-- OK --
-----------------> BLESK ver.2021-02-27 run on Sun May 28 00:03:17 2023 UTC
-- File 20230527_175309_1226.dat, 18  packets found.
APID:
      1226 (SR) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.000000 to 14.177029
-each separately from 0s to 14s and from 0us to 841254us.
   Differences in neigboring packets: 
      0.833939 in 3 packets: 1, 7, 13
      0.833943 in 4 packets: 2, 5, 10, 16
      0.833944 in 10 packets: 3, 4, 6, 8, 9, 11, 12, 14, 15, 17
MIU packet time from 2023-05-27T17:53:12.000.000.000 to 2023-05-27T17:53:26.177.029.000
   Differences in neigboring packets: 
      0.833939 in 3 packets: 1, 7, 13
      0.833943 in 4 packets: 2, 5, 10, 16
      0.833944 in 10 packets: 3, 4, 6, 8, 9, 11, 12, 14, 15, 17
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.5456710 to 15.5456211
   Differences in neigboring packets: 
      0.0000000 in 3 packets: 1, 7, 13
      0.9999964 in 9 packets: 2, 4, 6, 9, 11, 12, 15, 16, 17
      0.9999965 in 5 packets: 3, 5, 8, 10, 14
   XO PPM: 
      -3.6 in 9 packets: 2, 4, 6, 9, 11, 12, 15, 16, 17
      -3.5 in 5 packets: 3, 5, 8, 10, 14
Temperature FPGA: 
      01 -> 53.7 degC in 10 packets: 1, 3, 5, 6, 9, 10, 11, 12, 15, 16
      00 -> 55.0 degC in 8 packets: 2, 4, 7, 8, 13, 14, 17, 18
Temperature XO: 
      C3 -> 40.4 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230527_175309_1226.dat, 1 SR data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      0 in 18 packets (all packets)  -- OK --
   10 MHz counter from 1.5456711 to 15.7226511
      Difference between neigboring packets: 
      0.8339400 in 17 packets (all packets)  -- OK --
   MIU time from 2023-05-27T17:53:12.000.000.000 to 2023-05-27T17:53:26.177.029.000
      Difference between neigboring packets: 
      0.833939 in 3 packets: 1, 7, 13
      0.833943 in 4 packets: 2, 5, 10, 16
      0.833944 in 10 packets: 3, 4, 6, 8, 9, 11, 12, 14, 15, 17
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 1.545671100 corrected by 40bit 10MHz counter of 1.545671000 obtained at 1Hz counter of 0. Result: 0.000000100
--
   MIU or XO time of the 1st SR sample = header (MIU or corrected XO) time - 75 ns delay
   Samplig frequency correction factor in all SR packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SR packets: 2023-05-27T17:53:11.999.999.925
   Max 1st sample MIU time among all SR packets: 2023-05-27T17:53:26.177.028.925
   Min 1st sample XO time among all SR packets: 0.000000025
   Max 1st sample XO time among all SR packets: 14.177030562
   TM values from -148 to 144 median:-4.00000
--
Characteristics of separate SR data packets:

 - Packet number: 1 starting at 2023-05-27T17:53:11.999.999.925 (XO: 0.000000025 - 0.000017025, t=40.4 degC)
   TM values: Max =   111  Min =  -113  Ave =    -4.37  Std =    43.56

 - Packet number: 2 starting at 2023-05-27T17:53:12.833.938.925 (XO: 0.833943027 - 0.833960027, t=40.4 degC)
   TM values: Max =   144  Min =  -145  Ave =    -4.48  Std =    54.18

 - Packet number: 3 starting at 2023-05-27T17:53:13.667.881.925 (XO: 1.667885963 - 1.667902963, t=40.4 degC)
   TM values: Max =   105  Min =  -104  Ave =    -4.42  Std =    43.60

 - Packet number: 4 starting at 2023-05-27T17:53:14.501.825.925 (XO: 2.501828932 - 2.501845932, t=40.4 degC)
   TM values: Max =   126  Min =  -134  Ave =    -4.51  Std =    54.90

 - Packet number: 5 starting at 2023-05-27T17:53:15.335.769.925 (XO: 3.335771900 - 3.335788900, t=40.4 degC)
   TM values: Max =   118  Min =  -132  Ave =    -4.39  Std =    53.91

 - Packet number: 6 starting at 2023-05-27T17:53:16.169.712.925 (XO: 4.169714836 - 4.169731836, t=40.4 degC)
   TM values: Max =   126  Min =  -130  Ave =    -4.33  Std =    54.60

 - Packet number: 7 starting at 2023-05-27T17:53:17.003.656.925 (XO: 5.003657838 - 5.003674838, t=40.4 degC)
   TM values: Max =   111  Min =  -117  Ave =    -4.46  Std =    54.10

 - Packet number: 8 starting at 2023-05-27T17:53:17.837.595.925 (XO: 5.837600757 - 5.837617757, t=40.4 degC)
   TM values: Max =   110  Min =  -118  Ave =    -4.51  Std =    54.96

 - Packet number: 9 starting at 2023-05-27T17:53:18.671.539.925 (XO: 6.671543743 - 6.671560743, t=40.4 degC)
   TM values: Max =   121  Min =  -130  Ave =    -4.25  Std =    54.68

 - Packet number: 10 starting at 2023-05-27T17:53:19.505.483.925 (XO: 7.505486694 - 7.505503694, t=40.4 degC)
   TM values: Max =   139  Min =  -148  Ave =    -4.53  Std =    59.83

 - Packet number: 11 starting at 2023-05-27T17:53:20.339.426.925 (XO: 8.339429647 - 8.339446647, t=40.4 degC)
   TM values: Max =   113  Min =  -126  Ave =    -4.51  Std =    49.73

 - Packet number: 12 starting at 2023-05-27T17:53:21.173.370.925 (XO: 9.173372649 - 9.173389649, t=40.4 degC)
   TM values: Max =   109  Min =  -127  Ave =    -4.45  Std =    48.42

 - Packet number: 13 starting at 2023-05-27T17:53:22.007.314.925 (XO: 10.007315651 - 10.007332651, t=40.4 degC)
   TM values: Max =   119  Min =  -119  Ave =    -4.58  Std =    48.72

 - Packet number: 14 starting at 2023-05-27T17:53:22.841.253.925 (XO: 10.841258569 - 10.841275569, t=40.4 degC)
   TM values: Max =   111  Min =  -120  Ave =    -4.21  Std =    47.03

 - Packet number: 15 starting at 2023-05-27T17:53:23.675.197.925 (XO: 11.675201556 - 11.675218556, t=40.4 degC)
   TM values: Max =   132  Min =  -144  Ave =    -4.32  Std =    66.77

 - Packet number: 16 starting at 2023-05-27T17:53:24.509.141.925 (XO: 12.509144558 - 12.509161558, t=40.4 degC)
   TM values: Max =    93  Min =   -95  Ave =    -4.29  Std =    42.52

 - Packet number: 17 starting at 2023-05-27T17:53:25.343.084.925 (XO: 13.343087560 - 13.343104560, t=40.4 degC)
   TM values: Max =   111  Min =  -111  Ave =    -4.63  Std =    49.55

 - Packet number: 18 starting at 2023-05-27T17:53:26.177.028.925 (XO: 14.177030562 - 14.177047562, t=40.4 degC)
   TM values: Max =   116  Min =  -116  Ave =    -4.44  Std =    48.50
