-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon May 22 21:52:58 2023 UTC
-- File 20230522_120650_1227.dat, 108  packets found.
APID:
      1227 (SL) in 108 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 108 packets (all packets)
Total number of packets: 
      18 in 108 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 10.519660 to 80.573132
-each separately from 10s to 80s and from 333340us to 823205us.
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.889211 in 1 packets: 18
      9.923891 in 1 packets: 36
      26.353726 in 1 packets: 54
      7.135272 in 1 packets: 72
      14.749638 in 1 packets: 90
MIU packet time from 2023-05-22T12:07:03.519.660.000 to 2023-05-22T12:08:13.573.132.000
   Differences in neigboring packets: 
      0.000017 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.889211 in 1 packets: 18
      9.923891 in 1 packets: 36
      26.353726 in 1 packets: 54
      7.135272 in 1 packets: 72
      14.749638 in 1 packets: 90
MIU orbit number: 
      60000 in 108 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 108 packets (all packets)
MIU orbit information / number of events: 
      0 in 108 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 108 packets (all packets)
Instrument anomalies: 
      0 (OK) in 108 packets (all packets)
FPGA configuration version: 
      211 in 108 packets (all packets)
10 MHz counter at the last 1Hz pulse from 11.7442409 to 81.7439865
   Differences in neigboring packets: 
      0.0000000 in 102 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      11.9999563 in 1 packets: 18
      9.9999637 in 1 packets: 36
      25.9999056 in 1 packets: 54
      6.9999745 in 1 packets: 72
      14.9999455 in 1 packets: 90
Temperature FPGA: 
      04 -> 49.8 degC in 54 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      03 -> 51.1 degC in 54 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 108 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 108 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 108 packets (all packets)
Number of external events: 
      1 in 37 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, ...
      2 in 71 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54, 55, ...
Number of generated alerts: 
      1 in 1 packets: 1
      2 in 36 packets: 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18, 19, ...
      3 in 17 packets: 38, 39, 40, 41, 42, 43, 44, 45, 46, 47, 48, 49, 50, 51, 52, 53, 54
      4 in 54 packets: 55, 56, 57, 58, 59, 60, 61, 62, 63, 64, 65, 66, 67, 68, 69, 70, 71, 72, ...
Survey memory error: 
      0 in 108 packets (all packets)
Event memory error: 
      0 in 108 packets (all packets)
Configuration error: 
      0 in 108 packets (all packets)
SDRAM CRC error: 
      0 in 108 packets (all packets)
Radio gain: 
      0 in 108 packets (all packets)
Radio bandwidth: 
      0 in 108 packets (all packets)
Digital gain: 
      0 in 108 packets (all packets)
Analog gain: 
      0 in 108 packets (all packets)
 
SURVEY DATA READER
-- File 20230522_120650_1227.dat, 6 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      104 in 18 packets (all packets)  -- OK --
   10 MHz counter from 12.2639019 to 12.2641909
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 11.8891670  -- OK for SL --
   MIU time from 2023-05-22T12:07:03.519.660.000 to 2023-05-22T12:07:03.519.949.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 11.889211

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      55 in 18 packets (all packets)  -- OK --
   10 MHz counter from 24.1533579 to 24.1536469
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 9.9238550  -- OK for SL --
   MIU time from 2023-05-22T12:07:15.409.160.000 to 2023-05-22T12:07:15.409.449.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 9.923891

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      167 in 18 packets (all packets)  -- OK --
   10 MHz counter from 34.0775019 to 34.0777909
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 26.3536310  -- OK for SL --
   MIU time from 2023-05-22T12:07:25.333.340.000 to 2023-05-22T12:07:25.333.629.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 26.353726

 - Block number: 4 from the sequence of 18 packets.
   Block numbers in the data:
      4 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      59 in 18 packets (all packets)  -- OK --
   10 MHz counter from 60.4314219 to 60.4317109
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.1352470  -- OK for SL --
   MIU time from 2023-05-22T12:07:51.687.355.000 to 2023-05-22T12:07:51.687.644.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.135272

 - Block number: 5 from the sequence of 18 packets.
   Block numbers in the data:
      5 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      56 in 18 packets (all packets)  -- OK --
   10 MHz counter from 67.5669579 to 67.5672469
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 14.7495830  -- OK for SL --
   MIU time from 2023-05-22T12:07:58.822.916.000 to 2023-05-22T12:07:58.823.205.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 14.749638

 - Block number: 6 from the sequence of 18 packets.
   Block numbers in the data:
      6 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      57 in 18 packets (all packets)  -- OK --
   10 MHz counter from 82.3168299 to 82.3171189
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-05-22T12:08:13.572.843.000 to 2023-05-22T12:08:13.573.132.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 108 points based on 89 XO calibration points at 1Hz counter from 0 to 88
   --- First point: 1Hz counter = 10, 32bit 10MHz counter = 12.263901900 corrected by 40bit 10MHz counter of 11.744240900 obtained at 1Hz counter of 10. Result: 10.519662923
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2023-05-22T12:07:03.519.519.725
   Max 1st sample MIU time among all SL packets: 2023-05-22T12:08:13.572.991.725
   Min 1st sample XO time among all SL packets: 10.519522647
   Max 1st sample XO time among all SL packets: 80.572994188
   TM values from -928 to 412 median:-4.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-05-22T12:07:03.519.519.725 (XO: 10.519522647 - 10.519828648, t=37.2 degC)  Rank: 00104
   TM values: Max =   412  Min =  -647  Ave =    -4.19  Std =    72.49
   Min TM value:-647 at an offset of 122.113 usec -> 2023-05-22T12:07:03.519.641.837 (XO: 10.519644760)
   Max TM value:412 at an offset of 289.426 usec -> 2023-05-22T12:07:03.519.809.151 (XO: 10.519812073)
   Max dTM/dt [1/us]:4.12e+03 at an offset of 289.226 usec -> 2023-05-22T12:07:03.519.808.951 (XO: 10.519811873)
   Min dTM/dt [1/us]:-5.43e+03 at an offset of 122.063 usec -> 2023-05-22T12:07:03.519.641.787 (XO: 10.519644710)

