-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri May 26 23:00:20 2023 UTC
-- File 20230526_142130_1227.dat, 18  packets found.
APID:
      1227 (SL) in 18 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 18 packets (all packets)
Total number of packets: 
      18 in 18 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 14.540153 to 14.540442
-each separately from 14s to 14s and from 540153us to 540442us.
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU packet time from 2023-05-26T14:21:47.540.153.000 to 2023-05-26T14:21:47.540.442.000
   Differences in neigboring packets: 
      0.000017 in 17 packets (all packets)
MIU orbit number: 
      60000 in 18 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 18 packets (all packets)
MIU orbit information / number of events: 
      0 in 18 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 18 packets (all packets)
Instrument anomalies: 
      0 (OK) in 18 packets (all packets)
FPGA configuration version: 
      211 in 18 packets (all packets)
10 MHz counter at the last 1Hz pulse from 15.7242956 to 15.7242956
   Differences in neigboring packets: 
      0.0000000 in 17 packets (all packets)
Temperature FPGA: 
      02 -> 52.4 degC in 18 packets (all packets)
Temperature XO: 
      C3 -> 40.4 degC in 18 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 18 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 18 packets (all packets)
Number of external events: 
      0 in 18 packets (all packets)
Number of generated alerts: 
      0 in 18 packets (all packets)
Survey memory error: 
      0 in 18 packets (all packets)
Event memory error: 
      0 in 18 packets (all packets)
Configuration error: 
      0 in 18 packets (all packets)
SDRAM CRC error: 
      0 in 18 packets (all packets)
Radio gain: 
      0 in 18 packets (all packets)
Radio bandwidth: 
      0 in 18 packets (all packets)
Digital gain: 
      0 in 18 packets (all packets)
Analog gain: 
      0 in 18 packets (all packets)
 
SURVEY DATA READER
-- File 20230526_142130_1227.dat, 1 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      67 in 18 packets (all packets)  -- OK --
   10 MHz counter from 16.2644500 to 16.2647390
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-05-26T14:21:47.540.153.000 to 2023-05-26T14:21:47.540.442.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 18 points based on 14 XO calibration points at 1Hz counter from 0 to 13
   --- First point: 1Hz counter = 14, 32bit 10MHz counter = 16.264450000 corrected by 40bit 10MHz counter of 14.724299100 obtained at 1Hz counter of 13. Result: 14.540156291
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-05-26T14:21:47.540.012.725
   Max 1st sample MIU time among all SL packets: 2023-05-26T14:21:47.540.301.725
   Min 1st sample XO time among all SL packets: 14.540016015
   Max 1st sample XO time among all SL packets: 14.540305016
   TM values from -195 to 152 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-05-26T14:21:47.540.012.725 (XO: 14.540016015 - 14.540322016, t=40.4 degC)  Rank: 00067
   TM values: Max =   152  Min =  -195  Ave =    -4.47  Std =    14.47
   Min TM value:-195 at an offset of 120.613 usec -> 2023-05-26T14:21:47.540.133.337 (XO: 14.540136628)
   Max TM value:152 at an offset of 125.638 usec -> 2023-05-26T14:21:47.540.138.362 (XO: 14.540141653)
   Max dTM/dt [1/us]:1.24e+03 at an offset of 123.150 usec -> 2023-05-26T14:21:47.540.135.875 (XO: 14.540139165)
   Min dTM/dt [1/us]:-1.62e+03 at an offset of 120.500 usec -> 2023-05-26T14:21:47.540.133.225 (XO: 14.540136515)
