-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Jun  1 08:00:36 2023 UTC
-- File 20230531_223157_1227.dat, 72  packets found.
APID:
      1227 (SL) in 72 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 72 packets (all packets)
Total number of packets: 
      18 in 72 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.044078 to 55.986238
-each separately from 14s to 55s and from 44078us to 986238us.
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.779734 in 1 packets: 18
      20.340505 in 1 packets: 36
      18.820765 in 1 packets: 54
MIU packet time from 2023-05-31T22:32:14.044.078.000 to 2023-05-31T22:32:55.986.238.000
   Differences in neigboring packets: 
      0.000017 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.779734 in 1 packets: 18
      20.340505 in 1 packets: 36
      18.820765 in 1 packets: 54
MIU orbit number: 
      60000 in 72 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 72 packets (all packets)
MIU orbit information / number of events: 
      0 in 72 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 72 packets (all packets)
Instrument anomalies: 
      0 (OK) in 72 packets (all packets)
FPGA configuration version: 
      211 in 72 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.8777376 to 56.8775941
   Differences in neigboring packets: 
      0.0000000 in 68 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      1.9999929 in 1 packets: 18
      20.9999265 in 1 packets: 36
      17.9999371 in 1 packets: 54
Temperature FPGA: 
      01 -> 53.7 degC in 72 packets (all packets)
Temperature XO: 
      C4 -> 42.0 degC in 72 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 72 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 72 packets (all packets)
Number of external events: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      0 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Survey memory error: 
      0 in 72 packets (all packets)
Event memory error: 
      0 in 72 packets (all packets)
Configuration error: 
      0 in 72 packets (all packets)
SDRAM CRC error: 
      0 in 72 packets (all packets)
Radio gain: 
      0 in 72 packets (all packets)
Radio bandwidth: 
      0 in 72 packets (all packets)
Digital gain: 
      0 in 72 packets (all packets)
Analog gain: 
      0 in 72 packets (all packets)
 
SURVEY DATA READER
-- File 20230531_223157_1227.dat, 4 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      71 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.9218170 to 15.9221060
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 2.7797270  -- OK for SL --
   MIU time from 2023-05-31T22:32:14.044.078.000 to 2023-05-31T22:32:14.044.367.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 2.779734

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 18.7018330 to 18.7021220
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 20.3404310  -- OK for SL --
   MIU time from 2023-05-31T22:32:16.824.101.000 to 2023-05-31T22:32:16.824.390.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 20.340505

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      91 in 18 packets (all packets)  -- OK --
   10 MHz counter from 39.0425530 to 39.0428420
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.8207030  -- OK for SL --
   MIU time from 2023-05-31T22:32:37.164.895.000 to 2023-05-31T22:32:37.165.184.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.820765

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 57.8635450 to 57.8638340
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-05-31T22:32:55.985.949.000 to 2023-05-31T22:32:55.986.238.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 72 points based on 59 XO calibration points at 1Hz counter from 0 to 58
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 15.921817000 corrected by 40bit 10MHz counter of 15.877737600 obtained at 1Hz counter of 14. Result: 14.044079559
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-05-31T22:32:14.043.937.725
   Max 1st sample MIU time among all SL packets: 2023-05-31T22:32:55.986.097.725
   Min 1st sample XO time among all SL packets: 14.043939283
   Max 1st sample XO time among all SL packets: 55.986103175
   TM values from -306 to 206 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-05-31T22:32:14.043.937.725 (XO: 14.043939283 - 14.044245284, t=42.0 degC)  Rank: 00071
   TM values: Max =   198  Min =  -274  Ave =    -4.45  Std =    21.48
   Min TM value:-274 at an offset of 95.888 usec -> 2023-05-31T22:32:14.044.033.612 (XO: 14.044035171)
   Max TM value:198 at an offset of 100.813 usec -> 2023-05-31T22:32:14.044.038.537 (XO: 14.044040096)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 98.250 usec -> 2023-05-31T22:32:14.044.035.975 (XO: 14.044037534)
   Min dTM/dt [1/us]:-2.27e+03 at an offset of 95.750 usec -> 2023-05-31T22:32:14.044.033.475 (XO: 14.044035034)

 - Block number: 2 starting at 2023-05-31T22:32:16.823.960.725 (XO: 16.823965109 - 16.824271110, t=42.0 degC)  Rank: 00080
   TM values: Max =   186  Min =  -262  Ave =    -4.39  Std =    20.83
   Min TM value:-262 at an offset of 117.263 usec -> 2023-05-31T22:32:16.824.077.987 (XO: 16.824082372)
   Max TM value:186 at an offset of 117.050 usec -> 2023-05-31T22:32:16.824.077.775 (XO: 16.824082159)
   Max dTM/dt [1/us]:1.66e+03 at an offset of 119.713 usec -> 2023-05-31T22:32:16.824.080.437 (XO: 16.824084822)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 117.175 usec -> 2023-05-31T22:32:16.824.077.900 (XO: 16.824082284)

 - Block number: 3 starting at 2023-05-31T22:32:37.164.754.725 (XO: 37.164756302 - 37.165062303, t=42.0 degC)  Rank: 00091
   TM values: Max =   206  Min =  -306  Ave =    -4.43  Std =    22.46
   Min TM value:-306 at an offset of 117.550 usec -> 2023-05-31T22:32:37.164.872.275 (XO: 37.164873852)
   Max TM value:206 at an offset of 105.050 usec -> 2023-05-31T22:32:37.164.859.775 (XO: 37.164861352)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 115.413 usec -> 2023-05-31T22:32:37.164.870.137 (XO: 37.164871715)
   Min dTM/dt [1/us]:-2.31e+03 at an offset of 117.400 usec -> 2023-05-31T22:32:37.164.872.125 (XO: 37.164873702)

 - Block number: 4 starting at 2023-05-31T22:32:55.985.808.725 (XO: 55.985814174 - 55.986120175, t=42.0 degC)  Rank: 00072
   TM values: Max =   205  Min =  -267  Ave =    -4.43  Std =    21.96
   Min TM value:-267 at an offset of 120.538 usec -> 2023-05-31T22:32:55.985.929.262 (XO: 55.985934712)
   Max TM value:205 at an offset of 120.275 usec -> 2023-05-31T22:32:55.985.929.000 (XO: 55.985934449)
   Max dTM/dt [1/us]:1.84e+03 at an offset of 122.938 usec -> 2023-05-31T22:32:55.985.931.662 (XO: 55.985937112)
   Min dTM/dt [1/us]:-2.50e+03 at an offset of 120.388 usec -> 2023-05-31T22:32:55.985.929.112 (XO: 55.985934562)
