-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed May 31 02:59:55 2023 UTC
-- File 20230530_150109_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 13.621860 to 40.427075
-each separately from 13s to 40s and from 327534us to 622149us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.705385 in 1 packets: 18
      24.098963 in 1 packets: 36
MIU packet time from 2023-05-30T15:01:25.621.860.000 to 2023-05-30T15:01:52.427.075.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.705385 in 1 packets: 18
      24.098963 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 14.6044925 to 41.6043979
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      2.9999895 in 1 packets: 18
      23.9999159 in 1 packets: 36
Temperature FPGA: 
      01 -> 53.7 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      00 -> 55.0 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
      02 -> 52.4 degC in 18 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Temperature XO: 
      C4 -> 42.0 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 53 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 53 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20230530_150109_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      100 in 18 packets (all packets)  -- OK --
   10 MHz counter from 15.2263536 to 15.2266426
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 2.7053750  -- OK for SL --
   MIU time from 2023-05-30T15:01:25.621.860.000 to 2023-05-30T15:01:25.622.149.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 2.705385

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 17.9320176 to 17.9323066
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.0988790  -- OK for SL --
   MIU time from 2023-05-30T15:01:28.327.534.000 to 2023-05-30T15:01:28.327.823.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.098963

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      72 in 18 packets (all packets)  -- OK --
   10 MHz counter from 42.0311856 to 42.0314746
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-05-30T15:01:52.426.786.000 to 2023-05-30T15:01:52.427.075.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 13, 32bit 10MHz counter = 15.226353600 corrected by 40bit 10MHz counter of 14.604492500 obtained at 1Hz counter of 13. Result: 13.621863277
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996600
   Min 1st sample MIU time among all SL packets: 2023-05-30T15:01:25.621.719.725
   Max 1st sample MIU time among all SL packets: 2023-05-30T15:01:52.426.934.725
   Min 1st sample XO time among all SL packets: 13.621723001
   Max 1st sample XO time among all SL packets: 40.426937877
   TM values from -263 to 206 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-05-30T15:01:25.621.719.725 (XO: 13.621723001 - 13.622029002, t=42.0 degC)  Rank: 00100
   TM values: Max =   117  Min =  -108  Ave =    -4.62  Std =    22.21
   Min TM value:-108 at an offset of 79.625 usec -> 2023-05-30T15:01:25.621.799.350 (XO: 13.621802626)
   Max TM value:117 at an offset of 87.450 usec -> 2023-05-30T15:01:25.621.807.175 (XO: 13.621810451)
   Max dTM/dt [1/us]:2.22e+02 at an offset of 80.438 usec -> 2023-05-30T15:01:25.621.800.162 (XO: 13.621803439)
   Min dTM/dt [1/us]:-2.17e+02 at an offset of 38.150 usec -> 2023-05-30T15:01:25.621.757.875 (XO: 13.621761151)

 - Block number: 2 starting at 2023-05-30T15:01:28.327.393.725 (XO: 16.327396471 - 16.327702472, t=42.0 degC)  Rank: 00078
   TM values: Max =   171  Min =  -233  Ave =    -4.49  Std =    18.11
   Min TM value:-233 at an offset of 110.475 usec -> 2023-05-30T15:01:28.327.504.200 (XO: 16.327506946)
   Max TM value:171 at an offset of 110.200 usec -> 2023-05-30T15:01:28.327.503.925 (XO: 16.327506671)
   Max dTM/dt [1/us]:1.77e+03 at an offset of 112.863 usec -> 2023-05-30T15:01:28.327.506.587 (XO: 16.327509334)
   Min dTM/dt [1/us]:-2.09e+03 at an offset of 110.338 usec -> 2023-05-30T15:01:28.327.504.062 (XO: 16.327506809)

 - Block number: 3 starting at 2023-05-30T15:01:52.426.645.725 (XO: 40.426648876 - 40.426954877, t=42.0 degC)  Rank: 00072
   TM values: Max =   206  Min =  -263  Ave =    -4.41  Std =    26.51
   Min TM value:-263 at an offset of 103.800 usec -> 2023-05-30T15:01:52.426.749.525 (XO: 40.426752676)
   Max TM value:206 at an offset of 101.313 usec -> 2023-05-30T15:01:52.426.747.037 (XO: 40.426750188)
   Max dTM/dt [1/us]:1.93e+03 at an offset of 103.913 usec -> 2023-05-30T15:01:52.426.749.637 (XO: 40.426752788)
   Min dTM/dt [1/us]:-2.39e+03 at an offset of 101.425 usec -> 2023-05-30T15:01:52.426.747.150 (XO: 40.426750301)
