-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu Jun  1 02:33:39 2023 UTC
-- File 20230531_104522_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 7.680401 to 43.422768
-each separately from 7s to 43s and from 422479us to 680690us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.900524 in 1 packets: 18
      22.840976 in 1 packets: 36
MIU packet time from 2023-05-31T10:45:32.680.401.000 to 2023-05-31T10:46:08.422.768.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.900524 in 1 packets: 18
      22.840976 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 8.8958134 to 44.8956879
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      12.9999547 in 1 packets: 18
      22.9999198 in 1 packets: 36
Temperature FPGA: 
      01 -> 53.7 degC in 36 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      02 -> 52.4 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C4 -> 42.0 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Number of generated alerts: 
      0 in 19 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 35 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20230531_104522_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      80 in 18 packets (all packets)  -- OK --
   10 MHz counter from 9.5762159 to 9.5765049
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 12.9004790  -- OK for SL --
   MIU time from 2023-05-31T10:45:32.680.401.000 to 2023-05-31T10:45:32.680.690.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 12.900524

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 22.4769839 to 22.4772729
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.8408950  -- OK for SL --
   MIU time from 2023-05-31T10:45:45.581.214.000 to 2023-05-31T10:45:45.581.503.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.840976

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      76 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.3181679 to 45.3184569
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-05-31T10:46:08.422.479.000 to 2023-05-31T10:46:08.422.768.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 7, 32bit 10MHz counter = 9.576215900 corrected by 40bit 10MHz counter of 8.895813400 obtained at 1Hz counter of 7. Result: 7.680404881
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-05-31T10:45:32.680.260.725
   Max 1st sample MIU time among all SL packets: 2023-05-31T10:46:08.422.627.725
   Min 1st sample XO time among all SL packets: 7.680264606
   Max 1st sample XO time among all SL packets: 43.422630204
   TM values from -253 to 196 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-05-31T10:45:32.680.260.725 (XO: 7.680264606 - 7.680570607, t=42.0 degC)  Rank: 00080
   TM values: Max =   182  Min =  -245  Ave =    -4.45  Std =    18.81
   Min TM value:-245 at an offset of 124.625 usec -> 2023-05-31T10:45:32.680.385.350 (XO: 7.680389231)
   Max TM value:182 at an offset of 122.075 usec -> 2023-05-31T10:45:32.680.382.800 (XO: 7.680386681)
   Max dTM/dt [1/us]:1.88e+03 at an offset of 124.725 usec -> 2023-05-31T10:45:32.680.385.450 (XO: 7.680389331)
   Min dTM/dt [1/us]:-2.08e+03 at an offset of 122.213 usec -> 2023-05-31T10:45:32.680.382.937 (XO: 7.680386819)

 - Block number: 2 starting at 2023-05-31T10:45:45.581.073.725 (XO: 20.581077559 - 20.581383560, t=42.0 degC)  Rank: 00081
   TM values: Max =   196  Min =  -253  Ave =    -4.52  Std =    17.79
   Min TM value:-253 at an offset of 106.375 usec -> 2023-05-31T10:45:45.581.180.100 (XO: 20.581183934)
   Max TM value:196 at an offset of 111.350 usec -> 2023-05-31T10:45:45.581.185.075 (XO: 20.581188909)
   Max dTM/dt [1/us]:1.49e+03 at an offset of 108.913 usec -> 2023-05-31T10:45:45.581.182.637 (XO: 20.581186472)
   Min dTM/dt [1/us]:-1.76e+03 at an offset of 106.263 usec -> 2023-05-31T10:45:45.581.179.987 (XO: 20.581183822)

 - Block number: 3 starting at 2023-05-31T10:46:08.422.338.725 (XO: 43.422341203 - 43.422647204, t=42.0 degC)  Rank: 00076
   TM values: Max =   193  Min =  -243  Ave =    -4.46  Std =    17.70
   Min TM value:-243 at an offset of 121.950 usec -> 2023-05-31T10:46:08.422.460.675 (XO: 43.422463154)
   Max TM value:193 at an offset of 121.688 usec -> 2023-05-31T10:46:08.422.460.412 (XO: 43.422462891)
   Max dTM/dt [1/us]:1.82e+03 at an offset of 124.313 usec -> 2023-05-31T10:46:08.422.463.037 (XO: 43.422465516)
   Min dTM/dt [1/us]:-2.25e+03 at an offset of 121.788 usec -> 2023-05-31T10:46:08.422.460.512 (XO: 43.422462991)
