-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri Jun  2 00:08:38 2023 UTC
-- File 20230601_095627_1229.dat, 36720  packets found.
APID:
      1229 (EW) in 36720 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36720 packets (all packets)
Total number of packets: 
      36720 in 36720 packets (all packets)
Packet No from 1 to 36720
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 60.097345 to 85.944438
-each separately from 60s to 85s and from 97345us to 974776us.
   Differences in neigboring packets: 
      0.000000 in 36715 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      4.969662 in 1 packets: 9792
      -25.847093 in 1 packets: 13260
      25.847093 in 1 packets: 25500
      -4.969662 in 1 packets: 34272
MIU packet time from 2023-06-01T09:57:30.097.345.000 to 2023-06-01T09:57:55.944.438.000
   Differences in neigboring packets: 
      0.000000 in 36715 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      4.969662 in 1 packets: 9792
      -25.847093 in 1 packets: 13260
      25.847093 in 1 packets: 25500
      -4.969662 in 1 packets: 34272
MIU orbit number: 
      60000 in 36720 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36720 packets (all packets)
MIU orbit information / number of events: 
      0 in 36720 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36720 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36720 packets (all packets)
FPGA configuration version: 
      211 in 36720 packets (all packets)
10 MHz counter at the last 1Hz pulse from 61.6047768 to 86.6046884
   Differences in neigboring packets: 
      0.0000000 in 36715 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      4.9999823 in 1 packets: 9792
      -24.9999116 in 1 packets: 13260
      24.9999116 in 1 packets: 25500
      -4.9999823 in 1 packets: 34272
Temperature FPGA: 
      01 -> 53.7 degC in 24480 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      02 -> 52.4 degC in 12240 packets: 13261, 13262, 13263, 13264, 13265, 13266, 13267, 13268, 13269, 13270, 13271, 13272, 13273, 13274, 13275, 13276, 13277, 13278, ...
Temperature XO: 
      C3 -> 40.4 degC in 36720 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36720 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36720 packets (all packets)
Number of external events: 
      2 in 12240 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      3 in 12240 packets: 9793, 9794, 9795, 9796, 9797, 9798, 9799, 9800, 9801, 9802, 9803, 9804, 9805, 9806, 9807, 9808, 9809, 9810, ...
      1 in 12240 packets: 13261, 13262, 13263, 13264, 13265, 13266, 13267, 13268, 13269, 13270, 13271, 13272, 13273, 13274, 13275, 13276, 13277, 13278, ...
Number of generated alerts: 
      5 in 12240 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      6 in 12240 packets: 9793, 9794, 9795, 9796, 9797, 9798, 9799, 9800, 9801, 9802, 9803, 9804, 9805, 9806, 9807, 9808, 9809, 9810, ...
      1 in 12240 packets: 13261, 13262, 13263, 13264, 13265, 13266, 13267, 13268, 13269, 13270, 13271, 13272, 13273, 13274, 13275, 13276, 13277, 13278, ...
Survey memory error: 
      0 in 36720 packets (all packets)
Event memory error: 
      0 in 36720 packets (all packets)
Configuration error: 
      0 in 36720 packets (all packets)
SDRAM CRC error: 
      0 in 36720 packets (all packets)
Radio gain: 
      0 in 36720 packets (all packets)
Radio bandwidth: 
      0 in 36720 packets (all packets)
Digital gain: 
      0 in 36720 packets (all packets)
Analog gain: 
      0 in 36720 packets (all packets)

EVENT DATA READER
-- File 20230601_095627_1229.dat, 180 EW pages found. Extra packets: 0.
   Design version: 211 => Data interpreted as:   Pre-final (EM4-FM1) data structure with variable 32bit C10MHz in the event data descriptor
--
Data structure of all event data packets in the sequence read from file:
   Event numbers from 1 to 3
      2 in 12240 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      3 in 12240 packets: 9793, 9794, 9795, 9796, 9797, 9798, 9799, 9800, 9801, 9802, 9803, 9804, 9805, 9806, 9807, 9808, 9809, 9810, ...
