-- OK --
-----------------> BLESK ver.2021-02-27 run on Wed May 31 00:45:35 2023 UTC
-- File 20230530_114427_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.451531 to 26.032027
-each separately from 8s to 26s and from 31738us to 451820us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.579918 in 1 packets: 18
MIU packet time from 2023-05-30T11:44:38.451.531.000 to 2023-05-30T11:44:56.032.027.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.579918 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 9.8890191 to 27.8889560
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999369 in 1 packets: 18
Temperature FPGA: 
      00 -> 55.0 degC in 36 packets (all packets)
Temperature XO: 
      C3 -> 40.4 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      C4 -> 42.0 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 17 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      2 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230530_114427_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      85 in 18 packets (all packets)  -- OK --
   10 MHz counter from 10.3405516 to 10.3408406
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 17.5798550  -- OK for SL --
   MIU time from 2023-05-30T11:44:38.451.531.000 to 2023-05-30T11:44:38.451.820.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 17.579918

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      75 in 18 packets (all packets)  -- OK --
   10 MHz counter from 27.9206956 to 27.9209846
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-05-30T11:44:56.031.738.000 to 2023-05-30T11:44:56.032.027.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 10.340551600 corrected by 40bit 10MHz counter of 9.889019100 obtained at 1Hz counter of 8. Result: 8.451534126
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-05-30T11:44:38.451.390.725
   Max 1st sample MIU time among all SL packets: 2023-05-30T11:44:56.031.886.725
   Min 1st sample XO time among all SL packets: 8.451393850
   Max 1st sample XO time among all SL packets: 26.031888437
   TM values from -281 to 192 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-05-30T11:44:38.451.390.725 (XO: 8.451393850 - 8.451699851, t=40.4 degC)  Rank: 00085
   TM values: Max =   190  Min =  -265  Ave =    -4.55  Std =    17.89
   Min TM value:-265 at an offset of 123.075 usec -> 2023-05-30T11:44:38.451.513.800 (XO: 8.451516925)
   Max TM value:190 at an offset of 127.963 usec -> 2023-05-30T11:44:38.451.518.687 (XO: 8.451521813)
   Max dTM/dt [1/us]:1.41e+03 at an offset of 125.463 usec -> 2023-05-30T11:44:38.451.516.187 (XO: 8.451519313)
   Min dTM/dt [1/us]:-1.78e+03 at an offset of 122.950 usec -> 2023-05-30T11:44:38.451.513.675 (XO: 8.451516800)

 - Block number: 2 starting at 2023-05-30T11:44:56.031.597.725 (XO: 26.031599436 - 26.031905437, t=42.0 degC)  Rank: 00075
   TM values: Max =   192  Min =  -281  Ave =    -4.52  Std =    17.78
   Min TM value:-281 at an offset of 128.475 usec -> 2023-05-30T11:44:56.031.726.200 (XO: 26.031727911)
   Max TM value:192 at an offset of 126.013 usec -> 2023-05-30T11:44:56.031.723.737 (XO: 26.031725449)
   Max dTM/dt [1/us]:2.22e+03 at an offset of 128.588 usec -> 2023-05-30T11:44:56.031.726.312 (XO: 26.031728024)
   Min dTM/dt [1/us]:-2.17e+03 at an offset of 128.400 usec -> 2023-05-30T11:44:56.031.726.125 (XO: 26.031727836)
