-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jun  2 05:13:59 2023 UTC
-- File 20230601_234248_1227.dat, 126  packets found.
APID:
      1227 (SL) in 126 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 126 packets (all packets)
Total number of packets: 
      18 in 126 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 2.300064 to 93.591441
-each separately from 2s to 93s and from 80582us to 740050us.
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.199491 in 1 packets: 18
      20.169430 in 1 packets: 36
      13.659552 in 1 packets: 54
      15.410068 in 1 packets: 72
      14.340532 in 1 packets: 90
      12.510281 in 1 packets: 108
MIU packet time from 2023-06-01T23:42:53.300.064.000 to 2023-06-01T23:44:24.591.441.000
   Differences in neigboring packets: 
      0.000017 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      15.199491 in 1 packets: 18
      20.169430 in 1 packets: 36
      13.659552 in 1 packets: 54
      15.410068 in 1 packets: 72
      14.340532 in 1 packets: 90
      12.510281 in 1 packets: 108
MIU orbit number: 
      60000 in 126 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 126 packets (all packets)
MIU orbit information / number of events: 
      0 in 126 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 126 packets (all packets)
Instrument anomalies: 
      0 (OK) in 126 packets (all packets)
FPGA configuration version: 
      211 in 126 packets (all packets)
10 MHz counter at the last 1Hz pulse from 3.4672120 to 94.4668916
   Differences in neigboring packets: 
      0.0000000 in 119 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      14.9999471 in 1 packets: 18
      19.9999296 in 1 packets: 36
      13.9999508 in 1 packets: 54
      14.9999472 in 2 packets: 72, 90
      11.9999577 in 1 packets: 108
Temperature FPGA: 
      00 -> 55.0 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      01 -> 53.7 degC in 72 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C4 -> 42.0 degC in 108 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C3 -> 40.4 degC in 18 packets: 91, 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Temperature ant 3: 
      01 -> 108.0 degC in 126 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 126 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 90 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 90 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 17 packets: 110, 111, 112, 113, 114, 115, 116, 117, 118, 119, 120, 121, 122, 123, 124, 125, 126
Survey memory error: 
      0 in 126 packets (all packets)
Event memory error: 
      0 in 126 packets (all packets)
Configuration error: 
      0 in 126 packets (all packets)
SDRAM CRC error: 
      0 in 126 packets (all packets)
Radio gain: 
      0 in 126 packets (all packets)
Radio bandwidth: 
      0 in 126 packets (all packets)
Digital gain: 
      0 in 126 packets (all packets)
Analog gain: 
      0 in 126 packets (all packets)
 
SURVEY DATA READER
-- File 20230601_234248_1227.dat, 7 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 3.7672769 to 3.7675659
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.1994390  -- OK for SL --
   MIU time from 2023-06-01T23:42:53.300.064.000 to 2023-06-01T23:42:53.300.353.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.199491

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.9670049 to 18.9672939
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.1693590  -- OK for SL --
   MIU time from 2023-06-01T23:43:08.499.844.000 to 2023-06-01T23:43:08.500.133.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.169430

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.1366529 to 39.1369419
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.6595030  -- OK for SL --
   MIU time from 2023-06-01T23:43:28.669.563.000 to 2023-06-01T23:43:28.669.852.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.659552

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 52.7964449 to 52.7967339
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.4100150  -- OK for SL --
   MIU time from 2023-06-01T23:43:42.329.404.000 to 2023-06-01T23:43:42.329.693.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.410068

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 68.2067489 to 68.2070379
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.3404790  -- OK for SL --
   MIU time from 2023-06-01T23:43:57.739.761.000 to 2023-06-01T23:43:57.740.050.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.340532

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 82.5475169 to 82.5478059
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.5102390  -- OK for SL --
   MIU time from 2023-06-01T23:44:12.080.582.000 to 2023-06-01T23:44:12.080.871.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.510281

 - Block number: 7 from the sequence of 18 packets.
   Block numbers in the data:
      7 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 95.0580449 to 95.0583339
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-01T23:44:24.591.152.000 to 2023-06-01T23:44:24.591.441.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 126 points based on 104 XO calibration points at 1Hz counter from 0 to 103
   --- First point: 1Hz counter = 2, 32bit 10MHz counter = 3.767276900 corrected by 40bit 10MHz counter of 3.467212000 obtained at 1Hz counter of 2. Result: 2.300065980
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-06-01T23:42:53.299.923.725
   Max 1st sample MIU time among all SL packets: 2023-06-01T23:44:24.591.300.725
   Min 1st sample XO time among all SL packets: 2.299925705
   Max 1st sample XO time among all SL packets: 93.591304095
   TM values from -273 to 235 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-01T23:42:53.299.923.725 (XO: 2.299925705 - 2.300231706, t=42.0 degC)  Rank: 00079
   TM values: Max =   183  Min =  -252  Ave =    -4.60  Std =    20.84
   Min TM value:-252 at an offset of 97.825 usec -> 2023-06-01T23:42:53.300.021.550 (XO: 2.300023530)
   Max TM value:183 at an offset of 97.563 usec -> 2023-06-01T23:42:53.300.021.287 (XO: 2.300023268)
   Max dTM/dt [1/us]:1.68e+03 at an offset of 100.263 usec -> 2023-06-01T23:42:53.300.023.987 (XO: 2.300025968)
   Min dTM/dt [1/us]:-2.12e+03 at an offset of 97.700 usec -> 2023-06-01T23:42:53.300.021.425 (XO: 2.300023405)

