-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jun  2 01:18:56 2023 UTC
-- File 20230601_123809_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.453102 to 50.141067
-each separately from 5s to 50s and from 140778us to 738076us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.242618 in 1 packets: 18
      10.041489 in 1 packets: 36
      17.402702 in 1 packets: 54
MIU packet time from 2023-06-01T12:38:17.453.102.000 to 2023-06-01T12:39:02.141.067.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.242618 in 1 packets: 18
      10.041489 in 1 packets: 36
      17.402702 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.7550464 to 51.7548902
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999410 in 1 packets: 18
      9.9999653 in 1 packets: 36
      17.9999375 in 1 packets: 54
Temperature FPGA: 
      01 -> 53.7 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      00 -> 55.0 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C3 -> 40.4 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C4 -> 42.0 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 54 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 17 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20230601_123809_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      84 in 18 packets (all packets)  -- OK --
   10 MHz counter from 7.2081496 to 7.2084386
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.2425590  -- OK for SL --
   MIU time from 2023-06-01T12:38:17.453.102.000 to 2023-06-01T12:38:17.453.391.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.242618

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.4509976 to 24.4512866
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.0414550  -- OK for SL --
   MIU time from 2023-06-01T12:38:34.696.009.000 to 2023-06-01T12:38:34.696.298.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.041489

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.4927416 to 34.4930306
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.4026390  -- OK for SL --
   MIU time from 2023-06-01T12:38:44.737.787.000 to 2023-06-01T12:38:44.738.076.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.402702

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 51.8956696 to 51.8959586
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-01T12:39:02.140.778.000 to 2023-06-01T12:39:02.141.067.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 7.208149600 corrected by 40bit 10MHz counter of 6.755046400 obtained at 1Hz counter of 5. Result: 5.453104786
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-06-01T12:38:17.452.961.725
   Max 1st sample MIU time among all SL packets: 2023-06-01T12:39:02.140.926.725
   Min 1st sample XO time among all SL packets: 5.452964510
   Max 1st sample XO time among all SL packets: 50.140928618
   TM values from -255 to 204 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-01T12:38:17.452.961.725 (XO: 5.452964510 - 5.453270511, t=40.4 degC)  Rank: 00084
   TM values: Max =   204  Min =  -246  Ave =    -4.36  Std =    18.42
   Min TM value:-246 at an offset of 117.075 usec -> 2023-06-01T12:38:17.453.078.800 (XO: 5.453081586)
   Max TM value:204 at an offset of 104.300 usec -> 2023-06-01T12:38:17.453.066.025 (XO: 5.453068811)
   Max dTM/dt [1/us]:1.60e+03 at an offset of 119.625 usec -> 2023-06-01T12:38:17.453.081.350 (XO: 5.453084136)
   Min dTM/dt [1/us]:-2.05e+03 at an offset of 104.413 usec -> 2023-06-01T12:38:17.453.066.137 (XO: 5.453068923)

 - Block number: 2 starting at 2023-06-01T12:38:34.695.868.725 (XO: 22.695872361 - 22.696178362, t=42.0 degC)  Rank: 00079
   TM values: Max =   182  Min =  -246  Ave =    -4.43  Std =    16.90
   Min TM value:-246 at an offset of 108.138 usec -> 2023-06-01T12:38:34.695.976.862 (XO: 22.695980498)
   Max TM value:182 at an offset of 105.575 usec -> 2023-06-01T12:38:34.695.974.300 (XO: 22.695977936)
   Max dTM/dt [1/us]:1.91e+03 at an offset of 108.238 usec -> 2023-06-01T12:38:34.695.976.962 (XO: 22.695980598)
   Min dTM/dt [1/us]:-2.00e+03 at an offset of 105.713 usec -> 2023-06-01T12:38:34.695.974.437 (XO: 22.695978073)

 - Block number: 3 starting at 2023-06-01T12:38:44.737.646.725 (XO: 32.737651207 - 32.737957208, t=40.4 degC)  Rank: 00077
   TM values: Max =   198  Min =  -228  Ave =    -4.45  Std =    18.39
   Min TM value:-228 at an offset of 104.025 usec -> 2023-06-01T12:38:44.737.750.750 (XO: 32.737755232)
   Max TM value:198 at an offset of 109.025 usec -> 2023-06-01T12:38:44.737.755.750 (XO: 32.737760232)
   Max dTM/dt [1/us]:1.43e+03 at an offset of 106.600 usec -> 2023-06-01T12:38:44.737.753.325 (XO: 32.737757807)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 91.638 usec -> 2023-06-01T12:38:44.737.738.362 (XO: 32.737742845)

 - Block number: 4 starting at 2023-06-01T12:39:02.140.637.725 (XO: 50.140639617 - 50.140945618, t=40.4 degC)  Rank: 00081
   TM values: Max =   196  Min =  -255  Ave =    -4.30  Std =    22.53
   Min TM value:-255 at an offset of 116.950 usec -> 2023-06-01T12:39:02.140.754.675 (XO: 50.140756568)
   Max TM value:196 at an offset of 116.700 usec -> 2023-06-01T12:39:02.140.754.425 (XO: 50.140756318)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 119.350 usec -> 2023-06-01T12:39:02.140.757.075 (XO: 50.140758968)
   Min dTM/dt [1/us]:-2.35e+03 at an offset of 116.813 usec -> 2023-06-01T12:39:02.140.754.537 (XO: 50.140756430)
