-- OK --
-----------------> BLESK ver.2021-02-27 run on Fri May 19 00:00:17 2023 UTC
-- File 20230518_170221_1227.dat, 54  packets found.
APID:
      1227 (SL) in 54 packets (all packets)
Mode:
      1 (EBEW-SRSL) in 54 packets (all packets)
Total number of packets: 
      18 in 54 packets (all packets)
Packet No from 1 to 18
   Sequence of packet numbers -- OK -- 
Hz and MHz counters from 5.349665 to 34.274909
-each separately from 5s to 34s and from 274620us to 996352us.
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.646109 in 1 packets: 18
      7.278268 in 1 packets: 36
MIU packet time from 2023-05-18T17:02:29.349.665.000 to 2023-05-18T17:02:58.274.909.000
   Differences in neigboring packets: 
      0.000017 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      21.646109 in 1 packets: 18
      7.278268 in 1 packets: 36
MIU orbit number: 
      60000 in 54 packets (all packets)
MIU orbit information / directon: 
      0 (DOWN) in 54 packets (all packets)
MIU orbit information / number of events: 
      0 in 54 packets (all packets)
Waveform overflows in 306us: 
      0 (none) in 54 packets (all packets)
Instrument anomalies: 
      0 (OK) in 54 packets (all packets)
FPGA configuration version: 
      211 in 54 packets (all packets)
10 MHz counter at the last 1Hz pulse from 6.3539335 to 35.3538266
   Differences in neigboring packets: 
      0.0000000 in 51 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 19, ...
      20.9999226 in 1 packets: 18
      7.9999705 in 1 packets: 36
Temperature FPGA: 
      02 -> 52.4 degC in 18 packets: 1, 2, 3, 4, 5, 6, 7, 8, 9, 10, 11, 12, 13, 14, 15, 16, 17, 18
      03 -> 51.1 degC in 36 packets: 19, 20, 21, 22, 23, 24, 25, 26, 27, 28, 29, 30, 31, 32, 33, 34, 35, 36, ...
Temperature XO: 
      C1 -> 37.2 degC in 54 packets (all packets)
Temperature ant 3: 
      01 -> 108.0 degC in 54 packets (all packets)
Temperature ant 4: 
      01 -> 108.0 degC in 54 packets (all packets)
Number of external events: 
      0 in 54 packets (all packets)
Number of generated alerts: 
      0 in 54 packets (all packets)
Survey memory error: 
      0 in 54 packets (all packets)
Event memory error: 
      0 in 54 packets (all packets)
Configuration error: 
      0 in 54 packets (all packets)
SDRAM CRC error: 
      0 in 54 packets (all packets)
Radio gain: 
      0 in 54 packets (all packets)
Radio bandwidth: 
      0 in 54 packets (all packets)
Digital gain: 
      0 in 54 packets (all packets)
Analog gain: 
      0 in 54 packets (all packets)
 
SURVEY DATA READER
-- File 20230518_170221_1227.dat, 3 SL data blocks found. Extra packets: 0.
--
Data structure of blocks of survey data packets:

 - Block number: 1 from the sequence of 18 packets.
   Block numbers in the data:
      1 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      56 in 18 packets (all packets)  -- OK --
   10 MHz counter from 6.7035997 to 6.7038887
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 21.6460310  -- OK for SL --
   MIU time from 2023-05-18T17:02:29.349.665.000 to 2023-05-18T17:02:29.349.954.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 21.646109

 - Block number: 2 from the sequence of 18 packets.
   Block numbers in the data:
      2 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      53 in 18 packets (all packets)  -- OK --
   10 MHz counter from 28.3499197 to 28.3502087
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
      Difference to the following block: 7.2782390  -- OK for SL --
   MIU time from 2023-05-18T17:02:50.996.063.000 to 2023-05-18T17:02:50.996.352.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
      Difference to the following block: 7.278268