      1 in 12240 packets: 13261, 13262, 13263, 13264, 13265, 13266, 13267, 13268, 13269, 13270, 13271, 13272, 13273, 13274, 13275, 13276, 13277, 13278, ...
   LS 32 bits of C10MHz from descriptor (period 430s)  from 61.6466354 to 87.7017024
      Difference between the neighboring packets: 
      0.0000170 in 36715 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      4.9107050 in 1 packets: 9792
      -26.0550670 in 1 packets: 13260
      25.6389410 in 1 packets: 25500
      -5.1187510 in 1 packets: 34272
   Memory page number from 0 to 237
      Difference between the neighboring packets: 
      0 in 36540 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 177 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
      16 in 1 packets: 13260
      44 in 1 packets: 25500
   Event page number from 1 to 60
      Difference between the neighboring packets: 
      0 in 36540 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 175 packets: 204, 408, 612, 816, 1020, 1224, 1428, 1632, 1836, 2040, 2244, 2448, 2652, 2856, 3060, 3264, 3468, 3672, ...
      -16 in 1 packets: 9792
      -59 in 2 packets: 13260, 25500
      -42 in 1 packets: 34272
   MIU event flag 
      1 in 36720 packets (all packets)
   Time triggered event flag 
      0 in 36720 packets (all packets)
   Absolute time of the 1st EW sample = header abs. time + offset - one page (3.468 ms) - 175 ns delay
   Min EW event (1st sample) abs. time found among all events: 2023-06-01T09:57:30.041.856.626
   Min EW header abs. time found among all events            : 2023-06-01T09:57:30.097.345.000
   Max EW event (1st sample) abs. time found among all events: 2023-06-01T09:57:55.888.949.626
   Max EW header abs. time found among all events            : 2023-06-01T09:57:55.944.438.000
   TM values from -308 to 310 median:-4.00000
--
Data structure of separate events (selected and re-ordered from the packet sequence):

 Event no. 1 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      80 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      81 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      82 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      83 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      84 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      85 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      86 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      87 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      88 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      89 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      90 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      91 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      92 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      93 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      94 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      95 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      96 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      97 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      98 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      99 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      100 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      101 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      102 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      103 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      104 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      105 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      106 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      107 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      108 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      109 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      110 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      111 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      112 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      113 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      114 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      115 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      116 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      117 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      118 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      119 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      120 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      121 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      122 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      123 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      124 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      125 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      126 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      127 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      128 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      129 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      130 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      131 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      132 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      133 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      134 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      135 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      136 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      137 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      138 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      139 in 204 packets (all packets)  --OK-- 
