-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jun  2 00:17:40 2023 UTC
-- File 20230601_100412_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.387403 to 81.476441
-each separately from 8s to 81s and from 150157us to 778710us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.953829 in 1 packets: 18
      15.436611 in 1 packets: 36
      14.780451 in 1 packets: 54
      14.590707 in 1 packets: 72
      17.325706 in 1 packets: 90
MIU packet time from 2023-06-01T10:04:23.387.403.000 to 2023-06-01T10:05:36.476.441.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.953829 in 1 packets: 18
      15.436611 in 1 packets: 36
      14.780451 in 1 packets: 54
      14.590707 in 1 packets: 72
      17.325706 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.8335188 to 82.8332653
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      10.9999619 in 1 packets: 18
      14.9999479 in 1 packets: 36
      14.9999481 in 1 packets: 54
      14.9999478 in 1 packets: 72
      16.9999408 in 1 packets: 90
Temperature FPGA: 
      01 -> 53.7 degC in 72 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      02 -> 52.4 degC in 36 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90, ...
Temperature XO: 
      C3 -> 40.4 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 71 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
      2 in 18 packets: 91, 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 5 packets: 20, 21, 22, 23, 24
      2 in 12 packets: 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      3 in 54 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
      4 in 1 packets: 91
      5 in 17 packets: 92, 93, 94, 95, 96, 97, 98, 99, 100, 101, 102, 103, 104, 105, 106, 107, 108
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20230601_100412_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      77 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.2209227 to 10.2212117
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 10.9537910  -- OK for SL --
   MIU time from 2023-06-01T10:04:23.387.403.000 to 2023-06-01T10:04:23.387.692.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 10.953829

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      221 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.1750027 to 21.1752917
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 15.4365590  -- OK for SL --
   MIU time from 2023-06-01T10:04:34.341.521.000 to 2023-06-01T10:04:34.341.810.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 15.436611

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      73 in 18 packets (all packets)  -- OK --
   10 MHz counter from 36.6118507 to 36.6121397
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.7803990  -- OK for SL --
   MIU time from 2023-06-01T10:04:49.778.421.000 to 2023-06-01T10:04:49.778.710.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.780451

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      71 in 18 packets (all packets)  -- OK --
   10 MHz counter from 51.3925387 to 51.3928277
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.5906550  -- OK for SL --
   MIU time from 2023-06-01T10:05:04.559.161.000 to 2023-06-01T10:05:04.559.450.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.590707

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 65.9834827 to 65.9837717
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.3256470  -- OK for SL --
   MIU time from 2023-06-01T10:05:19.150.157.000 to 2023-06-01T10:05:19.150.446.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.325706

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      130 in 18 packets (all packets)  -- OK --
   10 MHz counter from 83.3094187 to 83.3097077
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-06-01T10:05:36.476.152.000 to 2023-06-01T10:05:36.476.441.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.220922700 corrected by 40bit 10MHz counter of 9.833518800 obtained at 1Hz counter of 8. Result: 8.387405217
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-06-01T10:04:23.387.262.725
   Max 1st sample MIU time among all SL packets: 2023-06-01T10:05:36.476.300.725
   Min 1st sample XO time among all SL packets: 8.387264942
   Max 1st sample XO time among all SL packets: 81.476303792
   TM values from -421 to 650 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-06-01T10:04:23.387.262.725 (XO: 8.387264942 - 8.387570943, t=40.4 degC)  Rank: 00077
   TM values: Max =   183  Min =  -243  Ave =    -4.46  Std =    20.41
   Min TM value:-243 at an offset of 117.600 usec -> 2023-06-01T10:04:23.387.380.325 (XO: 8.387382542)
   Max TM value:183 at an offset of 115.050 usec -> 2023-06-01T10:04:23.387.377.775 (XO: 8.387379992)
   Max dTM/dt [1/us]:1.93e+03 at an offset of 117.713 usec -> 2023-06-01T10:04:23.387.380.437 (XO: 8.387382655)
   Min dTM/dt [1/us]:-2.03e+03 at an offset of 115.188 usec -> 2023-06-01T10:04:23.387.377.912 (XO: 8.387380130)