 - Block number: 3 from the sequence of 18 packets.
   Block numbers in the data:
      3 in 18 packets (all packets)  -- OK --
   Ranks for SL, 0 for SR and SA:
      59 in 18 packets (all packets)  -- OK --
   10 MHz counter from 35.6284477 to 35.6287367
      Difference between neigboring packets: 
      0.0000170 in 17 packets (all packets)  -- OK --
   MIU time from 2023-05-18T17:02:58.274.620.000 to 2023-05-18T17:02:58.274.909.000
      Difference between neigboring packets: 
      0.000017 in 17 packets (all packets)
   --- Rel. time correction for 54 points based on 44 XO calibration points at 1Hz counter from 0 to 43
   --- First point: 1Hz counter = 5, 32bit 10MHz counter = 6.703599700 corrected by 40bit 10MHz counter of 6.353933500 obtained at 1Hz counter of 5. Result: 5.349667459
--
   MIU or XO time of the 1st SL sample = header (MIU or corrected XO) time +offset(if negative) - 275 ns delay
   Samplig frequency correction factor in all SL packets: 0.999996300 - 0.999996400
   Min 1st sample MIU time among all SL packets: 2023-05-18T17:02:29.349.524.725
   Max 1st sample MIU time among all SL packets: 2023-05-18T17:02:58.274.768.725
   Min 1st sample XO time among all SL packets: 5.349527183
   Max 1st sample XO time among all SL packets: 34.274770842
   TM values from -292 to 257 median:-5.00000
--
Extremes and waveform characteristics in blocks of SL data packets:

 - Block number: 1 starting at 2023-05-18T17:02:29.349.524.725 (XO: 5.349527183 - 5.349833184, t=37.2 degC)  Rank: 00056
   TM values: Max =   193  Min =  -247  Ave =    -4.44  Std =    36.17
   Min TM value:-247 at an offset of 127.213 usec -> 2023-05-18T17:02:29.349.651.937 (XO: 5.349654396)
   Max TM value:193 at an offset of 126.925 usec -> 2023-05-18T17:02:29.349.651.650 (XO: 5.349654109)
   Max dTM/dt [1/us]:2.16e+03 at an offset of 129.775 usec -> 2023-05-18T17:02:29.349.654.500 (XO: 5.349656959)
   Min dTM/dt [1/us]:-1.83e+03 at an offset of 127.038 usec -> 2023-05-18T17:02:29.349.651.762 (XO: 5.349654221)

 - Block number: 2 starting at 2023-05-18T17:02:50.995.922.725 (XO: 26.995927010 - 26.996233011, t=37.2 degC)  Rank: 00053
   TM values: Max =   235  Min =  -274  Ave =    -4.37  Std =    42.32
   Min TM value:-274 at an offset of 118.413 usec -> 2023-05-18T17:02:50.996.041.137 (XO: 26.996045423)
   Max TM value:235 at an offset of 105.913 usec -> 2023-05-18T17:02:50.996.028.637 (XO: 26.996032923)
   Max dTM/dt [1/us]:2.02e+03 at an offset of 121.113 usec -> 2023-05-18T17:02:50.996.043.837 (XO: 26.996048123)
   Min dTM/dt [1/us]:-2.24e+03 at an offset of 106.025 usec -> 2023-05-18T17:02:50.996.028.750 (XO: 26.996033035)

 - Block number: 3 starting at 2023-05-18T17:02:58.274.479.725 (XO: 34.274481841 - 34.274787842, t=37.2 degC)  Rank: 00059
   TM values: Max =   257  Min =  -292  Ave =    -4.49  Std =    33.13
   Min TM value:-292 at an offset of 114.213 usec -> 2023-05-18T17:02:58.274.593.937 (XO: 34.274596054)
   Max TM value:257 at an offset of 113.913 usec -> 2023-05-18T17:02:58.274.593.637 (XO: 34.274595754)
   Max dTM/dt [1/us]:1.80e+03 at an offset of 124.488 usec -> 2023-05-18T17:02:58.274.604.212 (XO: 34.274606329)
   Min dTM/dt [1/us]:-2.37e+03 at an offset of 114.025 usec -> 2023-05-18T17:02:58.274.593.750 (XO: 34.274595866)