   Event no. 1 pages: 60  ordered packets: 12240
   TM values from -238 to 194 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=60 C1MHz=0.097345
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 12239 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 61.6466354
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2023-06-01T09:57:30.097.345.000
   Event 1st sample abs. time : 2023-06-01T09:57:30.041.856.626)
   --- Rel. time correction for 1 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 60, 32bit 10MHz counter = 61.646635400 corrected by 40bit 10MHz counter of 61.604776800 obtained at 1Hz counter of 60. Result: 60.041858751
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              60.041858576
   Event data descriptor XO time (reconstructed seconds from XP start)    :              60.041858751
   Sampling frequency correction factor for this event:     0.999996400

 Event no. 2 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      226 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      227 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      228 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      229 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      230 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      231 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      232 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      233 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      234 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      235 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      236 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      237 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      0 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      1 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      2 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      3 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      4 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      5 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      6 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      7 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      8 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      9 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      10 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      11 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      12 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      13 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      14 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      15 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      16 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      17 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      18 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      19 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      20 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      21 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      22 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      23 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      24 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      25 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      26 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      27 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      28 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      29 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      30 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      31 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      32 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      33 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      34 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      35 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      36 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      37 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      38 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      39 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      40 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      41 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      42 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      43 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      44 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      45 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      46 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      47 in 204 packets (all packets)  --OK-- 
