-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed May 31 01:33:00 2023 UTC
-- File 20230530_130236_1227.dat, 90  packets found.
APID:
      1227 (SL) in 90 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 90 packets (all packets)
Total number of packets: 
      18 in 90 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.783522 to 64.883024
-each separately from 10s to 64s and from 223333us to 883024us.
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.439522 in 1 packets: 18
      13.119452 in 1 packets: 36
      13.129581 in 1 packets: 54
      18.409502 in 1 packets: 72
MIU packet time from 2023-05-30T13:02:49.783.522.000 to 2023-05-30T13:03:43.883.024.000
   Differences in neigboring packets: 
      0.000017 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.439522 in 1 packets: 18
      13.119452 in 1 packets: 36
      13.129581 in 1 packets: 54
      18.409502 in 1 packets: 72
MIU orbit number: 
      60000 in 90 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 90 packets (all packets)
MIU orbit information / number of events: 
      0 in 90 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 90 packets (all packets)
Instrument anomalies: 
      0 (OK) in 90 packets (all packets)
FPGA configuration version: 
      211 in 90 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.3851632 to 65.3849741
   Differences in neigboring packets: 
      0.0000000 in 85 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      9.9999650 in 1 packets: 18
      12.9999545 in 1 packets: 36
      12.9999544 in 1 packets: 54
      17.9999370 in 1 packets: 72
Temperature FPGA: 
      01 -> 53.7 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      00 -> 55.0 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C4 -> 42.0 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      C3 -> 40.4 degC in 36 packets: 37, 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, ...
Temperature ant 3: 
      01 -> 108.0 degC in 90 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 90 packets (all packets)
Number of external events: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73
      2 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Number of generated alerts: 
      0 in 55 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      1 in 18 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73
      2 in 17 packets: 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Survey memory error: 
      0 in 90 packets (all packets)
Event memory error: 
      0 in 90 packets (all packets)
Configuration error: 
      0 in 90 packets (all packets)
SDRAM CRC error: 
      0 in 90 packets (all packets)
Radio gain: 
      0 in 90 packets (all packets)
Radio bandwidth: 
      0 in 90 packets (all packets)
Digital gain: 
      0 in 90 packets (all packets)
Analog gain: 
      0 in 90 packets (all packets)
 
SURVEY DATA READER
-- File 20230530_130236_1227.dat, 5 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      79 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.1686863 to 12.1689753
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.4394870  -- OK for SL --
   MIU time from 2023-05-30T13:02:49.783.522.000 to 2023-05-30T13:02:49.783.811.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.439522

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      78 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.6084623 to 21.6087513
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.1194070  -- OK for SL --
   MIU time from 2023-05-30T13:02:59.223.333.000 to 2023-05-30T13:02:59.223.622.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.119452

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      69 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.7281583 to 34.7284473
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.1295350  -- OK for SL --
   MIU time from 2023-05-30T13:03:12.343.074.000 to 2023-05-30T13:03:12.343.363.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.129581

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      81 in 18 packets (all packets)  -- OK --
   10 MHz counter from 47.8579823 to 47.8582713
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.4094390  -- OK for SL --
   MIU time from 2023-05-30T13:03:25.472.944.000 to 2023-05-30T13:03:25.473.233.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.409502

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      82 in 18 packets (all packets)  -- OK --
   10 MHz counter from 66.2677103 to 66.2679993
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-05-30T13:03:43.882.735.000 to 2023-05-30T13:03:43.883.024.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 90 points based on 74 XO calibration points at 1Hz counter from 0 to 73
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.168686300 corrected by 40bit 10MHz counter of 11.385163200 obtained at 1Hz counter of 10. Result: 10.783525842
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-05-30T13:02:49.783.381.725
   Max 1st sample MIU time among all SL packets: 2023-05-30T13:03:43.882.883.725
   Min 1st sample XO time among all SL packets: 10.783385567
   Max 1st sample XO time among all SL packets: 64.882888015
   TM values from -297 to 224 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-05-30T13:02:49.783.381.725 (XO: 10.783385567 - 10.783691568, t=42.0 degC)  Rank: 00079
   TM values: Max =   202  Min =  -258  Ave =    -4.41  Std =    20.72
   Min TM value:-258 at an offset of 120.300 usec -> 2023-05-30T13:02:49.783.502.025 (XO: 10.783505867)
   Max TM value:202 at an offset of 105.763 usec -> 2023-05-30T13:02:49.783.487.487 (XO: 10.783491330)
   Max dTM/dt [1/us]:1.96e+03 at an offset of 120.400 usec -> 2023-05-30T13:02:49.783.502.125 (XO: 10.783505967)
   Min dTM/dt [1/us]:-2.20e+03 at an offset of 117.913 usec -> 2023-05-30T13:02:49.783.499.637 (XO: 10.783503480)

