-- OK --
-----------------> BLESK ver.2021-02-27 run on Thu May 25 23:44:51 2023 UTC
-- File 20230525_140324_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 0.787581 to 17.113739
-each separately from 0s to 17s and from 113450us to 787870us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.325580 in 1 packets: 18
MIU packet time from 2023-05-25T14:03:27.787.581.000 to 2023-05-25T14:03:44.113.739.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.325580 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 1.9356322 to 18.9355716
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999394 in 1 packets: 18
Temperature FPGA: 
      03 -> 51.1 degC in 36 packets (all packets)
Temperature XO: 
      C2 -> 38.8 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Number of generated alerts: 
      0 in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      1 in 3 packets: 19, 20, 21
      2 in 15 packets: 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230525_140324_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      68 in 18 packets (all packets)  -- OK --
   10 MHz counter from 2.7232145 to 2.7235035
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.3255190  -- OK for SL --
   MIU time from 2023-05-25T14:03:27.787.581.000 to 2023-05-25T14:03:27.787.870.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.325580

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      209 in 18 packets (all packets)  -- OK --
   10 MHz counter from 19.0490225 to 19.0493115
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-05-25T14:03:44.113.450.000 to 2023-05-25T14:03:44.113.739.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 0, 32bit 10MHz counter = 2.723214500 corrected by 40bit 10MHz counter of 1.935632200 obtained at 1Hz counter of 0. Result: 0.787585057
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996500 - 0.999996500
   Min 1st sample MIU time among all SL packets: 2023-05-25T14:03:27.787.440.725
   Max 1st sample MIU time among all SL packets: 2023-05-25T14:03:44.113.598.725
   Min 1st sample XO time among all SL packets: 0.787444781
   Max 1st sample XO time among all SL packets: 17.113600023
   TM values from -701 to 654 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-05-25T14:03:27.787.440.725 (XO: 0.787444781 - 0.787750782, t=38.8 degC)  Rank: 00068
   TM values: Max =   469  Min =  -214  Ave =    -4.36  Std =    16.34
   Min TM value:-214 at an offset of 124.163 usec -> 2023-05-25T14:03:27.787.564.887 (XO: 0.787568944)
   Max TM value:469 at an offset of 123.775 usec -> 2023-05-25T14:03:27.787.564.500 (XO: 0.787568556)
   Max dTM/dt [1/us]:2.84e+03 at an offset of 123.675 usec -> 2023-05-25T14:03:27.787.564.400 (XO: 0.787568456)
   Min dTM/dt [1/us]:-2.52e+03 at an offset of 123.838 usec -> 2023-05-25T14:03:27.787.564.562 (XO: 0.787568619)

 - Block number: 2 starting at 2023-05-25T14:03:44.113.309.725 (XO: 17.113311022 - 17.113617023, t=38.8 degC)  Rank: 00209
   TM values: Max =   654  Min =  -701  Ave =    -4.06  Std =   114.09
   Min TM value:-701 at an offset of 244.876 usec -> 2023-05-25T14:03:44.113.554.600 (XO: 17.113555897)
   Max TM value:654 at an offset of 244.488 usec -> 2023-05-25T14:03:44.113.554.213 (XO: 17.113555510)
   Max dTM/dt [1/us]:5.07e+03 at an offset of 244.951 usec -> 2023-05-25T14:03:44.113.554.675 (XO: 17.113555972)
   Min dTM/dt [1/us]:-5.33e+03 at an offset of 234.338 usec -> 2023-05-25T14:03:44.113.544.063 (XO: 17.113545360)