   Event no. 2 pages: 60  ordered packets: 12240
   TM values from -308 to 310 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=80 C1MHz=0.974776
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 12239 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 82.5239954
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2023-06-01T09:57:50.974.776.000
   Event 1st sample abs. time : 2023-06-01T09:57:50.919.287.626)
   --- Rel. time correction for 1 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 80, 32bit 10MHz counter = 82.523995400 corrected by 40bit 10MHz counter of 74.604730900 obtained at 1Hz counter of 73. Result: 80.919293009
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              80.919292834
   Event data descriptor XO time (reconstructed seconds from XP start)    :              80.919293009
   Sampling frequency correction factor for this event:     0.999996400

 Event no. 3 total number of packets: 12240
    Page no. 1:  204 packets  -- OK;  memory page(s):
      183 in 204 packets (all packets)  --OK-- 
    Page no. 2:  204 packets  -- OK;  memory page(s):
      184 in 204 packets (all packets)  --OK-- 
    Page no. 3:  204 packets  -- OK;  memory page(s):
      185 in 204 packets (all packets)  --OK-- 
    Page no. 4:  204 packets  -- OK;  memory page(s):
      186 in 204 packets (all packets)  --OK-- 
    Page no. 5:  204 packets  -- OK;  memory page(s):
      187 in 204 packets (all packets)  --OK-- 
    Page no. 6:  204 packets  -- OK;  memory page(s):
      188 in 204 packets (all packets)  --OK-- 
    Page no. 7:  204 packets  -- OK;  memory page(s):
      189 in 204 packets (all packets)  --OK-- 
    Page no. 8:  204 packets  -- OK;  memory page(s):
      190 in 204 packets (all packets)  --OK-- 
    Page no. 9:  204 packets  -- OK;  memory page(s):
      191 in 204 packets (all packets)  --OK-- 
    Page no. 10:  204 packets  -- OK;  memory page(s):
      192 in 204 packets (all packets)  --OK-- 
    Page no. 11:  204 packets  -- OK;  memory page(s):
      193 in 204 packets (all packets)  --OK-- 
    Page no. 12:  204 packets  -- OK;  memory page(s):
      194 in 204 packets (all packets)  --OK-- 
    Page no. 13:  204 packets  -- OK;  memory page(s):
      195 in 204 packets (all packets)  --OK-- 
    Page no. 14:  204 packets  -- OK;  memory page(s):
      196 in 204 packets (all packets)  --OK-- 
    Page no. 15:  204 packets  -- OK;  memory page(s):
      197 in 204 packets (all packets)  --OK-- 
    Page no. 16:  204 packets  -- OK;  memory page(s):
      198 in 204 packets (all packets)  --OK-- 
    Page no. 17:  204 packets  -- OK;  memory page(s):
      199 in 204 packets (all packets)  --OK-- 
    Page no. 18:  204 packets  -- OK;  memory page(s):
      200 in 204 packets (all packets)  --OK-- 
    Page no. 19:  204 packets  -- OK;  memory page(s):
      201 in 204 packets (all packets)  --OK-- 
    Page no. 20:  204 packets  -- OK;  memory page(s):
      202 in 204 packets (all packets)  --OK-- 
    Page no. 21:  204 packets  -- OK;  memory page(s):
      203 in 204 packets (all packets)  --OK-- 
    Page no. 22:  204 packets  -- OK;  memory page(s):
      204 in 204 packets (all packets)  --OK-- 
    Page no. 23:  204 packets  -- OK;  memory page(s):
      205 in 204 packets (all packets)  --OK-- 
    Page no. 24:  204 packets  -- OK;  memory page(s):
      206 in 204 packets (all packets)  --OK-- 
    Page no. 25:  204 packets  -- OK;  memory page(s):
      207 in 204 packets (all packets)  --OK-- 
    Page no. 26:  204 packets  -- OK;  memory page(s):
      208 in 204 packets (all packets)  --OK-- 
    Page no. 27:  204 packets  -- OK;  memory page(s):
      209 in 204 packets (all packets)  --OK-- 
    Page no. 28:  204 packets  -- OK;  memory page(s):
      210 in 204 packets (all packets)  --OK-- 
    Page no. 29:  204 packets  -- OK;  memory page(s):
      211 in 204 packets (all packets)  --OK-- 
    Page no. 30:  204 packets  -- OK;  memory page(s):
      212 in 204 packets (all packets)  --OK-- 
    Page no. 31:  204 packets  -- OK;  memory page(s):
      213 in 204 packets (all packets)  --OK-- 
    Page no. 32:  204 packets  -- OK;  memory page(s):
      214 in 204 packets (all packets)  --OK-- 
    Page no. 33:  204 packets  -- OK;  memory page(s):
      215 in 204 packets (all packets)  --OK-- 
    Page no. 34:  204 packets  -- OK;  memory page(s):
      216 in 204 packets (all packets)  --OK-- 
    Page no. 35:  204 packets  -- OK;  memory page(s):
      217 in 204 packets (all packets)  --OK-- 
    Page no. 36:  204 packets  -- OK;  memory page(s):
      218 in 204 packets (all packets)  --OK-- 
    Page no. 37:  204 packets  -- OK;  memory page(s):
      219 in 204 packets (all packets)  --OK-- 
    Page no. 38:  204 packets  -- OK;  memory page(s):
      220 in 204 packets (all packets)  --OK-- 
    Page no. 39:  204 packets  -- OK;  memory page(s):
      221 in 204 packets (all packets)  --OK-- 
    Page no. 40:  204 packets  -- OK;  memory page(s):
      222 in 204 packets (all packets)  --OK-- 
    Page no. 41:  204 packets  -- OK;  memory page(s):
      223 in 204 packets (all packets)  --OK-- 
    Page no. 42:  204 packets  -- OK;  memory page(s):
