085822)

 - Block number: 3 starting at 2023-06-02T02:14:09.301.972.725 (XO: 34.301974682 - 34.302280683, t=42.0 degC)  Rank: 00081
   TM values: Max =   166  Min =  -238  Ave =    -4.54  Std =    18.34
   Min TM value:-238 at an offset of 119.800 usec -> 2023-06-02T02:14:09.302.092.525 (XO: 34.302094482)
   Max TM value:166 at an offset of 122.450 usec -> 2023-06-02T02:14:09.302.095.175 (XO: 34.302097132)
   Max dTM/dt [1/us]:1.54e+03 at an offset of 122.288 usec -> 2023-06-02T02:14:09.302.095.012 (XO: 34.302096970)
   Min dTM/dt [1/us]:-1.78e+03 at an offset of 119.650 usec -> 2023-06-02T02:14:09.302.092.375 (XO: 34.302094332)

 - Block number: 4 starting at 2023-06-02T02:14:30.661.949.725 (XO: 55.661953742 - 55.662259743, t=42.0 degC)  Rank: 00079
   TM values: Max =   191  Min =  -269  Ave =    -4.51  Std =    19.99
   Min TM value:-269 at an offset of 83.713 usec -> 2023-06-02T02:14:30.662.033.437 (XO: 55.662037455)
   Max TM value:191 at an offset of 71.313 usec -> 2023-06-02T02:14:30.662.021.037 (XO: 55.662025055)
   Max dTM/dt [1/us]:1.46e+03 at an offset of 86.200 usec -> 2023-06-02T02:14:30.662.035.925 (XO: 55.662039942)
   Min dTM/dt [1/us]:-1.98e+03 at an offset of 83.588 usec -> 2023-06-02T02:14:30.662.033.312 (XO: 55.662037330)
