-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun May 28 22:26:58 2023 UTC
-- File 20230528_071408_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.487051 to 43.753307
-each separately from 12s to 43s and from 344116us to 753307us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.856776 in 1 packets: 18
      24.408613 in 1 packets: 36
MIU packet time from 2023-05-28T07:14:23.487.051.000 to 2023-05-28T07:14:54.753.307.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.856776 in 1 packets: 18
      24.408613 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.8154797 to 44.8153685
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      6.9999749 in 1 packets: 18
      23.9999139 in 1 packets: 36
Temperature FPGA: 
      00 -> 55.0 degC in 54 packets (all packets)
Temperature XO: 
      C3 -> 40.4 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20230528_071408_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      61 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.3025318 to 14.3028208
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 6.8567510  -- OK for SL --
   MIU time from 2023-05-28T07:14:23.487.051.000 to 2023-05-28T07:14:23.487.340.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 6.856776

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      63 in 18 packets (all packets)  -- OK --
   10 MHz counter from 21.1595718 to 21.1598608
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 24.4085270  -- OK for SL --
   MIU time from 2023-05-28T07:14:30.344.116.000 to 2023-05-28T07:14:30.344.405.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 24.408613

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      58 in 18 packets (all packets)  -- OK --
   10 MHz counter from 45.5683878 to 45.5686768
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-05-28T07:14:54.753.018.000 to 2023-05-28T07:14:54.753.307.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.302531800 corrected by 40bit 10MHz counter of 13.815479700 obtained at 1Hz counter of 12. Result: 12.487053853
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2023-05-28T07:14:23.486.910.725
   Max 1st sample MIU time among all SL packets: 2023-05-28T07:14:54.753.166.725
   Min 1st sample XO time among all SL packets: 12.486913578
   Max 1st sample XO time among all SL packets: 43.753170812
   TM values from -238 to 188 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-05-28T07:14:23.486.910.725 (XO: 12.486913578 - 12.487219579, t=40.4 degC)  Rank: 00061
   TM values: Max =   179  Min =  -195  Ave =    -4.50  Std =    18.48
   Min TM value:-195 at an offset of 124.138 usec -> 2023-05-28T07:14:23.487.034.862 (XO: 12.487037716)
   Max TM value:179 at an offset of 111.375 usec -> 2023-05-28T07:14:23.487.022.100 (XO: 12.487024953)
   Max dTM/dt [1/us]:1.40e+03 at an offset of 111.263 usec -> 2023-05-28T07:14:23.487.021.987 (XO: 12.487024841)
   Min dTM/dt [1/us]:-1.67e+03 at an offset of 111.488 usec -> 2023-05-28T07:14:23.487.022.212 (XO: 12.487025066)

 - Block number: 2 starting at 2023-05-28T07:14:30.343.975.725 (XO: 19.343978163 - 19.344284164, t=40.4 degC)  Rank: 00063
   TM values: Max =   184  Min =  -223  Ave =    -4.42  Std =    18.69
   Min TM value:-223 at an offset of 126.925 usec -> 2023-05-28T07:14:30.344.102.650 (XO: 19.344105089)
   Max TM value:184 at an offset of 124.425 usec -> 2023-05-28T07:14:30.344.100.150 (XO: 19.344102589)
   Max dTM/dt [1/us]:1.76e+03 at an offset of 127.038 usec -> 2023-05-28T07:14:30.344.102.762 (XO: 19.344105201)
   Min dTM/dt [1/us]:-1.87e+03 at an offset of 124.525 usec -> 2023-05-28T07:14:30.344.100.250 (XO: 19.344102689)

 - Block number: 3 starting at 2023-05-28T07:14:54.752.877.725 (XO: 43.752881811 - 43.753187812, t=40.4 degC)  Rank: 00058
   TM values: Max =   188  Min =  -238  Ave =    -4.45  Std =    23.32
   Min TM value:-238 at an offset of 116.625 usec -> 2023-05-28T07:14:54.752.994.350 (XO: 43.752998436)
   Max TM value:188 at an offset of 158.501 usec -> 2023-05-28T07:14:54.753.036.225 (XO: 43.753040311)
   Max dTM/dt [1/us]:1.79e+03 at an offset of 116.738 usec -> 2023-05-28T07:14:54.752.994.462 (XO: 43.752998549)
   Min dTM/dt [1/us]:-1.78e+03 at an offset of 114.275 usec -> 2023-05-28T07:14:54.752.992.000 (XO: 43.752996086)