      224 in 204 packets (all packets)  --OK-- 
    Page no. 43:  204 packets  -- OK;  memory page(s):
      225 in 204 packets (all packets)  --OK-- 
    Page no. 44:  204 packets  -- OK;  memory page(s):
      48 in 204 packets (all packets)  --OK-- 
    Page no. 45:  204 packets  -- OK;  memory page(s):
      49 in 204 packets (all packets)  --OK-- 
    Page no. 46:  204 packets  -- OK;  memory page(s):
      50 in 204 packets (all packets)  --OK-- 
    Page no. 47:  204 packets  -- OK;  memory page(s):
      51 in 204 packets (all packets)  --OK-- 
    Page no. 48:  204 packets  -- OK;  memory page(s):
      52 in 204 packets (all packets)  --OK-- 
    Page no. 49:  204 packets  -- OK;  memory page(s):
      53 in 204 packets (all packets)  --OK-- 
    Page no. 50:  204 packets  -- OK;  memory page(s):
      54 in 204 packets (all packets)  --OK-- 
    Page no. 51:  204 packets  -- OK;  memory page(s):
      55 in 204 packets (all packets)  --OK-- 
    Page no. 52:  204 packets  -- OK;  memory page(s):
      56 in 204 packets (all packets)  --OK-- 
    Page no. 53:  204 packets  -- OK;  memory page(s):
      57 in 204 packets (all packets)  --OK-- 
    Page no. 54:  204 packets  -- OK;  memory page(s):
      58 in 204 packets (all packets)  --OK-- 
    Page no. 55:  204 packets  -- OK;  memory page(s):
      59 in 204 packets (all packets)  --OK-- 
    Page no. 56:  204 packets  -- OK;  memory page(s):
      60 in 204 packets (all packets)  --OK-- 
    Page no. 57:  204 packets  -- OK;  memory page(s):
      61 in 204 packets (all packets)  --OK-- 
    Page no. 58:  204 packets  -- OK;  memory page(s):
      62 in 204 packets (all packets)  --OK-- 
    Page no. 59:  204 packets  -- OK;  memory page(s):
      63 in 204 packets (all packets)  --OK-- 
    Page no. 60:  204 packets  -- OK;  memory page(s):
      64 in 204 packets (all packets)  --OK-- 
   Event no. 3 pages: 60  ordered packets: 12240
   TM values from -235 to 199 median: -4.00000
   1Hz and 1MHz counters -- OK -- (constant for all packets of this event):
          C1Hz=85 C1MHz=0.944438
   Differences of   LS 32 bits of C10MHz from descriptor (period 430s)  in neigboring packets of this event: 
      0.0000170 in 12239 packets (all packets)
      -- OK -- correct differences for all packets of this event
         LS 32 bits of C10MHz from descriptor (period 430s)  in the first packet of this event: 87.4936394
   Event type -- OK -- (constant for all packets of this event): External
   Gain -- OK -- (constant for all packets of this event): 1x
   Header abs. time  -- OK -- (constant for all packets of this event): 2023-06-01T09:57:55.944.438.000
   Event 1st sample abs. time : 2023-06-01T09:57:55.888.949.626)
   --- Rel. time correction for 1 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 85, 32bit 10MHz counter = 87.493639400 corrected by 40bit 10MHz counter of 74.604730900 obtained at 1Hz counter of 73. Result: 85.888954900
   XO time of the 1st EW sample = data descriptor XO time - 175 ns delay
   Event 1st sample XO time (reconstructed seconds from XP start):              85.888954725
   Event data descriptor XO time (reconstructed seconds from XP start)    :              85.888954900
   Sampling frequency correction factor for this event:     0.999996400
--
Extremes in separate events:

 - Event number: 1 from 2023-06-01T09:57:30.041.856.626 to 2023-06-01T09:57:30.249.937.375 (XO: 60.041858576 - 60.249939325)
   Min TM value:-238 at an offset of 156871.740 usec -> 2023-06-01T09:57:30.198.728.365 (XO: 60.198730315)
   Max TM value:194 at an offset of 66949.404 usec -> 2023-06-01T09:57:30.108.806.029 (XO: 60.108807979)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 206907.507 usec -> 2023-06-01T09:57:30.248.764.133 (XO: 60.248766083)
   Min dTM/dt [1/us]:-1.94e+03 at an offset of 66949.504 usec -> 2023-06-01T09:57:30.108.806.129 (XO: 60.108808079)
   Med: -4.00000   Ave: -4.4410158   Std: 6.9526126

 - Event number: 2 from 2023-06-01T09:57:50.919.287.626 to 2023-06-01T09:57:51.127.368.375 (XO: 80.919292834 - 81.127373584)
   Min TM value:-308 at an offset of 52737.015 usec -> 2023-06-01T09:57:50.972.024.640 (XO: 80.972029849)
   Max TM value:310 at an offset of 52732.265 usec -> 2023-06-01T09:57:50.972.019.890 (XO: 80.972025099)
   Max dTM/dt [1/us]:1.71e+03 at an offset of 90096.262 usec -> 2023-06-01T09:57:51.009.383.887 (XO: 81.009389096)
   Min dTM/dt [1/us]:-2.06e+03 at an offset of 80252.151 usec -> 2023-06-01T09:57:50.999.539.777 (XO: 80.999544986)
   Med: -4.00000   Ave: -4.4747249   Std: 8.2007000

 - Event number: 3 from 2023-06-01T09:57:55.888.949.626 to 2023-06-01T09:57:56.097.030.375 (XO: 85.888954725 - 86.097035474)
   Min TM value:-235 at an offset of 50800.795 usec -> 2023-06-01T09:57:55.939.750.421 (XO: 85.939755521)
   Max TM value:199 at an offset of 180858.551 usec -> 2023-06-01T09:57:56.069.808.177 (XO: 86.069813276)
   Max dTM/dt [1/us]:1.83e+03 at an offset of 50803.195 usec -> 2023-06-01T09:57:55.939.752.821 (XO: 85.939757921)
   Min dTM/dt [1/us]:-1.96e+03 at an offset of 180858.664 usec -> 2023-06-01T09:57:56.069.808.289 (XO: 86.069813389)
   Med: -4.00000   Ave: -4.4864809   Std: 8.8170688
