-- OK --
-----------------> BLESK ver.2021-02-27 run on Sat May 27 23:12:40 2023 UTC
-- File 20230527_134414_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 8.906617 to 44.787417
-each separately from 8s to 44s and from 143627us to 906906us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.236721 in 1 packets: 18
      22.643212 in 1 packets: 36
MIU packet time from 2023-05-27T13:44:25.906.617.000 to 2023-05-27T13:45:01.787.417.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.236721 in 1 packets: 18
      22.643212 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 10.2158981 to 46.2157715
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      13.9999507 in 1 packets: 18
      21.9999227 in 1 packets: 36
Temperature FPGA: 
      02 -> 52.4 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      01 -> 53.7 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C3 -> 40.4 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      3 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Number of generated alerts: 
      0 in 1 packets: 1
      1 in 18 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19
      2 in 18 packets: 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, 37
      3 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20230527_134414_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      134 in 18 packets (all packets)  -- OK --
   10 MHz counter from 11.1225169 to 11.1228059
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 13.2366710  -- OK for SL --
   MIU time from 2023-05-27T13:44:25.906.617.000 to 2023-05-27T13:44:25.906.906.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 13.236721

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      96 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.3594769 to 24.3597659
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 22.6431350  -- OK for SL --
   MIU time from 2023-05-27T13:44:39.143.627.000 to 2023-05-27T13:44:39.143.916.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 22.643212

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      127 in 18 packets (all packets)  -- OK --
   10 MHz counter from 47.0029009 to 47.0031899
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-05-27T13:45:01.787.128.000 to 2023-05-27T13:45:01.787.417.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 8, 32bit 10MHz counter = 11.122516900 corrected by 40bit 10MHz counter of 10.215898100 obtained at 1Hz counter of 8. Result: 8.906622064
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996400 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-05-27T13:44:25.906.476.725
   Max 1st sample MIU time among all SL packets: 2023-05-27T13:45:01.787.276.725
   Min 1st sample XO time among all SL packets: 8.906481788
   Max 1st sample XO time among all SL packets: 44.787280880
   TM values from -621 to 356 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-05-27T13:44:25.906.476.725 (XO: 8.906481788 - 8.906787789, t=40.4 degC)  Rank: 00134
   TM values: Max =   356  Min =  -621  Ave =    -4.48  Std =    62.95
   Min TM value:-621 at an offset of 121.225 usec -> 2023-05-27T13:44:25.906.597.950 (XO: 8.906603014)
   Max TM value:356 at an offset of 127.238 usec -> 2023-05-27T13:44:25.906.603.962 (XO: 8.906609026)
   Max dTM/dt [1/us]:5.32e+03 at an offset of 121.288 usec -> 2023-05-27T13:44:25.906.598.012 (XO: 8.906603076)
   Min dTM/dt [1/us]:-3.07e+03 at an offset of 158.576 usec -> 2023-05-27T13:44:25.906.635.300 (XO: 8.906640364)

 - Block number: 2 starting at 2023-05-27T13:44:39.143.486.725 (XO: 22.143488327 - 22.143794328, t=40.4 degC)  Rank: 00096
   TM values: Max =   235  Min =  -477  Ave =    -4.78  Std =    35.47
   Min TM value:-477 at an offset of 2.725 usec -> 2023-05-27T13:44:39.143.489.450 (XO: 22.143491052)
   Max TM value:235 at an offset of 115.863 usec -> 2023-05-27T13:44:39.143.602.587 (XO: 22.143604190)
   Max dTM/dt [1/us]:2.59e+03 at an offset of 246.076 usec -> 2023-05-27T13:44:39.143.732.800 (XO: 22.143734403)
   Min dTM/dt [1/us]:-3.41e+03 at an offset of 2.675 usec -> 2023-05-27T13:44:39.143.489.400 (XO: 22.143491002)

 - Block number: 3 starting at 2023-05-27T13:45:01.786.987.725 (XO: 44.786991879 - 44.787297881, t=40.4 degC)  Rank: 00127
   TM values: Max =   344  Min =  -310  Ave =    -4.41  Std =    62.01
   Min TM value:-310 at an offset of 132.713 usec -> 2023-05-27T13:45:01.787.120.437 (XO: 44.787124592)
   Max TM value:344 at an offset of 150.688 usec -> 2023-05-27T13:45:01.787.138.413 (XO: 44.787142567)
   Max dTM/dt [1/us]:2.59e+03 at an offset of 292.701 usec -> 2023-05-27T13:45:01.787.280.426 (XO: 44.787284580)
   Min dTM/dt [1/us]:-3.57e+03 at an offset of 292.826 usec -> 2023-05-27T13:45:01.787.280.551 (XO: 44.787284705)