 - Block number: 2 starting at 2023-05-22T12:07:15.409.019.725 (XO: 22.409021897 - 22.409327899, t=37.2 degC)  Rank: 00055
   TM values: Max =   170  Min =  -152  Ave =    -4.31  Std =    10.81
   Min TM value:-152 at an offset of 120.238 usec -> 2023-05-22T12:07:15.409.139.962 (XO: 22.409142135)
   Max TM value:170 at an offset of 122.438 usec -> 2023-05-22T12:07:15.409.142.162 (XO: 22.409144335)
   Max dTM/dt [1/us]:1.55e+03 at an offset of 122.350 usec -> 2023-05-22T12:07:15.409.142.075 (XO: 22.409144248)
   Min dTM/dt [1/us]:-1.38e+03 at an offset of 124.875 usec -> 2023-05-22T12:07:15.409.144.600 (XO: 22.409146773)

 - Block number: 3 starting at 2023-05-22T12:07:25.333.199.725 (XO: 32.333201958 - 32.333507959, t=37.2 degC)  Rank: 00167
   TM values: Max =   370  Min =  -928  Ave =    -4.51  Std =    42.09
   Min TM value:-928 at an offset of 103.900 usec -> 2023-05-22T12:07:25.333.303.625 (XO: 32.333305858)
   Max TM value:370 at an offset of 105.113 usec -> 2023-05-22T12:07:25.333.304.837 (XO: 32.333307071)
   Max dTM/dt [1/us]:5.83e+03 at an offset of 100.200 usec -> 2023-05-22T12:07:25.333.299.925 (XO: 32.333302158)
   Min dTM/dt [1/us]:-5.22e+03 at an offset of 100.088 usec -> 2023-05-22T12:07:25.333.299.812 (XO: 32.333302046)

 - Block number: 4 starting at 2023-05-22T12:07:51.687.214.725 (XO: 58.687217599 - 58.687523600, t=37.2 degC)  Rank: 00059
   TM values: Max =   178  Min =  -181  Ave =    -4.40  Std =    20.34
   Min TM value:-181 at an offset of 108.350 usec -> 2023-05-22T12:07:51.687.323.075 (XO: 58.687325949)
   Max TM value:178 at an offset of 110.425 usec -> 2023-05-22T12:07:51.687.325.150 (XO: 58.687328024)
   Max dTM/dt [1/us]:1.51e+03 at an offset of 110.325 usec -> 2023-05-22T12:07:51.687.325.050 (XO: 58.687327924)
   Min dTM/dt [1/us]:-1.53e+03 at an offset of 108.250 usec -> 2023-05-22T12:07:51.687.322.975 (XO: 58.687325849)

 - Block number: 5 starting at 2023-05-22T12:07:58.822.775.725 (XO: 65.822779587 - 65.823085588, t=37.2 degC)  Rank: 00056
   TM values: Max =   172  Min =  -173  Ave =    -4.42  Std =    20.13
   Min TM value:-173 at an offset of 99.675 usec -> 2023-05-22T12:07:58.822.875.400 (XO: 65.822879262)
   Max TM value:172 at an offset of 101.738 usec -> 2023-05-22T12:07:58.822.877.462 (XO: 65.822881325)
   Max dTM/dt [1/us]:1.44e+03 at an offset of 142.801 usec -> 2023-05-22T12:07:58.822.918.525 (XO: 65.822922388)
   Min dTM/dt [1/us]:-1.45e+03 at an offset of 101.863 usec -> 2023-05-22T12:07:58.822.877.587 (XO: 65.822881450)

 - Block number: 6 starting at 2023-05-22T12:08:13.572.702.725 (XO: 80.572705187 - 80.573011188, t=37.2 degC)  Rank: 00057
   TM values: Max =   149  Min =  -151  Ave =    -4.42  Std =    11.78
   Min TM value:-151 at an offset of 126.013 usec -> 2023-05-22T12:08:13.572.828.737 (XO: 80.572831200)
   Max TM value:149 at an offset of 123.388 usec -> 2023-05-22T12:08:13.572.826.112 (XO: 80.572828575)
   Max dTM/dt [1/us]:1.21e+03 at an offset of 111.475 usec -> 2023-05-22T12:08:13.572.814.200 (XO: 80.572816662)
   Min dTM/dt [1/us]:-1.31e+03 at an offset of 123.538 usec -> 2023-05-22T12:08:13.572.826.262 (XO: 80.572828725)
