-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun May 28 23:27:45 2023 UTC
-- File 20230528_120212_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 3.652758 to 86.943003
-each separately from 3s to 86s and from 74845us to 943003us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.214290 in 1 packets: 18
      19.918633 in 1 packets: 36
      13.758483 in 1 packets: 54
      5.529525 in 1 packets: 72
      25.867580 in 1 packets: 90
MIU packet time from 2023-05-28T12:02:18.652.758.000 to 2023-05-28T12:03:41.943.003.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      18.214290 in 1 packets: 18
      19.918633 in 1 packets: 36
      13.758483 in 1 packets: 54
      5.529525 in 1 packets: 72
      25.867580 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 4.7583395 to 87.7580313
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      17.9999336 in 1 packets: 18
      19.9999258 in 1 packets: 36
      13.9999479 in 1 packets: 54
      5.9999775 in 1 packets: 72
      24.9999070 in 1 packets: 90
Temperature FPGA: 
      01 -> 53.7 degC in 90 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      02 -> 52.4 degC in 18 packets: 73, 74, 75, 76, 77, 78, 79, 80, 81, 82, 83, 84, 85, 86, 87, 88, 89, 90
Temperature XO: 
      C3 -> 40.4 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 53 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 54 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, ...
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 53 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      2 in 1 packets: 55
      3 in 53 packets: 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, 73, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20230528_120212_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      100 in 18 packets (all packets)  -- OK --
   10 MHz counter from 5.4110989 to 5.4113879
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 18.2142230  -- OK for SL --
   MIU time from 2023-05-28T12:02:18.652.758.000 to 2023-05-28T12:02:18.653.047.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 18.214290

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      67 in 18 packets (all packets)  -- OK --
   10 MHz counter from 23.6256109 to 23.6258999
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 19.9185590  -- OK for SL --
   MIU time from 2023-05-28T12:02:36.867.337.000 to 2023-05-28T12:02:36.867.626.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 19.918633

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      68 in 18 packets (all packets)  -- OK --
   10 MHz counter from 43.5444589 to 43.5447479
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.7584310  -- OK for SL --
   MIU time from 2023-05-28T12:02:56.786.259.000 to 2023-05-28T12:02:56.786.548.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.758483

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      149 in 18 packets (all packets)  -- OK --
   10 MHz counter from 57.3031789 to 57.3034679
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 5.5295030  -- OK for SL --
   MIU time from 2023-05-28T12:03:10.545.031.000 to 2023-05-28T12:03:10.545.320.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 5.529525

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      70 in 18 packets (all packets)  -- OK --
   10 MHz counter from 62.8329709 to 62.8332599
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 25.8674870  -- OK for SL --
   MIU time from 2023-05-28T12:03:16.074.845.000 to 2023-05-28T12:03:16.075.134.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 25.867580

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      68 in 18 packets (all packets)  -- OK --
   10 MHz counter from 88.7007469 to 88.7010359
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-05-28T12:03:41.942.714.000 to 2023-05-28T12:03:41.943.003.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 3, 32bit 10MHz counter = 5.411098900 corrected by 40bit 10MHz counter of 4.758339500 obtained at 1Hz counter of 3. Result: 3.652761815
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996200 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2023-05-28T12:02:18.652.617.725
   Max 1st sample MIU time among all SL packets: 2023-05-28T12:03:41.942.862.725
   Min 1st sample XO time among all SL packets: 3.652621540
   Max 1st sample XO time among all SL packets: 86.942867908
   TM values from -360 to 236 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-05-28T12:02:18.652.617.725 (XO: 3.652621540 - 3.652927541, t=40.4 degC)  Rank: 00100
   TM values: Max =   209  Min =  -238  Ave =    -4.38  Std =    23.46
   Min TM value:-238 at an offset of 65.700 usec -> 2023-05-28T12:02:18.652.683.425 (XO: 3.652687240)
   Max TM value:209 at an offset of 196.776 usec -> 2023-05-28T12:02:18.652.814.500 (XO: 3.652818315)
   Max dTM/dt [1/us]:1.39e+03 at an offset of 196.688 usec -> 2023-05-28T12:02:18.652.814.413 (XO: 3.652818228)
   Min dTM/dt [1/us]:-1.16e+03 at an offset of 98.350 usec -> 2023-05-28T12:02:18.652.716.075 (XO: 3.652719890)