 - Block number: 2 starting at 2023-06-01T23:43:08.499.703.725 (XO: 17.499707274 - 17.500013275, t=42.0 degC)  Rank: 00085
   TM values: Max =   175  Min =  -261  Ave =    -4.51  Std =    19.83
   Min TM value:-261 at an offset of 109.825 usec -> 2023-06-01T23:43:08.499.813.550 (XO: 17.499817099)
   Max TM value:175 at an offset of 114.863 usec -> 2023-06-01T23:43:08.499.818.587 (XO: 17.499822137)
   Max dTM/dt [1/us]:1.61e+03 at an offset of 112.288 usec -> 2023-06-01T23:43:08.499.816.012 (XO: 17.499819562)
   Min dTM/dt [1/us]:-1.91e+03 at an offset of 109.688 usec -> 2023-06-01T23:43:08.499.813.412 (XO: 17.499816962)

 - Block number: 3 starting at 2023-06-01T23:43:28.669.422.725 (XO: 37.669426268 - 37.669732269, t=42.0 degC)  Rank: 00080
   TM values: Max =   235  Min =  -258  Ave =    -4.51  Std =    20.78
   Min TM value:-258 at an offset of 126.800 usec -> 2023-06-01T23:43:28.669.549.525 (XO: 37.669553068)
   Max TM value:235 at an offset of 124.363 usec -> 2023-06-01T23:43:28.669.547.087 (XO: 37.669550631)
   Max dTM/dt [1/us]:2.28e+03 at an offset of 126.900 usec -> 2023-06-01T23:43:28.669.549.625 (XO: 37.669553168)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 124.450 usec -> 2023-06-01T23:43:28.669.547.175 (XO: 37.669550718)

 - Block number: 4 starting at 2023-06-01T23:43:42.329.263.725 (XO: 51.329266277 - 51.329572279, t=42.0 degC)  Rank: 00078
   TM values: Max =   196  Min =  -186  Ave =    -4.49  Std =    18.87
   Min TM value:-186 at an offset of 110.388 usec -> 2023-06-01T23:43:42.329.374.112 (XO: 51.329376665)
   Max TM value:196 at an offset of 115.575 usec -> 2023-06-01T23:43:42.329.379.300 (XO: 51.329381853)
   Max dTM/dt [1/us]:1.18e+03 at an offset of 115.488 usec -> 2023-06-01T23:43:42.329.379.212 (XO: 51.329381765)
   Min dTM/dt [1/us]:-1.65e+03 at an offset of 115.713 usec -> 2023-06-01T23:43:42.329.379.437 (XO: 51.329381990)

 - Block number: 5 starting at 2023-06-01T23:43:57.739.620.725 (XO: 66.739624514 - 66.739930515, t=42.0 degC)  Rank: 00078
   TM values: Max =   190  Min =  -273  Ave =    -4.62  Std =    21.20
   Min TM value:-273 at an offset of 97.613 usec -> 2023-06-01T23:43:57.739.718.337 (XO: 66.739722127)
   Max TM value:190 at an offset of 97.350 usec -> 2023-06-01T23:43:57.739.718.075 (XO: 66.739721864)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 99.963 usec -> 2023-06-01T23:43:57.739.720.687 (XO: 66.739724477)
   Min dTM/dt [1/us]:-2.31e+03 at an offset of 97.463 usec -> 2023-06-01T23:43:57.739.718.187 (XO: 66.739721977)

 - Block number: 6 starting at 2023-06-01T23:44:12.080.441.725 (XO: 81.080443015 - 81.080749016, t=40.4 degC)  Rank: 00075
   TM values: Max =   181  Min =  -200  Ave =    -4.46  Std =    19.17
   Min TM value:-200 at an offset of 121.013 usec -> 2023-06-01T23:44:12.080.562.737 (XO: 81.080564028)
   Max TM value:181 at an offset of 125.950 usec -> 2023-06-01T23:44:12.080.567.675 (XO: 81.080568965)
   Max dTM/dt [1/us]:1.35e+03 at an offset of 123.688 usec -> 2023-06-01T23:44:12.080.565.412 (XO: 81.080566703)
   Min dTM/dt [1/us]:-1.62e+03 at an offset of 108.563 usec -> 2023-06-01T23:44:12.080.550.287 (XO: 81.080551577)

 - Block number: 7 starting at 2023-06-01T23:44:24.591.011.725 (XO: 93.591015094 - 93.591321095, t=42.0 degC)  Rank: 00083
   TM values: Max =   192  Min =  -221  Ave =    -4.44  Std =    22.43
   Min TM value:-221 at an offset of 143.038 usec -> 2023-06-01T23:44:24.591.154.763 (XO: 93.591158132)
   Max TM value:192 at an offset of 127.488 usec -> 2023-06-01T23:44:24.591.139.212 (XO: 93.591142581)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 143.176 usec -> 2023-06-01T23:44:24.591.154.900 (XO: 93.591158269)
   Min dTM/dt [1/us]:-1.75e+03 at an offset of 127.588 usec -> 2023-06-01T23:44:24.591.139.312 (XO: 93.591142681)
