-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jun  2 04:37:39 2023 UTC
-- File 20230601_223933_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.910157 to 77.006428
-each separately from 5s to 77s and from 6139us to 987440us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.667901 in 1 packets: 18
      5.929412 in 1 packets: 36
      18.478814 in 1 packets: 54
      17.618991 in 1 packets: 72
      7.399419 in 1 packets: 90
MIU packet time from 2023-06-01T22:39:41.910.157.000 to 2023-06-01T22:40:53.006.428.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.667901 in 1 packets: 18
      5.929412 in 1 packets: 36
      18.478814 in 1 packets: 54
      17.618991 in 1 packets: 72
      7.399419 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.4696797 to 78.4694264
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.9999226 in 1 packets: 18
      5.9999789 in 1 packets: 36
      17.9999367 in 2 packets: 54, 72
      7.9999718 in 1 packets: 90
Temperature FPGA: 
      00 -> 55.0 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      01 -> 53.7 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C4 -> 42.0 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 53 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 36 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 53 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20230601_223933_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.3798386 to 7.3801276
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.6678230  -- OK for SL --
   MIU time from 2023-06-01T22:39:41.910.157.000 to 2023-06-01T22:39:41.910.446.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.667901

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      83 in 18 packets (all packets)  -- OK --
   10 MHz counter from 29.0479506 to 29.0482396
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.9293910  -- OK for SL --
   MIU time from 2023-06-01T22:40:03.578.347.000 to 2023-06-01T22:40:03.578.636.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.929412

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      70 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.9776306 to 34.9779196
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.4787510  -- OK for SL --
   MIU time from 2023-06-01T22:40:09.508.048.000 to 2023-06-01T22:40:09.508.337.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.478814

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 53.4566706 to 53.4569596
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.6189270  -- OK for SL --
   MIU time from 2023-06-01T22:40:27.987.151.000 to 2023-06-01T22:40:27.987.440.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.618991

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 71.0758866 to 71.0761756
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.3993910  -- OK for SL --
   MIU time from 2023-06-01T22:40:45.606.431.000 to 2023-06-01T22:40:45.606.720.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.399419

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 78.4755666 to 78.4758556
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-01T22:40:53.006.139.000 to 2023-06-01T22:40:53.006.428.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.379838600 corrected by 40bit 10MHz counter of 6.469679700 obtained at 1Hz counter of 5. Result: 5.910162086
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-06-01T22:39:41.910.016.725
   Max 1st sample MIU time among all SL packets: 2023-06-01T22:40:53.006.287.725
   Min 1st sample XO time among all SL packets: 5.910021810
   Max 1st sample XO time among all SL packets: 77.006288947
   TM values from -276 to 213 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-01T22:39:41.910.016.725 (XO: 5.910021810 - 5.910327811, t=42.0 degC)  Rank: 00077
   TM values: Max =   185  Min =  -239  Ave =    -4.46  Std =    21.17
   Min TM value:-239 at an offset of 123.513 usec -> 2023-06-01T22:39:41.910.140.237 (XO: 5.910145323)
   Max TM value:185 at an offset of 128.513 usec -> 2023-06-01T22:39:41.910.145.237 (XO: 5.910150323)
   Max dTM/dt [1/us]:1.52e+03 at an offset of 126.113 usec -> 2023-06-01T22:39:41.910.142.837 (XO: 5.910147923)
   Min dTM/dt [1/us]:-1.46e+03 at an offset of 123.388 usec -> 2023-06-01T22:39:41.910.140.112 (XO: 5.910145198)