 - Block number: 2 starting at 2023-05-28T12:02:36.867.196.725 (XO: 21.867200734 - 21.867506735, t=40.4 degC)  Rank: 00067
   TM values: Max =   165  Min =  -205  Ave =    -4.33  Std =    15.49
   Min TM value:-205 at an offset of 120.613 usec -> 2023-05-28T12:02:36.867.317.337 (XO: 21.867321347)
   Max TM value:165 at an offset of 108.900 usec -> 2023-05-28T12:02:36.867.305.625 (XO: 21.867309634)
   Max dTM/dt [1/us]:1.24e+03 at an offset of 123.075 usec -> 2023-05-28T12:02:36.867.319.800 (XO: 21.867323809)
   Min dTM/dt [1/us]:-1.71e+03 at an offset of 120.475 usec -> 2023-05-28T12:02:36.867.317.200 (XO: 21.867321209)

 - Block number: 3 starting at 2023-05-28T12:02:56.786.118.725 (XO: 41.786122634 - 41.786428635, t=40.4 degC)  Rank: 00068
   TM values: Max =   163  Min =  -208  Ave =    -4.44  Std =    16.20
   Min TM value:-208 at an offset of 108.700 usec -> 2023-05-28T12:02:56.786.227.425 (XO: 41.786231334)
   Max TM value:163 at an offset of 96.063 usec -> 2023-05-28T12:02:56.786.214.787 (XO: 41.786218697)
   Max dTM/dt [1/us]:1.39e+03 at an offset of 111.138 usec -> 2023-05-28T12:02:56.786.229.862 (XO: 41.786233772)
   Min dTM/dt [1/us]:-1.86e+03 at an offset of 108.563 usec -> 2023-05-28T12:02:56.786.227.287 (XO: 41.786231197)

 - Block number: 4 starting at 2023-05-28T12:03:10.544.890.725 (XO: 55.544893841 - 55.545199842, t=40.4 degC)  Rank: 00149
   TM values: Max =   236  Min =  -360  Ave =    -4.49  Std =    45.81
   Min TM value:-360 at an offset of 106.263 usec -> 2023-05-28T12:03:10.544.996.987 (XO: 55.545000104)
   Max TM value:236 at an offset of 109.538 usec -> 2023-05-28T12:03:10.545.000.262 (XO: 55.545003379)
   Max dTM/dt [1/us]:1.51e+03 at an offset of 105.138 usec -> 2023-05-28T12:03:10.544.995.862 (XO: 55.544998979)
   Min dTM/dt [1/us]:-1.55e+03 at an offset of 116.138 usec -> 2023-05-28T12:03:10.545.006.862 (XO: 55.545009979)

 - Block number: 5 starting at 2023-05-28T12:03:16.074.704.725 (XO: 61.074706601 - 61.075012603, t=40.4 degC)  Rank: 00070
   TM values: Max =   156  Min =  -186  Ave =    -4.38  Std =    16.34
   Min TM value:-186 at an offset of 108.050 usec -> 2023-05-28T12:03:16.074.812.775 (XO: 61.074814652)
   Max TM value:156 at an offset of 113.188 usec -> 2023-05-28T12:03:16.074.817.912 (XO: 61.074819789)
   Max dTM/dt [1/us]:1.60e+03 at an offset of 110.575 usec -> 2023-05-28T12:03:16.074.815.300 (XO: 61.074817177)
   Min dTM/dt [1/us]:-1.59e+03 at an offset of 107.950 usec -> 2023-05-28T12:03:16.074.812.675 (XO: 61.074814552)

 - Block number: 6 starting at 2023-05-28T12:03:41.942.573.725 (XO: 86.942578907 - 86.942884908, t=40.4 degC)  Rank: 00068
   TM values: Max =   186  Min =  -216  Ave =    -4.50  Std =    15.50
   Min TM value:-216 at an offset of 125.363 usec -> 2023-05-28T12:03:41.942.699.087 (XO: 86.942704270)
   Max TM value:186 at an offset of 122.863 usec -> 2023-05-28T12:03:41.942.696.587 (XO: 86.942701770)
   Max dTM/dt [1/us]:1.70e+03 at an offset of 125.500 usec -> 2023-05-28T12:03:41.942.699.225 (XO: 86.942704407)
   Min dTM/dt [1/us]:-1.90e+03 at an offset of 122.950 usec -> 2023-05-28T12:03:41.942.696.675 (XO: 86.942701857)