 - Block number: 2 starting at 2023-06-01T10:04:34.341.380.725 (XO: 19.341382920 - 19.341688921, t=40.4 degC)  Rank: 00221
   TM values: Max =   650  Min =  -421  Ave =    -4.54  Std =    61.97
   Min TM value:-421 at an offset of 174.976 usec -> 2023-06-01T10:04:34.341.555.700 (XO: 19.341557895)
   Max TM value:650 at an offset of 121.850 usec -> 2023-06-01T10:04:34.341.502.575 (XO: 19.341504770)
   Max dTM/dt [1/us]:2.57e+03 at an offset of 148.076 usec -> 2023-06-01T10:04:34.341.528.800 (XO: 19.341530995)
   Min dTM/dt [1/us]:-3.13e+03 at an offset of 174.801 usec -> 2023-06-01T10:04:34.341.555.525 (XO: 19.341557720)

 - Block number: 3 starting at 2023-06-01T10:04:49.778.280.725 (XO: 34.778284549 - 34.778590550, t=40.4 degC)  Rank: 00073
   TM values: Max =   197  Min =  -259  Ave =    -4.51  Std =    25.93
   Min TM value:-259 at an offset of 117.775 usec -> 2023-06-01T10:04:49.778.398.500 (XO: 34.778402324)
   Max TM value:197 at an offset of 113.538 usec -> 2023-06-01T10:04:49.778.394.262 (XO: 34.778398087)
   Max dTM/dt [1/us]:2.20e+03 at an offset of 117.888 usec -> 2023-06-01T10:04:49.778.398.612 (XO: 34.778402437)
   Min dTM/dt [1/us]:-2.27e+03 at an offset of 115.450 usec -> 2023-06-01T10:04:49.778.396.175 (XO: 34.778399999)

 - Block number: 4 starting at 2023-06-01T10:05:04.559.020.725 (XO: 49.559023682 - 49.559329683, t=40.4 degC)  Rank: 00071
   TM values: Max =   186  Min =  -225  Ave =    -4.47  Std =    17.29
   Min TM value:-225 at an offset of 116.638 usec -> 2023-06-01T10:05:04.559.137.362 (XO: 49.559140319)
   Max TM value:186 at an offset of 114.125 usec -> 2023-06-01T10:05:04.559.134.850 (XO: 49.559137807)
   Max dTM/dt [1/us]:1.87e+03 at an offset of 116.738 usec -> 2023-06-01T10:05:04.559.137.462 (XO: 49.559140419)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 114.238 usec -> 2023-06-01T10:05:04.559.134.962 (XO: 49.559137919)

 - Block number: 5 starting at 2023-06-01T10:05:19.150.016.725 (XO: 64.150018450 - 64.150324451, t=40.4 degC)  Rank: 00080
   TM values: Max =   190  Min =  -217  Ave =    -4.46  Std =    17.28
   Min TM value:-217 at an offset of 112.800 usec -> 2023-06-01T10:05:19.150.129.525 (XO: 64.150131250)
   Max TM value:190 at an offset of 112.550 usec -> 2023-06-01T10:05:19.150.129.275 (XO: 64.150131000)
   Max dTM/dt [1/us]:1.44e+03 at an offset of 127.488 usec -> 2023-06-01T10:05:19.150.144.212 (XO: 64.150145938)
   Min dTM/dt [1/us]:-1.92e+03 at an offset of 112.650 usec -> 2023-06-01T10:05:19.150.129.375 (XO: 64.150131100)

 - Block number: 6 starting at 2023-06-01T10:05:36.476.011.725 (XO: 81.476014791 - 81.476320792, t=40.4 degC)  Rank: 00130
   TM values: Max =   226  Min =  -178  Ave =    -3.57  Std =    31.54
   Min TM value:-178 at an offset of 116.913 usec -> 2023-06-01T10:05:36.476.128.637 (XO: 81.476131704)
   Max TM value:226 at an offset of 113.700 usec -> 2023-06-01T10:05:36.476.125.425 (XO: 81.476128491)
   Max dTM/dt [1/us]:1.11e+03 at an offset of 305.651 usec -> 2023-06-01T10:05:36.476.317.376 (XO: 81.476320442)
   Min dTM/dt [1/us]:-7.17e+02 at an offset of 113.838 usec -> 2023-06-01T10:05:36.476.125.562 (XO: 81.476128629)