 - Block number: 2 starting at 2023-05-30T13:02:59.223.192.725 (XO: 20.223194629 - 20.223500630, t=42.0 degC)  Rank: 00078
   TM values: Max =   168  Min =  -246  Ave =    -4.48  Std =    17.47
   Min TM value:-246 at an offset of 116.775 usec -> 2023-05-30T13:02:59.223.309.500 (XO: 20.223311404)
   Max TM value:168 at an offset of 116.513 usec -> 2023-05-30T13:02:59.223.309.237 (XO: 20.223311141)
   Max dTM/dt [1/us]:1.90e+03 at an offset of 119.150 usec -> 2023-05-30T13:02:59.223.311.875 (XO: 20.223313779)
   Min dTM/dt [1/us]:-2.10e+03 at an offset of 116.650 usec -> 2023-05-30T13:02:59.223.309.375 (XO: 20.223311279)

 - Block number: 3 starting at 2023-05-30T13:03:12.342.933.725 (XO: 33.342936560 - 33.343242561, t=40.4 degC)  Rank: 00069
   TM values: Max =   189  Min =  -253  Ave =    -4.45  Std =    21.35
   Min TM value:-253 at an offset of 83.325 usec -> 2023-05-30T13:03:12.343.017.050 (XO: 33.343019885)
   Max TM value:189 at an offset of 83.600 usec -> 2023-05-30T13:03:12.343.017.325 (XO: 33.343020160)
   Max dTM/dt [1/us]:2.01e+03 at an offset of 83.438 usec -> 2023-05-30T13:03:12.343.017.162 (XO: 33.343019997)
   Min dTM/dt [1/us]:-1.88e+03 at an offset of 83.250 usec -> 2023-05-30T13:03:12.343.016.975 (XO: 33.343019810)

 - Block number: 4 starting at 2023-05-30T13:03:25.472.803.725 (XO: 46.472806580 - 46.473112581, t=42.0 degC)  Rank: 00081
   TM values: Max =   187  Min =  -243  Ave =    -4.45  Std =    21.43
   Min TM value:-243 at an offset of 116.250 usec -> 2023-05-30T13:03:25.472.919.975 (XO: 46.472922830)
   Max TM value:187 at an offset of 121.163 usec -> 2023-05-30T13:03:25.472.924.887 (XO: 46.472927743)
   Max dTM/dt [1/us]:1.64e+03 at an offset of 114.163 usec -> 2023-05-30T13:03:25.472.917.887 (XO: 46.472920743)
   Min dTM/dt [1/us]:-1.72e+03 at an offset of 103.850 usec -> 2023-05-30T13:03:25.472.907.575 (XO: 46.472910430)

 - Block number: 5 starting at 2023-05-30T13:03:43.882.594.725 (XO: 64.882599014 - 64.882905015, t=40.4 degC)  Rank: 00082
   TM values: Max =   224  Min =  -297  Ave =    -4.56  Std =    24.03
   Min TM value:-297 at an offset of 122.725 usec -> 2023-05-30T13:03:43.882.717.450 (XO: 64.882721740)
   Max TM value:224 at an offset of 120.300 usec -> 2023-05-30T13:03:43.882.715.025 (XO: 64.882719315)
   Max dTM/dt [1/us]:2.39e+03 at an offset of 122.825 usec -> 2023-05-30T13:03:43.882.717.550 (XO: 64.882721840)
   Min dTM/dt [1/us]:-2.55e+03 at an offset of 120.413 usec -> 2023-05-30T13:03:43.882.715.137 (XO: 64.882719427)
