-- OK --
-----------------> BLESK ver.2021-02-27 run on Mon May 22 00:12:20 2023 UTC
-- File 20230521_182905_1227.dat, 36  packets found.
APID:
      1227 (SL) in 36 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 36 packets (all packets)
Total number of packets: 
      18 in 36 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 12.461464 to 29.193896
-each separately from 12s to 29s and from 193607us to 461753us.
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.731854 in 1 packets: 18
MIU packet time from 2023-05-21T18:29:20.461.464.000 to 2023-05-21T18:29:37.193.896.000
   Differences in neigboring packets: 
      0.000017 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.731854 in 1 packets: 18
MIU orbit number: 
      60000 in 36 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 36 packets (all packets)
MIU orbit information / number of events: 
      0 in 36 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 36 packets (all packets)
Instrument anomalies: 
      0 (OK) in 36 packets (all packets)
FPGA configuration version: 
      211 in 36 packets (all packets)
10 MHz counter at the last 1Hz pulse from 13.7575489 to 30.7574860
   Differences in neigboring packets: 
      0.0000000 in 34 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      16.9999371 in 1 packets: 18
Temperature FPGA: 
      04 -> 49.8 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 18 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36
Temperature XO: 
      C1 -> 37.2 degC in 36 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 36 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 36 packets (all packets)
Number of external events: 
      0 in 36 packets (all packets)
Number of generated alerts: 
      0 in 36 packets (all packets)
Survey memory error: 
      0 in 36 packets (all packets)
Event memory error: 
      0 in 36 packets (all packets)
Configuration error: 
      0 in 36 packets (all packets)
SDRAM CRC error: 
      0 in 36 packets (all packets)
Radio gain: 
      0 in 36 packets (all packets)
Radio bandwidth: 
      0 in 36 packets (all packets)
Digital gain: 
      0 in 36 packets (all packets)
Analog gain: 
      0 in 36 packets (all packets)
 
SURVEY DATA READER
-- File 20230521_182905_1227.dat, 2 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      63 in 18 packets (all packets)  -- OK --
   10 MHz counter from 14.2190137 to 14.2193027
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 16.7317910  -- OK for SL --
   MIU time from 2023-05-21T18:29:20.461.464.000 to 2023-05-21T18:29:20.461.753.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 16.731854

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      64 in 18 packets (all packets)  -- OK --
   10 MHz counter from 30.9510937 to 30.9513827
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-05-21T18:29:37.193.607.000 to 2023-05-21T18:29:37.193.896.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 36 points based on 29 XO calibration points at 1Hz counter from 0 to 28
   --- First point: 1Hz counter = 12, 32bit 10MHz counter = 14.219013700 corrected by 40bit 10MHz counter of 13.757548900 obtained at 1Hz counter of 12. Result: 12.461466507
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996300
   Min 1st sample MIU time among all SL packets: 2023-05-21T18:29:20.461.323.725
   Max 1st sample MIU time among all SL packets: 2023-05-21T18:29:37.193.755.725
   Min 1st sample XO time among all SL packets: 12.461326232
   Max 1st sample XO time among all SL packets: 29.193757142
   TM values from -419 to 597 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-05-21T18:29:20.461.323.725 (XO: 12.461326232 - 12.461632233, t=37.2 degC)  Rank: 00063
   TM values: Max =   597  Min =  -419  Ave =    -4.50  Std =    56.76
   Min TM value:-419 at an offset of 102.538 usec -> 2023-05-21T18:29:20.461.426.262 (XO: 12.461428770)
   Max TM value:597 at an offset of 100.525 usec -> 2023-05-21T18:29:20.461.424.250 (XO: 12.461426757)
   Max dTM/dt [1/us]:4.44e+03 at an offset of 248.701 usec -> 2023-05-21T18:29:20.461.572.425 (XO: 12.461574933)
   Min dTM/dt [1/us]:-9.77e+03 at an offset of 100.563 usec -> 2023-05-21T18:29:20.461.424.287 (XO: 12.461426795)

 - Block number: 2 starting at 2023-05-21T18:29:37.193.466.725 (XO: 29.193468141 - 29.193774142, t=37.2 degC)  Rank: 00064
   TM values: Max =   159  Min =  -178  Ave =    -4.54  Std =    49.16
   Min TM value:-178 at an offset of 130.138 usec -> 2023-05-21T18:29:37.193.596.862 (XO: 29.193598279)
   Max TM value:159 at an offset of 129.150 usec -> 2023-05-21T18:29:37.193.595.875 (XO: 29.193597291)
   Max dTM/dt [1/us]:1.07e+03 at an offset of 136.188 usec -> 2023-05-21T18:29:37.193.602.913 (XO: 29.193604329)
   Min dTM/dt [1/us]:-1.07e+03 at an offset of 125.813 usec -> 2023-05-21T18:29:37.193.592.537 (XO: 29.193593954)