 - Block number: 2 starting at 2023-06-01T22:40:03.578.206.725 (XO: 27.578210049 - 27.578516050, t=42.0 degC)  Rank: 00083
   TM values: Max =   213  Min =  -250  Ave =    -4.47  Std =    21.23
   Min TM value:-250 at an offset of 114.875 usec -> 2023-06-01T22:40:03.578.321.600 (XO: 27.578324924)
   Max TM value:213 at an offset of 112.400 usec -> 2023-06-01T22:40:03.578.319.125 (XO: 27.578322449)
   Max dTM/dt [1/us]:2.18e+03 at an offset of 114.988 usec -> 2023-06-01T22:40:03.578.321.712 (XO: 27.578325037)
   Min dTM/dt [1/us]:-2.21e+03 at an offset of 112.513 usec -> 2023-06-01T22:40:03.578.319.237 (XO: 27.578322562)

 - Block number: 3 starting at 2023-06-01T22:40:09.507.907.725 (XO: 33.507910903 - 33.508216904, t=42.0 degC)  Rank: 00070
   TM values: Max =   212  Min =  -247  Ave =    -4.52  Std =    22.49
   Min TM value:-247 at an offset of 87.850 usec -> 2023-06-01T22:40:09.507.995.575 (XO: 33.507998753)
   Max TM value:212 at an offset of 61.125 usec -> 2023-06-01T22:40:09.507.968.850 (XO: 33.507972028)
   Max dTM/dt [1/us]:1.81e+03 at an offset of 63.750 usec -> 2023-06-01T22:40:09.507.971.475 (XO: 33.507974653)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 61.250 usec -> 2023-06-01T22:40:09.507.968.975 (XO: 33.507972153)

 - Block number: 4 starting at 2023-06-01T22:40:27.987.010.725 (XO: 51.987015880 - 51.987321881, t=42.0 degC)  Rank: 00082
   TM values: Max =   210  Min =  -276  Ave =    -4.49  Std =    20.48
   Min TM value:-276 at an offset of 125.250 usec -> 2023-06-01T22:40:27.987.135.975 (XO: 51.987141130)
   Max TM value:210 at an offset of 112.763 usec -> 2023-06-01T22:40:27.987.123.487 (XO: 51.987128642)
   Max dTM/dt [1/us]:2.08e+03 at an offset of 127.638 usec -> 2023-06-01T22:40:27.987.138.362 (XO: 51.987143518)
   Min dTM/dt [1/us]:-2.51e+03 at an offset of 125.113 usec -> 2023-06-01T22:40:27.987.135.837 (XO: 51.987140992)

 - Block number: 5 starting at 2023-06-01T22:40:45.606.290.725 (XO: 69.606293847 - 69.606599848, t=42.0 degC)  Rank: 00076
   TM values: Max =   211  Min =  -269  Ave =    -4.53  Std =    19.74
   Min TM value:-269 at an offset of 101.550 usec -> 2023-06-01T22:40:45.606.392.275 (XO: 69.606395397)
   Max TM value:211 at an offset of 106.513 usec -> 2023-06-01T22:40:45.606.397.237 (XO: 69.606400360)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 104.050 usec -> 2023-06-01T22:40:45.606.394.775 (XO: 69.606397897)
   Min dTM/dt [1/us]:-1.99e+03 at an offset of 89.088 usec -> 2023-06-01T22:40:45.606.379.812 (XO: 69.606382935)

 - Block number: 6 starting at 2023-06-01T22:40:53.005.998.725 (XO: 77.005999946 - 77.006305947, t=42.0 degC)  Rank: 00073
   TM values: Max =   200  Min =  -261  Ave =    -4.44  Std =    24.48
   Min TM value:-261 at an offset of 101.763 usec -> 2023-06-01T22:40:53.006.100.487 (XO: 77.006101709)
   Max TM value:200 at an offset of 144.563 usec -> 2023-06-01T22:40:53.006.143.288 (XO: 77.006144509)
   Max dTM/dt [1/us]:1.51e+03 at an offset of 104.175 usec -> 2023-06-01T22:40:53.006.102.900 (XO: 77.006104121)
   Min dTM/dt [1/us]:-2.14e+03 at an offset of 101.625 usec -> 2023-06-01T22:40:53.006.100.350 (XO: 77.006101571)
